Development of a Magnetic Refrigerator Operating between 2 K and 10 K
A prototype adiabatic demagnetization refrigerator (ADR) operating between 2 K and 10 K has been built and tested. The refrigerant for the ADR is Gadolinium Gallium Garnet (GGG). In the design thick slices of single-crystal GGG are sandwiched together with strips of high-purity copper. The copper strips are used to transfer heat between the GGG and the 2 K and the 10 K heat switches. The heat switches employed in the ADR have no moving parts. The 10 K heat switch is a helium gas-gap heat switch; while, the 2 K heat switch is a He II-gap heat switch. Preliminary tests indicate that the prototype ADR can provide 27 mW of cooling at 2 K for 165 s; where, the cycle period is 12.6 minutes. The efficiency of the ADR relative to Carnot is 25%. Methods for improving the efficiency of the system are discussed.
KeywordsHeat Sink Heat Load Heat Transfer Area Magnetic Refrigeration Heat Leak
Unable to display preview. Download preview PDF.
- 1.Kashani, A., Helvensteijn, B.P.M., McCormack, F.J., and Spivak, A.L., “Helium Liquid and Gas-Gap Heat switches for Applications at 2 K”, Proceedings of the CEC, (1994).Google Scholar
- 2.Kashani, A., Helvensteijn, B.P.M., McCormack, F.J., and Spivak, A.L., “Helium Liquid and Gas-Gap Heat switches”, Proceedings of the ICCC 7. (1993), pp. 355-370.Google Scholar
- 3.Chan, C.K., “Gas Adsorption/Absorption Heat Switch”, JPL Publication 87-7 (1987).Google Scholar
- 4.Helvensteijn, B.P.M., Kashani, A., and McCormack, F.J., ” Methods for Improving the Transfer of Heat Across the Copper and Gadolinium Gallium Garnet Interface”, Proceeding of the CEC, (1994).Google Scholar
- 5.Lounasmaa, O.V., “Experimental Principles and Methods Below 1K”, Academic Press (1974) p.87.Google Scholar
- 6.Fisher, R.A., Brodale, G.E., Hornung, E.W., Giauque, W.F., “Magnetothermodynamics of Gadolinium Gallium Garnet. I. Heat Capacity, Entropy, Magnetic Moment from 0.5 to 4.2K, with fields to 90 kG, along the  Axis”, Journal of Chemical Physics, Vol. 59 (1973) pp. 4652–4663.ADSCrossRefGoogle Scholar
- 7.Barclay, J.A., Steyert, W.A., “Materials for Magnetic Refrigeration between 2K and 20K”, Cryogenics. (1982) pp. 73–80.Google Scholar