A portable system for collecting anatomical joint angles during stair ascent: a comparison with an optical tracking device
- 12k Downloads
Assessments of stair climbing in real-life situations using an optical tracking system are lacking, as it is difficult to adapt the system for use in and around full flights of stairs. Alternatively, a portable system that consists of inertial measurement units (IMUs) can be used to collect anatomical joint angles during stair ascent. The purpose of this study was to compare the anatomical joint angles obtained by IMUs to those calculated from position data of an optical tracking device.
Anatomical joint angles of the thigh, knee and ankle, obtained using IMUs and an optical tracking device, were compared for fourteen healthy subjects. Joint kinematics obtained with the two measurement devices were evaluated by calculating the root mean square error (RMSE) and by calculating a two-tailed Pearson product-moment correlation coefficient (r) between the two signals.
Strong mean correlations (range 0.93 to 0.99) were found for the angles between the two measurement devices, as well as an average root mean square error (RMSE) of 4 degrees over all the joint angles, showing that the IMUs are a satisfactory system for measuring anatomical joint angles.
These highly portable body-worn inertial sensors can be used by clinicians and researchers alike, to accurately collect data during stair climbing in complex real-life situations.
KeywordsRoot Mean Square Error Inertial Measurement Unit Stair Climbing Knee Angle Optical Tracking
In terms of self-rated health, the most important activities of daily living are those involving mobility . Self-reported difficulty in stair climbing has shown to be useful in assessing and defining functional status of older adults . Obtaining accurate data about mobility is therefore of great clinical relevance and could lead to further improvements in various rehabilitation treatments . Compared to level walking only a limited number of studies have investigated the kinematics and kinetics of normal stair climbing [4, 5, 6, 7, 8, 9, 10, 11]. In general, kinematics and biomechanical aspects of stair climbing are studied using laboratory staircases combined with an optical motion analysis system [4, 11]. Although this kind of research yields valuable information, the results only remain valid in conditions where no anticipation or reaction to a real-world environment is required. In addition, it is almost impossible to use any form of optical tracking on stairwells, as the vertical shaft which contains the staircase limits the placement of cameras. Collecting data during stair climbing in a more real-life, complex environment requires a portable and lightweight measuring device. Zhou et al (2006) showed that inertial measurement units (IMUs) consisting of gyroscopes, accelerometers and magnetometers used to measure upper limb motion can accurately estimate arm position . Accelerometers and gyroscopes have also been proven to be able to correctly record shank, thigh and knee angles during level walking and a variety of lower leg exercises [13, 14]. Although, a miniature gyroscope attached to the shank is able to detect different cycles during stair ascent  and position data of the foot can be gathered with the combination of a gyroscope and two accelerometers , a portable system that can collect anatomical joint angles during stair climbing has not yet been reported.
The purpose of this study is to compare the anatomical joint angles determined by IMUs during stair ascent, to those joint angles acquired with an optical tracking device. Measuring stair climbing can be of great clinical relevance, as according to the Canadian Institute for Health Information the most common specified type of falls (23%) for people of 65 years and over are falls on or from stairs and steps . Furthermore, it has also been shown that, for certain patient groups, stair climbing can be a more critical pre-clinical assessment than walking .
Fourteen healthy subjects, nine men and five women, with a mean age of 27 years (range 20 to 37) voluntarily participated in this study. Their mean (± standard deviation) height and weight were 175 (± 8) cm and 69 (± 10) kg. The protocol was approved by the College Research Ethics Committee. All subjects gave written informed consent before the experiment. Each subject was asked to ascend a staircase consisting of four steps during twelve separate trials. Subjects were instructed to climb the stairs in the way they felt most comfortable. Each step was 62 cm wide, 23 cm long and 15 cm high giving the stair a pitch angle of 31 degrees. The subject stood in front of the stair and started ascending the stair when a verbal signal was given.
During static stance, the X-axis of each IMU coordinate system was physically placed to be in the sagittal plane after an analytically alignment of the axes by software (MT Software V2.8.1, Xsens Technologies B. V., Enschede, Netherlands). The software program placed the Z-axis of each IMU in line with gravity (vertical plane) with the new X-axis of the sensor perpendicular to the Z-axis and along the line of the original X-axis . The non-orthogonality between the axes of the body-fixed co-ordinate system is less then 0.1° .
Active Codamotion (Codamotion, Charnwood Dynamics, Leicestershire, UK) markers were placed (Figure 1) on the toe (5th metatarsal head), ankle (lateral malleolus), knee (fibula head and lateral femoral condyle), hip (trochanter major) and on the side of stairs. These markers were fixed using double-sided adhesive tape. The Bilateral Segmental Gait Analysis system configuration was used for data acquisition by the Codamotion and Motion Tracker software. The cameras of the optical tracking device were positioned in such a way, that the position data of the markers on the right side could always be obtained during stair ascent. Data for both the Codamotion and the IMUs was acquired at 100 Hz and an electronic pulse was used to synchronize the two measurement devices. All further data analysis was done using Matlab (MathWorks, Inc, Natick, Massachussetts, USA).
The lower extremity could be approximated as a multi-link chain, with each body part as a rigid segment represented by one IMU . Only movements around the transverse axis (resulting in flexion-extension kinematics) were studied, as the largest range of motions of the lower extremity occur around this axis during stair climbing .
The knee angle (α) was determined with the IMUs, by subtracting the angle around the transverse axis of the shank from that of the thigh. The flexion-extension angle of the ankle was found by subtracting the lower leg angle from the foot angle in the sagittal plane, while the thigh angle was represented by the upper leg angle with respect to the vertical axis .
In which x6, y6, z6 are spatial coordinates of the trochanter major; x5, y5, z5 are spatial coordinates of the epicondylus fermoris lateralis; x4, y4, z4 are spatial coordinates of the head of the fibula; x3, y3, z3 are spatial coordinates of the malleolus lateralis. L65 is the length between the epicondylus fermoris lateralis and the trochanter major, while L43 is the distance between the head of the fibula and the malleolus lateralis. This calculation method described by Kiss, Kocsis and Knoll determines a knee angle (α) which only depends on the relative position of the shank to the thigh .
The same calculation can be applied for the determination of the ankle angle (β), by using the spatial coordinates of the most lateral aspect of the calcaneus (x2, y2, z2) and the 5th metatarsal head (x1, y1, z1) instead of those of the epicondylus fermoris lateralis and the trochanter major and by replacing the distance between the head of the fibula and the malleolus lateralis with the distance between the most lateral aspect of the calcaneus and the 5th metatarsal head. The thigh angle (γ) was defined as the angle between a spatial vector joining the epicondylus fermoris lateralis and the trochanter major and a vertical spatial vector.
Time was converted to percentages, starting from the onset of movement until the top of the stairs was reached, to allow accurate comparisons within subjects. All angles were normalized in time per trial and subject by calculating the mean angle per percentage of time.
Data was normally distributed as observed in the probability plots and histograms. All anatomical joint angles on the right leg, obtained with the two measurement devices, were evaluated by calculating a two-tailed Pearson product-moment correlation coefficient (r) and by calculating the root mean square error (RMSE) between the two signals [14, 25]. A paired t-test was used to compare the maximum range of motion obtained by IMUs with those obtained by the optical tracking device per subject (n = 14) and to determine if the slopes of the linear regressions differed from one. The significance level was set at 0.05.
Pearson correlations, Root Mean Square Errors and maximum Range of Motions
Pearson correlation coefficient
(p < 0.01) Mean (SD)
Root Mean Square Error in degrees Mean (SD)
Maximum Range of Motion in degrees
4 (± 2)
63 (± 8) *
54 (± 8)
4 (± 3)
91 (± 8)
92 (± 6)
5 (± 3)
56 (± 5) *
49 (± 4)
The aim of the study was to investigate if the anatomical joint angles determined by IMUs sufficiently approximate the anatomical joint angles that were gathered with an optical tracking device. Strong correlations and mean RMSE of 4 to 5 degrees were found for all angles, comparable to those obtained using a similar system to track upper limb motion . Similar correlations were also found for linear acceleration trajectories obtained from IMUs, when compared with those derived from optical tracking position data . Yet, in this study the mean RMSE as a percentage of the maximum value (4 to 9 percent) was higher than the percentages found for the linear acceleration trajectories, which were in the region of 1 to 6 percent.
In the unpublished pilot study to this paper, which compared the equipment component of both systems, Pearson's correlation coefficients of 0.999 (p < 0.001) between the IMUs and optical tracking device were found, with a RMSE of 1°. As the RMSE during stair climbing (4–5°) was greater than the RMSE found in the pilot study (1°), a small misalignment between the two coordinate systems could have been present. This misalignment might be explained by the fact that the IMUs were placed in the middle of the body segments, whilst the active markers of the optical tracking device were positioned on bony landmarks. However, these locations were chosen in order to minimize any motion artefact. Despite optimizing placement of markers, both systems suffer from motion artefacts. A translational displacement between bony landmark and marker is likely to occur during stair climbing, causing errors in estimating position during movement [27, 28], which in turn leads to inaccuracies in determining angles. The IMUs measure orientation rather than position and are consequently less prone to errors caused by translational displacement of the sensors. Errors related to movement can however still occur in the IMUs, because of rotational displacement of the sensor relative to the body segment, due to for example change in muscle contour. Future research is needed to investigate to what extent the IMUs are prone to this kind of error.
High accelerations can easily lead to an increase in errors, as many IMUs use the accelerometer unit as an inclinometer . If the magnitude of the acceleration can no longer be neglected with respect to the gravity, the accuracy of the orientation measurement will be reduced, making this kind of IMUs unsuitable to measure human movements during which high accelerations are occurring.
The range of motion, measured with the optical tracking device, at the ankle (54° ± 8) and the knee (92° ± 6) was comparable to the range (respectively, 56° ± 7 and 86° ± 5) observed by Mian et al (2007) in young adults climbing stairs . However, the range of motion at the thigh (49° ± 4) was higher then the range (30° ± 4) reported by Mian et al (2007), which might be related to a difference in the method used to calculate joint angles, at the hip, from optical tracking data.
Maximum range of motion of the knee angle was similar between the two measurement devices, but did differ in the thigh and ankle angles. Any inaccuracies in range of motion of the ankle angles were further increased by taking the foot as a single rigid segment, as motion occurs between the different parts of the foot . The fact that the foot is multi segmental, might explain why an IMU placed on the dorsal part of the foot provides a different range of motion compared to the optical markers placed on the heel and the toe. Some further clues about the differences found between the two systems are provided by inspection of individual traces which deviate strongly from the rest (Figure 3). Data inspection showed that these deviations occurred in the optical tracking marker position, presumably due to movement of the marker with respect to the bony landmark.
Activities of daily living have been previously investigated using mobile sensors, consisting of uniaxial accelerometers . More recently, inertial sensors, similar to those used in this study, were utilized to track upper limb motion without showing any notable drift in the estimation of the movements . No significant drift problems arose during data collection in this study, although future work is needed to determine how well the proposed method works during longer data collection periods.
In general, IMUs provide a good alternative for measuring joint angles of the lower extremity during stair ascent when compared to positional markers. In addition, they provide the opportunity to perform accurate measurements in complex real-life environments using a non constraining measurement device. Furthermore, the IMUs were easy to set up, giving rise to the opportunity for clinicians and researchers to measure stair climbing out of the laboratory setting.
This work was funded by a joint Medical Research Council and Graduate School Studentship of King's College London.
- 17.Canadian Institute for Health Information: National Trauma Registry Report Major Injury in Canada (includes 2001–2002 data). 2003, Ottawa: Canadian Institute for Health InformationGoogle Scholar
- 19.Kwakkel S, Godha S, Lachapelle G: Foot and Ankle Kinematics During Gait Using Foot Mounted Inertial System. ION NTM San Diego CA. 2007Google Scholar
- 21.XsensTechnologies: MTi and MTx User Manual and Technical Documentation. Enschede. 2006Google Scholar
- 22.Winter DA: Biomechanics and motor control of human movement. 1990, New York WileyGoogle Scholar
- 23.Liang Dong JW, Xiaoming Bao, Wendong Xiao: Tracking of Thigh Flexion Angle during Gait Cycles in an Ambulatory Activity Monitoring Sensor Network. ACTA AUTOMATICA SINICA – Special issue on Sensor Networks and Applications. 2006, 32: 938-946.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.