The tests for both contamination monitor and exposure meter are similar. They aim to check for obvious physical damage, battery failure and the response to a known activity. Operational checks should check the probe for damage, the state of the battery; and the measured count rate or exposure rate, for a known activity standard source, against expected count rate for contamination and exposure rate for the exposure meter. Routine tests should involve calibration of the scale against known activities and exposure rates respectively; and often have to be undertaken by authorised institutions.
KeywordsSyringe Radionuclide IAEA Cali
full width at half maximum
pulse height analyser
Unable to display preview. Download preview PDF.
- Fleming J, Williams N, Skrypniuk J, Thorley P, Rose M, Driver I (2002) Report 85 radioactive sample counting — principles and practice. Institute of Physics and Engineering in Medicine, YorkGoogle Scholar
- IAEA (1991) Tecdoc 602 quality control of nuclear medicine instruments. International Atomic Energy Agency, ViennaGoogle Scholar
- IEC (1992) International Electrotechnical Commission publication IEC61145 calibration and usage of ionization chamber systems for assay of radionuclides. IEC, GenevaGoogle Scholar
- IEC (1994) International Electrotechnical Commission publication IEC 1303 medical electrical equipment: particular methods for declaring the performance of radionuclide calibrators. IEC, GenevaGoogle Scholar
- IPSM (1992) Report 65 protocol for establishing and maintaining the calibration of medical radionuclide calibrators and their quality control. Institute of Physical Sciences in Medicine, YorkGoogle Scholar
- Todd JH, Short MD, Bessent RG, Hawkins LA, Leach KG, Ottewell D, Morgan DW (1980) HPA topic group report 33 an introduction to automatic radioactive sample counters. Hospital Physicists’ Association, LondonGoogle Scholar
- Zanzonico P, Heller S (2000) The intraoperative gamma probe: basic principles and choices available. Semin Nucl Med XXX: 33–48Google Scholar