Closed-Loop Control of Artificial Ventilation
Since 1956, researchers have developed numerous closed-loop controllers to adjust mechanical ventilation [1, 2]. In almost all systems, the end-tidal CO2 is measured on a breath-to-breath basis and the ventilator’s tidal volume or respiratory rate is adjusted using classical PID control. These systems reach the set end-tidal CO2 in less than lmin and maintain the end-tidal CO2 within ±0.1 vol% of the set point. The variability is less than half that seen during periods of open-loop control . Closed-loop control has also been used to adjust the inspired oxygen concentration to maintain the desired arterial oxygen concentration in neonates . Once again, the closed-loop controller out-performs the human operator.
KeywordsDioxide Monit Frumin
Unable to display preview. Download preview PDF.
- 4.Morozoff PE, Evans RW (1992) Closed-loop control of Sa02 in the neonate. Biomed Instr Tech 26: 117–123Google Scholar
- 11.East TD, Tolle CR, Farrell RM, Brunner JX (1991) A non-linear closed-loop controller for oxygenation based on a clinically proven fifth-dimensional quality surface. Crit Care Med 19: S61Google Scholar