Sensitive Helium Leak Detection in Cryogenic Vacuum Systems
Conventional leak detectors are not useful for cold leak detection of LHe cooled cryogenic vacuum systems since He is adsorbed onto the cold walls. Even though the He desorption leak detection system1 developed at the Continuous Electron Beam Accelerator Facility (CEBAF) is very sensitive, the cryogenic system needs to be warmed up to at least 10.5 K for leak detection purposes, and it still would be impossible to identify the location of the leak. Recently hot cathode extractor gauges have been found to be reliable, very useful and convenient for very low pressure measurement (10-14 torr) at LHe temperatures. This paper discusses the operation of extractor gauges at LHe temperatures and their in situ use for fast and convenient cold leak detection of cryogenic vacuum systems. Further, the location and identification of the leaks based on the principle of the pressure wave travel of He and time-of-flight measurements of the He signal are also discussed.
KeywordsHelium Flange Extractor
Unable to display preview. Download preview PDF.
- 2.M.G. Rao and P. Kneisel, The Use of Extractor Gauge at LHe Temperatures, CEB AF Tech Note #93-057, available from the Librarian, CEBAF, 12000 Jefferson Avenue, Newport News, VA 23606 USA.Google Scholar
- 3.M.G. Rao, Recent Advances in UHV Techniques for Particle Accelerators, CEBAF Tech Note #95-011, available from the Librarian, CEBAF, 12000 Jefferson Avenue, Newport News, VA 23606 USA.Google Scholar
- 4.J.G. Hengevoss, Gas Sorption by Surfaces Cooled to Low Temperatures, Trans. 3rd Int. Vac. Cong., Vol. 1, 51 (1965).Google Scholar
- 7.M.G. Rao, P. Kneisel, and J. Susta, Cryosorption Pumping of H2 and He With Metals and Metal Oxides at 4.3 K, Cryogenics ICEC Supplement 34, 377 (1994).Google Scholar
- 8.M.G. Rao and W.C. Turner, to be published in J. Vac. Sci. Technol. Google Scholar
- 9.M.G. Kaganer, Proc. Acad. Sci. USSR Phys. Chem. Sect. 116, 603 (1957).Google Scholar