Inline Real-Time Near-Infrared Granule Moisture Measurements of a Continuous Granulation–Drying–Milling Process
- 688 Downloads
The purpose of this research was to use inline real-time near-infrared (NIR) to measure the moisture content of granules manufactured using a commercial production scale continuous twin-screw granulator fluid-bed dryer milling process. A central composite response surface statistical design was used to study the effect of inlet air temperature and dew point on granule moisture content. The NIR moisture content was compared to Karl Fischer (KF) and loss on drying (LOD) moisture determinations. Using multivariate analysis, the data showed a statistically significant correlation between the conventional methods and NIR. The R 2 values for predicted moisture content by NIR versus KF and predicted moisture values by NIR versus LOD were 0.94 (p < 0.00001) and 0.85 (p < 0.0002), respectively. The adjusted R 2 for KF versus LOD correlation was 0.85 (p < 0.0001). Analysis of the response surface design data showed that inlet air temperature over a range of 35–55°C had a significant linear impact on granule moisture content as measured by predicted NIR (adjusted R 2 = 0.84, p < 0.02), KF (adjusted R 2 = 0.91, p < 0.0001), and LOD (adjusted R 2 = 0.85, p < 0.0006). The inlet air dew point range of 10–20°C did not have a significant impact on any of the moisture measurements.
Key wordscontinuous granulation–drying–milling near-infrared (NIR) spectroscopy process analytical technology (PAT) real-time moisture determination
We appreciate the generous support of GlaxoSmithKline (GSK). We also appreciate the support of Mr. Scott Staton, operation and formulation manager at Campbell University Pharmaceutical Sciences Institute for editing the flow diagram of ConsiGma™ kit provided by GSK (Zebulon).
- 1.Herschel W. Investigation of the powers of the prismatic colours to heat and illuminate objects; with remarks, that prove the different refrangibility of radiant heat, to which is added, an inquiry into the method of viewing the sun advantageously, with telescopes of large apertures and high magnifying powers. Phil Trans Roy Soc. 1800;90:255–83.CrossRefGoogle Scholar
- 6.Gupta A, Peck GE, Miller RW, Morris KR. Influence of ambient moisture on the compaction behavior of microcrystalline cellulose powder undergoing uni-axial compression and roller compaction: a comparative study using near-infrared spectroscopy. J Pharm Sci. 2005;94:2301–13.PubMedCrossRefGoogle Scholar
- 12.Poli JE, Hoag SW, Flank S. Comparison of authentic and suspect pharmaceuticals. Pharm Technol. 2009;33:46–52.Google Scholar
- 13.U.S. FDA. Office of Pharmaceutical Science (OPS). Process analytical technology (PAT) initiative. http://www.fda.gov/cder/OPS/PAT.htm. Accessed 1 Mar 2011.
- 14.De Beer T, Burggraeve A, Fonteyne M, Saerens L, Remon JP, Vervaet C. Near infrared and Raman spectroscopy for the in-process monitoring of pharmaceutical production processes. Int J Pharm. 2010. doi: 10.1016/j.ijphar.2010.12.012.
- 15.Simca-P+ version 11.5. Umetrics AB. Umea, Sweden.Google Scholar
- 16.JMP® 8 Design of experiments software. SAS Institute Inc. Cary, NC: SAS Publishing, 2008.Google Scholar