Noninvasive Vascular Blood Sound Monitoring Through Flexible Microphone
- 41 Downloads
An arteriovenous vascular access is used to provide hemodialysis to people suffering from acute or chronic kidney disease. Monitoring of vascular access is essential to reduce the risk of sudden clotting caused by vascular stenosis, to maintain patency of the access, and to avoid hospitalization or central catheter placement. Current imaging technologies are unsuitable for widespread monitoring of vascular access at a frequency greater than monthly and cannot detect rapidly progressing access dysfunction. Early detection can identify patients for imaging and enable surgical treatment planning.
In this chapter, we review the signal processing methods of analyzing the sounds of blood flow (bruits) using phonoangiography. This method relies on acoustic features produced from turbulent blood flow. We discuss the custom signal processing techniques used for de-noising of the signal and feature extraction to reduce the dimensionality of wavelet-processed audio signals. To move toward a device suitable for use in dialysis centers, we developed a multichannel-flexible piezoelectric microphone to record bruits from multiple locations. This piezoelectric microphone in combination with the signal processing technique could localize stenosis within 1 cm of the actual location and differentiate degree of stenosis into three grades (mild, moderate, and severe) in vitro. These results were confirmed with statistical analysis and the performance of the feature was also tested with a threshold-based stenosis detection classifier to mimic a clinical screening scenario aimed at identifying at-risk patients.
KeywordsPhonoangiogram Bruit Vascular access Stenosis Flexible microphone PVDF Microphone Frequency-domain linear prediction Wavelet Spectral centroid Spectral flux
This work was supported in part by RX001968-01 from US Dept. of Veterans Affairs Rehabilitation Research and Development Service, the Advanced Platform Technology Center of the Louis Stokes Cleveland Veterans Affairs Medical Center, and Case Western Reserve University. The contents do not represent the views of the US Government.
- 3.Feldman, H. I., Kobrin, S., & Wasserstein, A. (1996). Hemodialysis vascular access morbidity. Journal of Amerian Society of Nephrology, 7(4), 523–535.Google Scholar
- 9.Medicare claims processing manual. Publication # 100-04. Retrieved June 1, 2019, from https://www.cms.gov/Regulations-and-Guidance/Guidance/Manuals/downloads/clm104c08.pdf
- 10.Hemodialysis | NIDDK. [Online]. Retrieved January 4, 2019, from https://www.niddk.nih.gov/health-information/kidney-disease/kidney-failure/hemodialysis.
- 18.Majerus, S. J. A., Knauss, T., Mandal, S., Vince, G., & Damaser, M. S. (2018). Bruit-enhancing phonoangiogram filter using sub-band autoregressive linear predictive coding. In 2018 40th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC) (pp. 1416–1419).CrossRefGoogle Scholar
- 30.Wang, Y.-N., Chan, C.-Y., & Chou, S.-J. (2011). The detection of arteriovenous fistula stenosis for hemodialysis based on wavelet transform. International Journal of Advanced Computer Science, 1(1), 16–22.Google Scholar
- 32.Chen, W.-L., Kan, C.-D., Lin, C.-H., Chen, W.-L., Kan, C.-D., & Lin, C.-H. (2014). Arteriovenous shunt stenosis evaluation using a fractional-order Fuzzy Petri net based screening system for long-term hemodialysis patients. Journal of Biomedical Science and Engineering, 07(05), 258–275.CrossRefGoogle Scholar
- 33.Gram, M., et al. (2011). Stenosis detection algorithm for screening of arteriovenous fistulae. In K. Dremstrup, S. Rees, & M. Ø. Jensen (Eds.), 15th Nordic-Baltic Conference on Biomedical Engineering and Medical Physics (NBC 2011). IFMBE Proceedings (Vol. 34). Berlin: Springer.Google Scholar
- 34.Todo, A., et al. (2012). Frequency analysis of shunt sounds in the arteriovenous fistula on hemodialysis patients. In The 6th International Conference on Soft Computing and Intelligent Systems, and The 13th International Symposium on Advanced Intelligence Systems (pp. 1113–1118).CrossRefGoogle Scholar
- 37.Grochowina, M., Leniowska, L., & Dulkiewicz, P. (2014). Application of artificial neural networks for the diagnosis of the condition of the arterio-venous fistula on the basis of acoustic signals. Brain Informatics and Health, pp. 400–411. https://doi.org/10.1007/978-3-319-09891-3_37.CrossRefGoogle Scholar
- 38.Rousselot, L. (2014). Acoustical monitoring of model system for vascular access in haemodialysis. Master thesis. Retrieved from http://resolver.tudelft.nl/uuid:bcd26bbd-e2a1-48d9-9b9f-dbb1138b8009
- 40.Gårdhagen, R. (2013). Turbulent flow in constricted blood vessels: Quantification of wall shear stress using large Eddy simulation. PhD dissertation, Linköping. https://doi.org/10.3384/diss.diva-100918
- 41.Athineos, M., & Ellis, D. P. W. (2003). Frequency-domain linear prediction for temporal features. In 2003 IEEE Workshop on Automatic Speech Recognition and Understanding (IEEE Cat. No.03EX721), St Thomas, VI, (pp. 261–266). https://doi.org/10.1109/ASRU.2003.1318451
- 44.Chin, S., Panda, B., Damaser, M. S., & Majerus, S. J. A. (2018). Stenosis characterization and identification for dialysis vascular access. In IEEE Signal Processing in Medicine and Biology Symposium 2018. https://doi.org/10.1109/SPMB.2018.8615597.
- 45.Panda, B., Chin, S., Mandal, S., & Majerus, S. (2018). Skin-coupled pvdf microphones for noninvasive vascular blood sound monitoring. In 2018 IEEE Signal Processing in Medicine and Biology Symposium (SPMB) (pp. 1–4).Google Scholar