Comparison of normalisation methods for surfaceenhanced laser desorption and ionisation (SELDI) timeofflight (TOF) mass spectrometry data
 5.8k Downloads
 23 Citations
Abstract
Background
Mass spectrometry for biological data analysis is an active field of research, providing an efficient way of highthroughput proteome screening. A popular variant of mass spectrometry is SELDI, which is often used to measure sample populations with the goal of developing (clinical) classifiers. Unfortunately, not only is the data resulting from such measurements quite noisy, variance between replicate measurements of the same sample can be high as well. Normalisation of spectra can greatly reduce the effect of this technical variance and further improve the quality and interpretability of the data. However, it is unclear which normalisation method yields the most informative result.
Results
In this paper, we describe the first systematic comparison of a wide range of normalisation methods, using two objectives that should be met by a good method. These objectives are minimisation of interspectra variance and maximisation of signal with respect to class separation. The former is assessed using an estimation of the coefficient of variation, the latter using the classification performance of three types of classifiers on realworld datasets representing twoclass diagnostic problems. To obtain a maximally robust evaluation of a normalisation method, both objectives are evaluated over multiple datasets and multiple configurations of baseline correction and peak detection methods. Results are assessed for statistical significance and visualised to reveal the performance of each normalisation method, in particular with respect to using no normalisation. The normalisation methods described have been implemented in the freely available MASDA Rpackage.
Conclusion
In the general case, normalisation of mass spectra is beneficial to the quality of data. The majority of methods we compared performed significantly better than the case in which no normalisation was used. We have shown that normalisation methods that scale spectra by a factor based on the dispersion (e.g., standard deviation) of the data clearly outperform those where a factor based on the central location (e.g., mean) is used. Additional improvements in performance are obtained when these factors are estimated locally, using a sliding window within spectra, instead of globally, over full spectra. The underperforming category of methods using a globally estimated factor based on the central location of the data includes the method used by the majority of SELDI users.
Keywords
Support Vector Machine Normalisation Method Chronic Fatigue Syndrome Classification Performance Peak DetectionBackground
A wide range of mass spectrometry techniques is available, of which MatrixAssisted Laser Desorption and Ionisation (MALDI) and SurfaceEnhanced Laser Desorption and Ionisation (SELDI) [1] coupled with TimeOfFlight (TOF) tubes are widely used for proteome screening. In both of these techniques a biological sample of interest, e.g., a serum sample, is applied to a plate or chip, left to incubate and subsequently cocrystallised with a matrix material. A laser is then fired at the cocrystallised mixture, causing it to desorb. The energy of the laser beam is transferred via the matrix to the analyte sample, thereby ionising it. An electrical field causes the desorbed and ionised material to fly through the TOF tube. Lower mass peptides travel faster through the tube than higher mass peptides, causing the former to arrive earlier at a detector placed at the end of the flight tube. Using a quadratic equation, the mass to charge ratio (m/z) of a peptide can be calculated. Because of the indirect (i.e., via the matrix) ionisation, laser desorption and ionisation is considered to be a "soft" ionisation method. As a consequence, most peptides will be singlecharged, i.e., z = 1.
The main difference between MALDI and SELDI is that the latter normally uses a chip with a chromatographic surface, making the purification of the sample implicit. For MALDI, the purification needs to be done before application to the plate, for example through the use of chromatographic beads. Data resulting from mass spectrometry measurements usually contain a substantial amount of noise and show large intermeasurement variation [2]. Technical variance in mass spectra can be caused by a number of factors, including the preprocessing of samples in the wetlab and the mass spectrometer itself. The best way to get maximal power from a statistical analysis is to minimise the level of experimental error and noise. Variance introduced during the preprocessing stages should therefore be minimised as much as possible using strict lab protocols. Furthermore, it makes sense to perform multiple replicate measurements of the same patient. Doing so will give a better estimation of the "true" spectrum of a patient, leading to a better characterisation of the population.
Independent of whether replicate measurements are performed or not, normalisation is usually conducted in order to increase comparability of spectra resulting from different measurements. Normalisation of mass spectra typically entails subtracting an (optional) offset and dividing by a scaling factor. Such offset and scaling parameters can be defined and applied globally, over the full spectrum, or locally, using a sliding window, encompassing a contiguous subset of spectral positions. The rationale behind this is that the global approach may be better able to capture the general characteristics of the data, whereas the local approach may be beneficial for spectra with varying (massdependent) noise levels.
Although extensive effort has been devoted towards understanding and modeling the noise [3, 4] and variation [5, 6, 7] observed in mass spectra, it is unclear which normalisation method is most favourable. Here we perform an extensive comparison of 16 normalisation methods. Please refer to the Methods section for an overview of the methods evaluated. In a comparison of normalisation methods one should employ a measure of the reduction in interspectra variation after normalisation to assess the performance of different normalisation methods. However, this may actually promote degradation of the signal in case normalisation entails dividing the original spectrum by a quantity that is proportional to itself. In such a case, the resulting coefficient of variance will be low, although the amplitude of the remaining signal, will, of course, be very low as well; a situation that is clearly highly undesirable.
 1.
minimises the variance between spectra;
 2.
maximises the classification signal, i.e., the association of spectra with their respective class labels.
To assess the first objective, the coefficient of variance between spectra is used. The second objective is measured using the performance of three classifier, or "classifierlike", systems. These are the Globaltest, the Support Vector Machine (SVM) with a radial basis kernel and a decision tree (CART).
Note that we assess both objectives separately, yielding two performance indicators, instead of combining them to yield one objective score. The reason for this is that the used objectives are partly conflicting, in the sense that a reduction in variance does not neccessarily lead to an improvement in classification performance. This makes it nontrivial to combine them in a straightforward manner and forces one to attach weights to each of them, as also illustrated in Additional file 1. Instead of choosing these weights ourselves, we prefer to let the reader decide which of the two objectives is most important.
Results and Discussion
For this comparison, four different realworld datasets representing twoclass diagnostic problems have been analysed, before and after normalisation. For robustness, the preprocessing of spectra was performed using 42 different configurations, the product of seven methods for baseline correction and six peak detection approaches. Normalisation of spectra using the 16 methods under study was carried out after baseline correction and prior to peak detection.
For each of the 42 configurations, both objectives, i.e., interspectra variance minimisation and classification performance maximisation, were assessed for each of the normalisation methods as well as for unnormalised data. This results for each normalisation method in two vectors of (ranked) scores, one for each objective, across all 42 preprocessing approaches and four datasets. Please refer to the Methods section for a detailed explanation of how these vectors (Equations 6 and 7) are obtained.
Comparison to no normalisation
As becomes apparent from the figure, the majority of normalisation methods lie within the (nonshaded) lower left region, indicating that they significantly outperform the case when no normalisation is used. An interesting observation here is that the default normalisation method used by most SELDI users ("Global zero mean") slightly reduces the variance but does not improve classification performance significantly with respect to unnormalised data.
Perclassifier performance
Global vs. local normalisation methods
The normalisation methods we studied employ two parameters, an offset and a scaling parameter. The methods can be divided into two groups, depending on whether they are based on global or on local estimates of characteristics of the data. To study whether this parameter (global vs. local) has a significant effect on the two performance objectives, we compared the performance of these two groups of methods. This was done by pairing the performance ranks obtained for local methods with those of their global counterparts, e.g., "Local zero mean" was paired with "Global zero mean", for each of the two objectives. We then used the Wilcoxon signed rank test to test the nullhypothesis that the number of cases in which a method from one group outperforms its paired method from the other group is equal to the number of cases in which the opposite is the case.
Offset and scaling parameters
A number of the normalisation methods studied use a zerovalued offset parameter. To study the influence of this on the performance, we again divided the methods into two groups, i.e., methods that employ a nonzero parameter versus those for which this parameter has been set to zero. We used a twosample Wilcoxon rank sum (i.e., MannWhitney U) test to test the nullhypothesis that the two groups outperform the case in which no normalisation is performed, an equal number of times. As before, we tested both tails separately for both objectives, resulting in four pvalues, again corrected for multiple testing using the Bonferroni method. Figure 4b shows these values, indicating no significant effect for either of the two objectives.
The same analysis was done for the scaling parameter, which can also be divided into two groups; one containing methods that use the central location of the data (i.e., mean or median) and another containing methods that use some measure of dispersion of the data around that central location (i.e., standard deviation, median absolute deviation or (quantile) range). Figure 4c shows that there is a significant difference here; in terms of both variance reduction (p = 5.23 × 10^{6}) and classification performance (p = 1.15 × 10^{6}) it is beneficial to choose a scaling parameter based on the dispersion of the data.
Conclusion
We have performed a systematic comparison of 16 normalisation methods for (SELDITOF) mass spectrometry data. For robustness, a large number (42) of configurations for baseline correction and peak detection methods was used, as well as multiple datasets. We used two objectives to assess the benefit of applying a particular normalisation method, namely minimisation of interspectra variance and maximisation of classification performance. The latter has proven to be very helpful in safeguarding against methods that reduce the variance between spectra but at the same time do not improve, or even worsen, the amount of meaningful signal left after normalisation.
We have shown that in the general case, normalisation of mass spectra is beneficial to both objectives; most methods we compared performed significantly better than the case in which no normalisation was used. We have shown that normalisation methods that scale spectra by a factor based on the dispersion (e.g., standard deviation) of the data clearly outperform those where a factor based on the central location (e.g., mean) is used. Additional improvements in performance are obtained when these factors are estimated locally, using a sliding window within spectra, instead of globally, over full spectra. The underperforming category of methods using a globally estimated factor based on the central location of the data includes the method used by the majority of SELDI users, i.e., "Global zero mean".
Methods
Normalisation methods
where Θ and Δ denote offset and scaling parameters respectively. Both of these parameters can be estimated globally over a full spectrum or locally using a sliding window, which encompasses a contiguous subset of spectral positions. In the former case, Θ and Δ assume scalar values, while in the latter case, Θ and Δ are vectors of length S, i.e., the number of measurement points in the spectrum.
Normalisation methods used in the comparison. Normalisation methods used in the comparison. Offset and scaling parameters are used as in Equation 1. Note that for each method these parameters are estimated globally as well as locally and evaluated separately. Formal definitions of these methods can be found in Additional file 2.
No  Offset (Θ)  Scaling (Δ)  Working name  Citation 

1  0  mean  Zero mean  [15,16] 
2  0  median  Zero median  [8] 
3  0  sd  Zero SD  
4  0  mad  Zero MAD  
5  mean  sd  Mean SD  
6  median  mad  Median MAD  
7  median  iqr  Median quantile  [17] 
8  min  range  Min range  ([18]) 
The last column of Table 1 contains, when possible, citations of studies in which the normalisation methods were used. Note that for each method the global and local variants have been used, resulting in a total of 16 methods. Additional file 2 contains formal definitions of the normalisation methods.
The first method in Table 1 ("zero mean") is a derivation of the method as implemented in the software supplied with SELDITOFMS machines. Strictly speaking, this software (Ciphergen^{©} ProteinChip Software 3.1) uses the mean intensity per spectrum to scale individual spectra and subsequently rescales all spectra by the mean intensity across all spectra. Because this second scaling is done using a constant scale factor for all spectra, it can easily be left out, as we do here. Although this second scale factor is constant for one normalisation run, it is completely dependent on all spectra, i.e., on their means. This makes it hard to introduce new samples at a later stage or test any built classifier on another dataset. For this reason we only focus on normalisation factors calculated per single spectrum.
As an alternative to using the mean intensity as a scale factor, the median can be used (i.e., method "zero median"). Some studies [8] suggest this may be more robust against outlying peaks.
Another way of normalising spectra is by using the data dispersion, such as the standard deviation (methods "zero SD" and "mean SD"), the median absolute deviation (methods "zero MAD" and "median MAD") or ranges of the data. Method "min range" essentially rescales the data to the interval [0, 1], either globally or in a local window. As a more robust alternative to this, method "median quantile" uses quantiles of the data.
The citation for method "min range" is parenthesised in Table 1 because it does not strictly follow the normalisation described. In that particular study, normalisation is performed over random subsets of (binned) intensity values instead of over all values (globally) or sliding windows of adjacent values (locally).
Objective 1: interspectra variance
Given the linear relationship between means and standard deviations of peaks (illustrated in last section of Additional file 3), a reasonable measure to compare interspectra variation between methods is the coefficient of variation (CV). Defined as the ratio between the standard deviation and the mean $(\text{CV}=\frac{s}{\mu})$ it represents a more scaleindependent estimation of the variance. However, it may also yield very unstable results for mean values around zero. For this reason, we use an alternative way of computing the CV, adopted from [8]:
which is essentially the slope of a regression line through the origin in a scatter plot of s vs. μ, such as shown by the dashed line in the figure in Additional file 3.
A sensible way to estimate interspectra variance after normalisation would be to inspect the CV calculated over a number of artificially spikedin peaks and between samesample replicates. In (existing) realworld datasets however, technical replicates may not be present, let alone artificial spiking peaks. We hypothesised that the CV calculated over all peaks and all available spectra (from all classes) should be a reasonable approximation of the average samesample replicate CV calculated over spiking peaks only. This is indeed the case, proven by a high Pearson correlation (0.98) between the CVs calculated over the four spiking peaks and over all peaks present in the spectra of this dataset. For this reason, it is justified to use the CV calculated over all peaks. This allows us to use multiple datasets in our comparison of normalisation methods, including datasets lacking spikedin peptides and/or technical replicates.
Objective 2: classification performance
In order to assess class separation, and more specifically how it changes under the influence of different normalisation methods, we employ a number of classifiers or classifierlike systems. We employed a single classifier at a time in combination with the variance objective. All classifier analyses were performed using R and various packages that are available for it, and below we elaborate on each of the employed systems.
The Globaltest
The Globaltest [9] tests the association of groups of features with a given outcome. Simply put, it tests whether the normalized sum of the correlation coefficients resulting from correlating the peak heights and the class label is sufficiently high to reject the null hypothesis that there is no association between the peak heights and the class label. It is not a classifier, in the sense that it does not output a mapping to assign new samples to one of the outcome classes. It does, however, give an indication of the quality of a dataset with respect to class separation. On top of that, it makes minimal assumptions about the data and has no parameters to be set. The result of using the Globaltest is a single pvalue for each group of features, where a group can be defined as all features in a dataset or a subset thereof. Here, we apply this test to all peaks found in a normalised mass spectrum. The globaltest package was used as an implementation of the Globaltest.
Support Vector Machine (SVM) with radial basis kernel
A SVM [10] is a classifier that is widely used because of its generally good performance in complex classification problems and especially in applications, such as the problem being studied here, where the number of features is larger than the total number of samples. In essence, it is a classifier in linear space, however transformation to a nonlinear space is achieved by using a nonlinear kernel, such as the radial basis kernel, as we do here. We used the implementation provided by the e1071 package. To assess the performance of this classifier we use the leaveoneout cross validation error. This was implemented using the ipred package.
Classification And Regression Trees (CART)
CART [11] is an algorithm that is widely used within the SELDI community because it is implemented in the software that comes with the mass spectrometer itself and allows for easy interpretation of results. Trees were obtained by a twostep process, making use of the rpart package. Initially, large trees are grown using the Gini index for node impurity. Resulting trees are then, again using a leaveoneout cross validation approach, pruned back to the number of nodes at which the improvement in fit is not significant anymore. This process has the goal of avoiding overfitting. Like with the SVM, we employ the leaveoneout cross validation error to assess the performance of built classifiers.
Datasets
For robustness, the comparison of normalisation methods has been performed using a total of four SELDI datasets.

The first dataset is the result of an experiment designed and executed specifically for this study. It is the dataset used to study the relation between mean and standard deviation of peak intensities. For this dataset, realworld serum samples from patients suffering from colorectal cancer (four samples) and controls (four samples) have been artificially spiked with additional peptides. For each sample a total of 10 technical replicates was obtained, yielding a total of 80 spectra, 40 per class. Please refer to Additional file 3 for a more elaborate discussion on (the preparation of) this dataset.

The second and third datasets resulted from an earlier study [12] concerned with finding biomarkers in human serum to differentiate between colorectal cancer patients and controls. The datasets consist of SELDI measurements of serum samples from 37 colorectal cancer patients vs. 31 controls and 40 colorectal cancer patients vs. 49 controls. Detailed information on the used experimental setup and preprocessing methods can be found in [12].

The fourth dataset is from the public repository of the Critical Assessment of Microarray Data Analysis (CAMDA) conference 2006 [13]. In this case, the dataset is the result of measuring serum samples of patients suffering from chronic fatigue syndrome as well as control persons, with 62 and 64 samples in each class, respectively. Specifically, the second liquid chromatography fraction of the CM10 CAMDA dataset was used. The used preprocessing protocols can be obtained from the CAMDA website.
Spectrum preprocessing
All preprocessing of spectra was done using the R statistical software with the MASDA Rpackage. Preprocessing includes correcting the baseline effect of raw spectra, normalisation and peak detection, filtering and clustering.
We used seven different methods of correcting the baseline of raw spectra before normalisation. These methods include the one implemented in the PROcess Rpackage, the method used by the manufacturer of SELDI machines and methods based on various smoothing approaches.
After normalisation, peaks are detected in individual spectra, effectively by detecting changes in the first derivative of the intensity curve. This is followed by a process in which peaks are filtered by using estimated noise thresholds, depending on the parameter settings. During the comparison, six different parameter settings were used for robustness. These settings ranged from using no filtering at all to filtering out all peaks below a local noise threshold estimated by a robust local estimator plus five times its median absolute deviation (MAD).
Please refer to Additional file 4 for a more detailed description of the baseline correction and peak detection methods used.
After peak detection and filtering, peaks from different spectra are clustered together using complete linkage hierarchical onedimensional clustering with a configurable, mass dependent, cutoff point. By default, this point is 0.3% of the peak m/z value. This mass dependent value is turned into a constant by log10 transforming the m/z values of detected peaks. For the default value of 0.3%, this yields a cutoff point roughly equal to 0.0013. Resulting peakclusters, containing one peak per processed spectrum, are used as units in subsequent analyses.
The MASDA mass spectrometry data analysis Rpackage is freely available [14]. All code to process the datasets is included as Additional file 5.
Ranked score vector calculation
We use a nonparametric paired Wilcoxon signed rank test to compare two normalisation approaches, separately for the two objectives. Tests are performed on all possible pairings of normalisation methods, to assess whether rankings are significantly different between methods, irrespective of the used baseline correction method, peak detection approach and dataset. More specifically, we use onetailed tests in order to obtain information on relative performance differences between normalisation methods. Bonferroni multiple testing corrected pvalues were obtained by multiplying the raw pvalues with a correction factor equal to the number of normalisation methods employed, i.e., 17.
All code to analyse the datasets as described above, and to exactly generate the figures used in this paper, is included as Additional file 6.
Construction of quadrant plots
Quadrant plots (Figures 2, 3 and 4) aim to visualise the performance of normalisation methods, simultaneously for the two objectives. For each objective, we use two onetailed tests to assess whether the performance of a particular normalisation method is significantly better or worse than when using no normalisation. We employed onetailed tests, since we wanted to test the directionality of the association, and then correct for multiple testing. Because we are interested in differences between normalisation methods, we only use the lowest of the two pvalues obtained from the two onetailed tests per objective. For each normalisation method, we then plot a symbol representing it in the quadrant plot, using the two resulting pvalues (one for each objective) as coordinates.
Notes
Acknowledgements
The authors wish to thank Miranda van Uitert and Domenico Bellomo for useful discussions. This work is part of the BioRange programme of the Netherlands Bioinformatics Centre (NBIC), which is supported by a BSIK grant through the Netherlands Genomics Initiative (NGI).
Supplementary material
References
 1.Wright GL Jr, Cazares LH, Leung SM, Nasim S, Adam BL, Yip TT, Schellhammer PF, Gong L, Vlahou A: Proteinchip(R) surface enhanced laser desorption/ionization (SELDI) mass spectrometry: a novel protein biochip technology for detection of prostate cancer biomarkers in complex protein mixtures. Prostate Cancer Prostatic Dis 1999, 2(5/6):264–276.Google Scholar
 2.Semmes OJ, Feng Z, Adam BL, Banez LL, Bigbee WL, Campos D, Cazares LH, Chan DW, Grizzle WE, Izbicka E, Kagan J, Malik G, McLerran D, Moul JW, Partin A, Prasanna P, Rosenzweig J, Sokoll LJ, Srivastava S, Srivastava S, Thompson I, Welsh MJ, White N, Winget M, Yasui Y, Zhang Z, Zhu L: Evaluation of Serum Protein Profiling by SurfaceEnhanced Laser Desorption/Ionization TimeofFlight Mass Spectrometry for the Detection of Prostate Cancer: I. Assessment of Platform Reproducibility. Clin Chem 2005, 51: 65–74. 10.1373/clinchem.2005.053405PubMedCentralCrossRefPubMedGoogle Scholar
 3.Malyarenko DI, Cooke WE, Adam BL, Malik G, Chen H, Tracy ER, Trosset MW, Sasinowski M, Semmes OJ, Manos DM: Enhancement of sensitivity and resolution of surfaceenhanced laser desorption/ionization timeofflight mass spectrometric records for serum peptides using timeseries analysis techniques. Clin Chem 2005, 51: 65–74. 10.1373/clinchem.2004.037283PubMedCentralCrossRefPubMedGoogle Scholar
 4.Morris JS, Coombes KR, Koomen J, Baggerly KA, Kobayashi R: Feature extraction and quantification for mass spectrometry in biomedical applications using the mean spectrum. Bioinformatics 2005, 21(9):1764–1775. 10.1093/bioinformatics/bti254CrossRefPubMedGoogle Scholar
 5.Baggerly KA, Morris JS, Coombes KR: Reproducibility of SELDITOF protein patterns in serum: comparing datasets from different experiments. Bioinformatics 2004, 20(5):777–785. 10.1093/bioinformatics/btg484CrossRefPubMedGoogle Scholar
 6.Hong H, Dragan Y, Epstein J, Teitel C, Chen B, Xie Q, Fang H, Shi L, Perkins R, Tong W: Quality control and quality assessment of data from surfaceenhanced laser desorption/ionization (SELDI) timeof flight (TOF) mass spectrometry (MS). BMC Bioinformatics 2005, 6(Suppl 2):S5. 10.1186/147121056S2S5PubMedCentralCrossRefPubMedGoogle Scholar
 7.Dijkstra M, Vonk RJ, Jansen RC: SELDITOF mass spectra: A view on sources of variation. J Chromatogr B Analyt Technol Biomed Life Sci 2006.Google Scholar
 8.de Noo M, Tollenaar R, Ozalp A, Kuppen P, Bladergroen M, Eilers P, Deelder A: Reliability of Human Serum Protein Profiles Generated with C8 Magnetic Beads Assisted MALDITOF Mass Spectrometry. Anal Chem 2005, 77(22):7232–7241. 10.1021/ac050571fCrossRefPubMedGoogle Scholar
 9.Goeman JJ, van de Geer SA, de Kort F, van Houwelingen HC: A global test for groups of genes: testing association with a clinical outcome. Bioinformatics 2004, 20: 93–99. 10.1093/bioinformatics/btg382CrossRefPubMedGoogle Scholar
 10.Vapnik VN: The nature of statistical learning theory. New York, NY, USA: SpringerVerlag New York, Inc; 1995.CrossRefGoogle Scholar
 11.Breiman L, Friedman J, Olshen R, Stone C: Classification and regression trees. Pacific Grove, CA, USA: Wadsworth & Brooks/Cole Advanced Book and Software; 1984.Google Scholar
 12.Engwegen JYMN, Helgason HH, Cats A, Harris N, Bonfrer JMG, Schellens JHM, Beijnen JH: Identification of serum proteins discriminating colorectal cancer patients and healthy controls using surfaceenhanced laser desorption ionisationtime of flight mass spectrometry. World J Gastroenterol 2006, 12(10):1536–1544.PubMedCentralPubMedGoogle Scholar
 13.CAMDA2006 website[http://www.camda.duke.edu]
 14.NKI Bioinformatics website[http://bioinformatics.nki.nl]
 15.Fung ET, Enderwick C: ProteinChip clinical proteomics: computational challenges and solutions. Biotechniques 2002, (Suppl):34–8. 40–1Google Scholar
 16.Sauve AC, Speed TP: Normalisation, baseline correction and alignment of highthrougput mass spectrometry data. Proceedings Gensips 2004.Google Scholar
 17.de Noo ME, Mertens BJA, Ozalp A, Bladergroen MR, van der Werff MPJ, van de Velde CJH, Deelder AM, Tollenaar RAEM: Detection of colorectal cancer using MALDITOF serum protein profiling. Eur J Cancer 2006, 42(8):1068–1076. 10.1016/j.ejca.2005.12.023CrossRefPubMedGoogle Scholar
 18.Petricoin EF, Liotta LA: SELDITOFbased serum proteomic pattern diagnostics for early detection of cancer. Current Opinion in Biotechnology 2004, 15: 24–30. 10.1016/j.copbio.2004.01.005CrossRefPubMedGoogle Scholar
Copyright information
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.