CO2 DIAL Sensitivity Studies for Measurements of Atmospheric Trace Gases
There is great need for atmospheric trace constituent measurements with higher resolution than attainable with passive radiometers. Infrared (IR) differential absorption lidar (DIAL), which depends on Mie scattering from aerosols, has special advantages for tropospheric and lower stratospheric applications and has great potential importance for measurements from Shuttle and aircraft (refs. 1, 2, 3, 4). DIAL data reduction involves comparing large amplitude signals which have small differences. The accuracy of the trace constituent concentration inferred from DIAL measurements depends strongly on the errors in determining the amplitude of the signals. Thus, the commonly used SNR expression (signal divided by noise in the absence of signal) is not adequate to describe DIAL-measurement accuracy and must be replaced by an expression which includes the random coherent (speckle) noise within the signal (refs. 5, 6). A comprehensive DIAL computer algorithm (ref. 7) is modified to include heterodyne and direct detection with speckle noise. Results of studies for monitoring vertical distribution of O3, H2O, and NH3 from ground aircraft and Shuttle platforms are extended to include measurements of ground level plumes with larger than ambient densities, from ground and aircraft platforms. Results are shown for plumes of NH3 but these results can be generalized in terms of the plume optical depth.
KeywordsDirect Detection Speckle Noise Pulse Pair Heterodyne Detection Ground Reflection
Unable to display preview. Download preview PDF.
- 1.Brockman, P; Hess, R.V.; Staton, L.D.; and Bair, C.H.: DIAL with Heterodyne Detection Including Speckle Noise: Aircraft/Shuttle Measurement of O3, H2O and NH3 with Pulsed Tunable Lasers. Heterodyne Systems and Technology Part II, p. 557 NASA CP 2138, March 1980.Google Scholar
- 2.Hardesty, R.M.: A Comparison of Heterodyne and Direct Detection CO2 DIAL Systems for Ground Based Humidity Profiling. NOAA Technical Memorandum ERL WPL-64, October 1980.Google Scholar
- 4.Hess, R.V.: CO2 Lidar for Measurements of Trace Gases and Wind Velocities. SPIE 25th Annual Technical Symposium, Physics and Technology of Coherent Infrared Radar. Paper 300–05, August 1981.Google Scholar
- 11.Killinger, D.K. and Menyuk, N.: Remote Probing of the Atmosphere Using a CO2 DIAL System. IEEE Jl. Quantum Electronics, p. 1917, Sep. 1981.Google Scholar
- 15.Deirmendjian, D.: Electromagnetic Scattering on Spherican Polydispersions. American Elsevier Publishing Co., New York, NY, 1969.Google Scholar
- 16.McClatchey, Robert A. and Shelby, John E. A.: Atmospheric Attentuation of Laser Radiation from 0.76 to 31.25 μm. AFCRL-TR-74–0003, U.S. Air Force, Jan. 1974.Google Scholar
- 17.Hardesty, R.M.: Feasibility of Combined DIAL — Doppler Coherent Lidar Systems for Moisture Flux Measurement. Topical Meeting on Coherent Laser Radar for Atmospheric Sensing, Aspen, July 1980.Google Scholar
- 18.Menzies, R.T. and Chahine, M.T.: Remote Atmospheric Sensing with an Airborne Laser Absorption Spectrometer. Appl. Opt., Vol. 13, no. 12, Dec. 1974.Google Scholar