Structure Design and Motion Simulation of a Microprocessor-Controlled Prosthetic Knee
Prosthetic knee is the most important component of lower limb prosthesis. This work aims to design a novel knee joint prosthesis and provide theoretical model of the hydraulic microprocessor-controlled prosthetic knee. The intelligent knee prosthesis with hydraulic damper is designed. Innovative valve structure is proposed to realize damping adjustment with single motor. The dynamics model of the lower limb prosthesis is established. Motion simulation is done to verify the correctness of the knee joint structure. The flexion and extension damping can be adjusted continuously and independently. The simulation shows that the motion of the knee joint is steady. It means that the structure of the knee joint prosthesis is reasonable.
KeywordsProsthesis Knee joint Damping adjustment Motion simulation
The work reported in this paper is supported by National Natural Science Foundation of China, number: 61473193 and Shanghai Engineering Research Center of Assistive Devices, number: 15DZ2251700.
- 1.Shirsath VB, Dongare MP (2017) Neural network based gait phases of above knee prosthesis. In: IEEE international conference on advances in electronics, communication and computer technology. IEEEGoogle Scholar
- 3.Yu HL, Shen L, Hu JH et al (2009) Design of electronically controlled hydraulic damper for prosthetic knee. J Clin Rehab Tissue Eng Res 13(39):7635–7638Google Scholar
- 4.Wei L (2012) Working mechanism research and prototype design of intelligent prosthetic knee joint. Southeast UniversityGoogle Scholar
- 5.Lambrecht et al. (2009) Design of a semi-active knee prosthesis. IEEE international conference on robotics and automation. IEEE, pp 639–645Google Scholar
- 6.Zhang F, Liu M, Huang H (2012) Preliminary study of the effect of user intent recognition errors on volitional control of powered lower limb prostheses. Annual international conference of the IEEE engineering in medicine and biology society, pp 2768–2771Google Scholar
- 8.Budaker B (2012) Active driven prosthesis using a bevel helical gearbox in combination with a brushless DC-motor. Biomed Technol 57Google Scholar
- 11.Pejhan S, Farahmand F, Parnianpour M (2008) Design optimization of an above-knee prosthesis based on the kinematics of gait. Engineering in Medicine and Biology Society, 2008. EMBS 2008. International Conference of the IEEE. IEEE, pp 4274–4277Google Scholar
- 12.Fu H, Zhang X, Wang X et al (2016) A novel prosthetic knee joint with a parallel spring and damping mechanism. Int J Adv Rob Syst 13(4):1729881416658174Google Scholar