Wide-neck renal artery aneurysm: parenchymal sparing endovascular treatment with a new device
- 2.2k Downloads
Renal artery aneurysm is a rare disorder with a high mortality rate in the event of rupture, the most frequent complication, which can also occur in lesions smaller than those indicated for treatment by current criteria. Surgery is still the first-line treatment, although a growing trend toward endovascular management of visceral artery aneurysms has emerged because of the high efficacy and low invasiveness that has been demonstrated by several authors. Treatment of wide-necked aneurysms and, depending on location, those at renal artery bifurcations or distal branches is more complex and may require invasive surgical techniques, such as bench surgery.
We describe the successful use of a new neurointerventional coil to treat an enlarging wide-necked segmental-branch renal aneurysm in an elderly woman who was not a candidate for surgery because of several comorbidities.
The technique described allowed safe, successful treatment of a wide-necked aneurysm in an unfavorable vascular territory, reducing the risk of downstream artery embolization and consequent parenchymal damage and decreased renal function. In similar cases, other endovascular devices have often proven to be ineffective at nephron sparing. To validate the safety and efficacy of this system, more cases treated in this manner should be studied.
KeywordsRenal artery aneurysm Visceral artery aneurysm Wide-neck aneurysm Embolization Penumbra Microcoil embolization
Renal artery aneurysm
Multidetector computed tomography.
Renal artery aneurysm (RAA) is a rare disorder, occurring in 0.01%–1.3% of the general population. It represents 1% of all aneurysms but is the second most common visceral artery aneurysm, accounting for approximately 25% [1, 2].
RAAs are classified into four categories: saccular (70%); fusiform (22.5%); dissecting (7.5%); and mixed, including microaneurysms . Saccular aneurysms, the most common, typically occur at the primary or secondary bifurcations of renal arteries ; more rarely they are located intraparenchymally in a segmental artery or more distal branches. Etiologies for saccular aneurysms include atherosclerosis, fibromuscular dysplasia, and neurofibromatosis. RAAs are often noted incidentally but can also be accompanied by hypertension, hematuria, or pain. Hypertension related to RAA remains a controversial topic; however, most case series on RAA treatment report improvement or resolution of hypertension after treatment .
The indication for treatment is currently based on a size threshold that continues to be debated, although most authors believe that aneurysms 2 cm or less in diameter do not require treatment. However, the decision to treat an aneurysm should be made not only on the basis of size, but also after considering other issues, such as the presence of clinical symptoms (hematuria, hypertension, back pain, and renal infarction), anatomical and morphological characteristics (location, wall calcification, enlarging lesion), and general clinical features (life expectancy, comorbidities, planned pregnancy). Finally, it should be considered that the mortality rate for spontaneous rupture is about 80% . This outcome has also been described for aneurysms smaller than 1 cm [6, 7].
Many treatment options are currently available. Surgical management still represents the standard of care, particularly for aneurysms located at the renal artery bifurcation as well as those involving distal branches . Surgical repair may involve in situ or extracorporeal bench surgery , and may require a long operation time and have substantial perioperative complications such as unplanned nephrectomy, with a failure rate of 6.6%. After successful early surgical repair, there is a reintervention rate of 5.8% resulting from such complications as anastomotic stenosis or graft thrombosis .
Despite the relative success of surgical repair of symptomatic aneurysms, there is a growing trend toward endovascular alternatives over open procedures because of their low invasiveness and reduced morbidity. A recent review of endovascular treatment reported good clinical and angiographic success rates without major complications, loss of kidney function, or nephrectomy . However, no large case series or case reports with long-term follow-up are currently available.
Several endovascular embolization techniques have recently been developed to manage aneurysms with heterogeneous anatomical features. Indeed, while narrow-necked saccular aneurysms can be successfully treated with transcatheter coil embolization regardless of type , endovascular treatment of wide-necked aneurysms is not yet well established and may require the use of more sophisticated techniques. These techniques, originally conceived in neurointerventional practice, are based on balloon- or stent-assisted embolization. During these procedures, a suitable device is released within the feeding vessel across the neck of the aneurysm to prevent coil prolapse during deployment and facilitate denser packing of the aneurysm, with the aim of reducing the likelihood of parent-vessel embolization. Stents employed are either self-expandable (bifurcation and segmental artery) or balloon-expandable (main stem) depending on the reference vessels and operator preference [7, 11].
Complications associated with these techniques include artery dissection, thrombosis, and rupture; microthrombi embolization; and renal infarction [8, 11]. Endovascular techniques are not often used in distal aneurysms, since the smaller the reference vessel, the harder it is to release a stent or use a balloon without damaging the vessel wall or inducing thrombosis.
The use of stent graft has also been reported for treatment of aneurysms beyond the first renal artery division but has a low success rate in terms of sparing the segmental artery and parenchyma [10, 12, 13].
To protect the renal vasculature, three complex-shaped bare nitinol microcoils (Penumbra Coil 400™ 18 mm × 57-cm and 15 mm × 57 cm Complex Standard and 13 mm × 48-cm Complex Soft; Penumbra, Inc.) were then sequentially deployed in the aneurysm until it was densely packed to 10.5% of the aneurysm volume. Each coil was detached only when angiography clearly showed complete and exclusive coiling inside the sac. The first coil, which was used to prevent coil migration through the neck into the parent artery, was the stiffest available and was sized slightly larger than the aneurysmal dome, thus creating a cage through which the other, more flexible coils could be safely deployed.
In the present case, considering the patient’s comorbidities and location of the aneurysm, surgical repair would have been very invasive and would likely have required bench surgery, so endovascular exclusion was preferable. The anatomical features and location of the aneurysm, and the need for accurate coil deployment before release, justified the use of this new type of neurointerventional detachable coil, which has not previously been reported to treat RAA. A high packing density, with fewer coils and shorter procedure time than are required for other procedures using different materials, was also obtained.
To our knowledge, this is the first case of a visceral aneurysm treated with the Penumbra Coil 400™ Embolization System (Penumbra Inc.) that includes 3 years of follow-up. These 0.020-inch coils are larger in diameter than other neurointerventional coils and are inherently softer because of their diameter (softness increases by the third power of the diameter) and because of the different thickness of the nitinol stretch-resistant wire. The coil-within-a-coil design increases the ability of the device to occupy the aneurysmal sac with less risk of misplacement .
As previously reported, in the treatment of complex renal aneurysms in difficult locations, a neurointerventional coil system may facilitate complete sac embolization with lower morbidity and less risk of decreased renal function than can be achieved with surgery or other endovascular techniques. The Penumbra system in particular, because of its conformability and greater length-volume ratio, allows rapid and complete embolization with less risk of parent-vessel damage.
More cases should be treated in this manner to validate the safety and efficacy of this system.
Written informed consent was obtained from the patient for publication of this Case report and any accompanying images. A copy of the written consent is available for review by the Editor of this journal.
- 14.Milburn J, Pansara AL, Vidal G, Martinez RC: Initial experience using the Penumbra coil 400: comparison of aneurysm packing, cost effectiveness, and coil efficiency. J Neurointerv Surg. 2013, in pressGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2490/14/42/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.