Mössbauer Spectrometer with Novel Moving System and Resonant Detection of Gamma Rays
A Mössbauer spectrometer with the collective synchronous motion of the radioactive source and resonant detector has been built. The new special transducer with four drive coils and one velocity pickup coil has been developed. The polyamide fibers serve as suspension brackets, barium ferrite magnets are used. The mechanical construction of transducer allows using different cryostats and furnaces, because the sample is immovable. The resonant detector consists of the thin foil of the organic plastic scintillator with the dissolved substance converting the resonant gamma rays to conversion electrons.
KeywordsFurnace Ferrite Barium Reso
Unable to display preview. Download preview PDF.
- 3.Mitrofanov, K. P., Illarionova, N. V. and Shpinel, Y. S., Prib. Tekhn. Eksp. 30 (1963), 49.Google Scholar
- 5.Maltsev, Y., Mehner, H.,Menzel, M. and Rogozev, B., In: Program and Abstracts, ICAME’99,Garmisch—Partenkirchen, 29 August-03 September 1999, T9/35.Google Scholar
- 6.Evdokimov, V. A., Mashlan, M., Zak, D., Fyodorov, A. A., Kholmetskii, A. L. and Misevich, O. V., Nucl. Instrum. Meth. B 124 (1995), 287.Google Scholar