The stationary magnetic field B 0 of an MR scanner can be extremely strong: Superconducting magnets easily attain field strengths of 1.5–3.0 T (15,000–30,000 Gauss). This magnetic field bears great risks for both female and male patients, as well as for the staff operating the magnet. Ferromagnetic items in the vicinity of the magnet are particularly hazardous. They can become dangerous missiles (“missile effect”).
KeywordsStationary Magnetic Field Cochlear Implant Dental Prosthesis Ultra High Field Magnetic Field Interaction
Unable to display preview. Download preview PDF.
- Elster AD, Link KM, Carr JJ (1994) Patient screening prior to MR imaging: a practical approach synthesized from protocols at 15 U.S. medical centers. Am J Roentgenol 162: 195Google Scholar
- Hilfiker PR, Weishaupt D, Debatin JF (2002) Intravascular implants: safety and artifacts. In: Arlart IP, Bongartz GM, Marchal G (eds) Magnetic resonance angiography, 2nd edn. Springer, Berlin Heidelberg New York, p 454Google Scholar
- Shellock FG (2001) Reference manual for magnetic resonance safety: 2002 edition,1st edn. Amirsys, Salt Lake City, UtahGoogle Scholar