Skin-Friction Measurements in Two- and Three-Dimensional Highly Turbulent Flows with Separation
This paper presents comparative skin-friction measurements obtained by a Preston tube, a surface fence, a single film and a McCROSKEY hot-film probe, and a wall pulsed wire in two- and three-dimensional highly turbulent flows (near wall turbulence level > 40%) with separation and instantaneous reverse flow. Fluctuating values of skin friction were obtained by the wall pulse wire and the hot films in addition to the mean values and the direction of the wall stream line. Differences in the results of the measuring techniques are explained and the failure of devices under these severe conditions is displayed.
KeywordsPolyethylene Tungsten Verse Dianat
Unable to display preview. Download preview PDF.
- 2.R.B. Ginder, L.J.S. Bradbury: Preliminary investigation of a pulsed-gauge technique for skin friction measurements in highly turbulent flow. ARC Paper 34 448, 1973.Google Scholar
- 3.J.K. Eaton, J.P. Johnston: Turbulent flow reattachment. An experimental study of the flow and structure behind a backward-facing step. Rep. MD-39 Stanford Univ., 1980.Google Scholar
- 5.P. Dengel, H.H. Fernholz, J.-D. Vagt: Turbulent and mean flow measurement in an incompressible axisymmetric boundary layer with incipient separation. In Turbulent Shear Flows 3, ed. Bradbury et al. Springer Verlag, 1982.Google Scholar
- 7.I.P. Castro, M. Dianat: The measurement of skin friction with a pulsed-wire gauge. Final report on University of Surrey/ RAE agreement 2064/069 XR Aero.Google Scholar
- 8.P.H. Alfredson, A.V. Johansson, J.H. Haritonidis, H. Eckelmann: On the fluctuating wall shear stress and the velocity field in the viscous sublayer. Submitted to J. Fluid Mech., 1986.Google Scholar
- 9.The McCroskey probe was kindly provided by Dr. H. U. Meier of DFVLR Göttingen.Google Scholar
- 10.J.-D. Vagt, H.H. Fernholz: Use of surface fences to measure wall shear stress in three-dimensional boundary layers. Aero. Quart. 24, 87–91, 1973.Google Scholar
- 11.E. Froebel, J.-D. Vagt: Ein automatisch abgleichendes Flüssigkeitsmanometer mit digitaler Anzeige. DLR-FB 74–10, 1974.Google Scholar
- 14.X. Gu: Diplomarbeit am Hermann-Föttinger-Institut für Thermo-und Fluiddynamik, Technische Universität Berlin.Google Scholar