Phase contrast flow imaging in Cardiac Magnetic Resonance (CMR): Imaging turbulent blood flow
KeywordsCardiac Magnetic Resonance Flow Volume Pulmonary Valve Retrograde Flow Blood Flow Volume
Accurately measuring blood flow volume is paramount in patients with arterial valve disease. Flow measurements are important components of a CMR report and provide the referring clinicians with valuable information from which to base further clinical management. There are many factors which influence the quality and reliability of flow data including; slice positioning, velocity encoding mismatching, temporal resolution, and turbulence.
Antegrade and retrograde flow volumes, Anatomical slice positions
Degree of antegrade/retrograde turbulence
Variation in antegrade flow = Max - Min antegrade flow volume
Variation in regurgitant fraction = Max- Min calculated regurgitant fraction
The ratio of variation in flow volume to the mean antegrade flow for each patient was calculated.
The mean antegrade flow volume variation was 11.2 ml (13.1% of mean antegrade flow).
Mean retrograde flow volume variation was 7.1 ml (33% of mean retrograde flow).
The mean variation in regurgitant fraction was 7.4%.
These results show a significant variation in estimated antegrade and retrograde flow volume depending on the position across the valve at which the data was acquired. This would have a significant impact on any subsequent interventional decisions.
When turbulent flow is present no single slice position can be assumed to give an accurate result. Typically flow distal to the valve underestimates antegrade flow, and retrograde flow is underestimated proximal to the valve. Assuming the maximum measured antegrade and retrograde flow is closest to the true flow it would be appear to be more accurate to obtain a stack of phase contrast images through the valve to establish the true antegrade and retrograde values, and therefore better define the regurgitation fraction
National Institute of Health Research, Biomedical Research Unit.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.