In Vivo Testing of a Roller-Screw Type Electric Total Artificial Heart
Work was begun on the development of an electric motor drive total artificial heart at the Pennsylvania State University in 1978. The first systems developed utilized a low-speed high-torque brushless DC motor rotating a triple-track cam mechanism. This mechanism translated a rotary force into a rectilinear motion to actuate alternately the sac type blood pumps located on either end of the motor drive mechanism. The prototype of this system weighed slightly over 1 kg. Implantation of the cam type electric motor driven artificial heart began in 1983 and 222-day calf survival was obtained late in that year . In 1984, work was begun on a roller-screw type electric motor driven artificial heart. The roller-screw device had the advantages of being smaller, lighter, and less expensive to produce. Extensive in vitro testing of the roller-screw device was performed from 1984 through 1986. In mid-1986 implantation of the roller-screw device was begun in calves. Over the next year the device was implanted in six animals with a maximum survival time of greater than 85 days.
KeywordsBlood Pump Artificial Heart Battery Pack Secondary Coil Total Artificial Heart
Unable to display preview. Download preview PDF.
- 3.Rosenberg G, Donachy JH, Geselowitz D, Snyder A, Landis DL, Pierce WS (1982) Recent progress of the artificial heart. Eur Rev Biomed Technol 4:283–284Google Scholar
- 4.Richenbacher WE, Rosenberg G, Landis DL, Weiss WJ, Donachy JH, Pierce WS (1983) Development of an implantable electric motor artificial heart. Surg Forum 34:250–253Google Scholar
- 5.Rosenberg G, Cleary TJ, Snyder AJ, Landis DL, Gaselowitz DB, Pierce WS (1985) A totally implantable artificial heart design. Transactions of the American society of mechanical engineers, 85-WA/DE-11Google Scholar
- 6.Rosenberg G, Snyder A, Pierce WS, Geselowitz D (1982) Engineering development of the mechanical heart. In: McNeil BJ, Cravalho EG (eds) Critical issues in medical technology. Cravalho. Auburn House, Boston, pp 381–397Google Scholar
- 7.Gaines WE, Donachy JH, Rosenberg G, Landis DL, Pierce WS (1984) Studies leading to an artificial heart for clinical application. Contemp Surg 24:41–48Google Scholar
- 8.Jorge E, Snyder AJ, Rosenberg G, Fehr D, Geselowitz DB, Pierce WS (1986) Subcutaneous access port for the compliance chamber of the electric ventricular assist device and total artificial heart. In: Nose Y, Kjellstrand C, Ivanovich P (eds) Progress in artificial organ. Isao, Anaheim, pp 537–540Google Scholar
- 11.Wisman CB, Rosenberg G, Weiss WJ, Landis DL, Donachy JH, Snyder AJ, Richenbacher WE, Pierce WS (1983) The development and successful application of an intrathoracic compliance chamber for the implantable electric motor-driven ventricular assist pump. Surg Forum 34:253–256Google Scholar
- 12.Phillips WM, Donachy JH, Rosenberg G, Pierce WS (1980) The use of segmented polyurethanes in ventricular assist devices and artificial hearts. In: Szycher M, Robinson W (eds) Synthetic biomedical polymers; concepts and practice. Technomic, Westport, pp 39–57Google Scholar
- 13.Snyder A, Rosenberg G, Landis D (1985) Indirect estimation of circulatory pressures for control of an electric motor-driven total artificial heart. In: Langrana NA (ed) 1985 advances in bioengineering. The American society of mechanical engineers, New York, pp 87–88Google Scholar
- 14.Snyder AJ, Rosenberg G, Landis DL, Weiss W, Pierce WS (1984) Introductory lecture on control. Presented at the second world symposium on the artificial heart, July 3, Berlin, pp 167–190Google Scholar