Surface and Volume Resistivity of Pyrex Glass Used for Liquid Membrane Ion-Sensitive Microelectrodes
The essence of a liquid membrane ion-sensitive microelectrode is a column of sensor lodged in the tip of a glass micropipette. For it to function well, the electrical resistivity of the surface of the glass crmust be high, and so must the resistance through the wall of the pipette. The behavior of a fine-tipped microelectrode depends on many factors, and direct examination of electrode responses has led to conflicting ideas about leaks (e.g., [11, 17]). We have therefore made measurements of resistance on the most commonly used glass, Pyrex (Corning 7740), on samples of defined geometry (tubes), and with the atmospheric contamination of a typical physiological laboratory. In addition, we have drawn on literature describing the use of new analytical techniques for studying glass surfaces, much of this research having been done with a view to applications in capillary column chromatography or fiber optics. We will suggest that the physiologist’s problems are confined to the extreme tip of the microelectrode, that they are unlikely to be due to hydration of the glass, and that they might be partly due to cracks.
KeywordsHydration Hydroxyl Silane Foam Shipping
Unable to display preview. Download preview PDF.
- 2.Ammann D, Morf WE, Anker P, Meier PC, Pretsch E, Simon W (1983) Neutral carrier based ion-selective electrodes. Ion-Selective Electrode Rev 5: 3–92Google Scholar
- 4.Coles J A, Orkand RK, Munoz JL (1983) When the photoreceptors in the retina of the honeybee drone are stimulated, K+ activity in the glial cells rises more than Na+ activity falls. Experientia 39:630Google Scholar
- 5.Hulanicki A, Lewandowski R (1974) Some properties of ion-selective electrodes based on poly (vinyl chloride) membranes with liquid-ion-exchanger. Chemia Anal (Warsaw) 19: 53–61Google Scholar
- 6.Khuri RN, Agulian SK (1981) Intracellular electrochemical potentials: skeletal muscle vs epithelial, steady-state vs kinetics. In: Syková E, Hnik P, Vyclický L (eds) Ion-selective microelec- trodes and their use in excitable tissues. Plenum, New York pp 67–89Google Scholar
- 10.Levy S (1979) Mesure du Ca intracellulaire libre dans les photorécepteurs du faux-bourdon (Apis mellifera) au moyen de microélectrodes sélectives au Ca: effets de la photostimulation. M. Sc. Thesis, Geneva UniversityGoogle Scholar
- 13.Oehme M, Kessler M, Simon W (1976) Neutral carrier Ca2+- microelectrode. Chimia 30: 204–206Google Scholar
- 15.Thomas RC (1978) Ion-sensitive intracellular microelectrodes. Academic, LondonGoogle Scholar
- 18.Wiederhorn SM, Evans AG, Roberts DE (1974) A fracture mechanics study of the Skylab windows. In: Bradt RC, Hasselman DPH, Lange FF (eds) Fracture mechanics of ceramics, vol 2. Plenum, New York, pp 829–841Google Scholar