Stress on facial skin of class III subjects during maxillary protraction: a finite element analysis
Maxillary protraction with facemask (FM) is an orthopedic approach for treatment of Class III growing patients. Aim of the present investigation was to analyze tension loads produced by two different facial mask (FM) designs on facial skin of subject with skeletal Class III.
A three-dimensional (3D) geometry of Delaire and Petit FM models were reconstructed from the original Computer Aided Design (CAD) 3D prototype using software package (ANSYS 5.7). A traction load of 9.8 N inclined of 30° to the occlusal plane was applied combining analytical FM models with a 3D facial model. Resulting stresses and deformations on the skin layer were tested through the von Mises yield criterion.
Overall tensions were mostly developed on the chin area, while lower stresses were observed on forehead area for both FM designs. When Delaire FM model was tested, maximum stresses were observed on the upper border of the chin cup corresponding to the inferior lip and to marginal gingiva of lower incisors. After Petit FM application, maximum stresses were more extensively localized at the level of both upper border and central area of the chin. Stresses measured on the chin area were significantly higher with Petit FM when compared with Delaire FM (44 KPa versus 29 KPa, respectively).
Delaire FM determined lower stresses and tensile tensions than Petit FM model. Highest tensions were observed at the level of chin cup area for both Delaire and Petit FM. Stresses following Delaire FM application were mostly observed on the upper border of the chin cup, while Petit FM determined stresses more extensively distributed to the central area of the chin.
KeywordsFEM analysis Face mask Maxillary protraction Facial skin stresses
Computer Aided Design
Finite Element Analysis
Face-mask (FM) therapy is the most recommended orthopedic approach for early treatment of skeletal Class III malocclusion [1, 2, 3, 4, 5]. Many articles [1, 2, 3, 4, 5] described favorable and stable outcomes on dento-skeletal structures and on soft tissue profile of maxillary protraction protocol. During orthopedic treatment, tensile forces are applied on maxillary sutures allowing a forward displacement of the maxilla and an improvement of the sagittal dento-skeletal relationship [6, 7]. Orthopedic therapy requires the application of heavy forces ranging from 7.8 N to 9.8 N (800–1000 g, respectively) that often cause skin irritation and even mild swelling [7, 8]. Original FM model was firstly described by Jean Delaire [9, 10] and it consists of two extra-oral plastic supports connected to a metallic framework composed by two lateral vertical bars and one cross bar. Some years later, Petit  introduced a modified model of Delaire FM with a single rod running in the midline between the chin and forehead plastic cups. Recently, three-dimensional (3D) Finite Element Analysis (FEA) [6, 7, 12] has been used in order to evaluate the displacement and stress distribution of orthopedic forces applied on maxillofacial structures. Gautam et al.  highlighted that the highest loads produced by FM therapy were mostly localized along naso-maxillary, frontonasal and, fronto-maxillary sutures. Gazzani et al.  evaluated mechanical properties of Delaire FM and the stresses generated on device components during orthopedic treatment, by means of FEA. Some of the major complications related to maxillary protraction treatment are due to poor adaptation and low fit of FM on the face and to skin irritation determined by the extra-oral cups. The discomfort in wearing the FM device affects patients’ compliance with negative consequences on treatment effectiveness.
Purpose of the study
Hence, the aim of the present FEA study was to analyze tension loads produced by two FM designs and their effects on the chin and forehead skin area during orthopedic maxillary protraction.
Finite element analysis (FEA)
geometrical features of both Delaire and Petit FMs;
geometrical feature of 3D facial model prototype;
material properties for each element of the FMs;
support bone properties;
mesh (number, shape and size of the elements used to discretize the FMs);
constrains and loads applied on the system.
Evaluation of load and stress distribution
In order to underline the load and stress distribution on the forehead and chin areas, von Mises yield criterion was evaluated on the skin layer delimitated externally by the FMs cups and internally by the bone structure. The amount of elastic energy absorption was calculated for 9.8 N inclined of 30° to the occlusal plane to quantify the deformation state induced on the skin areas involved by the two FM models.
Maximum tensions on the skin areas
Max. tension on chin skin surface
Max. tension on forehead skin surface
FM therapy represents the gold standard for correcting skeletal Class III malocclusion in growing subjects [1, 2, 3, 4, 5, 16, 17, 18, 19]. Several factors play a substantial role in terms of efficacy and effectiveness of orthopedic Class III treatment, including individual skeletal pattern, protraction device adaptation, patients’ compliance, and device wear-time [20, 21, 22, 23, 24]. Stocker et al.  and Ozkalayci and Cicek  analyzed wearing time and patient compliance with FM reporting a significantly lower wearing time than the prescribed instructions. The not customized design of both FM models can cause poor device adaptation to the patient’s face. The less adhesion of the device could determine skin irritations and mild swelling, with negative effects on patient compliance [1, 2, 3, 4, 5]. Stresses and load distribution developed on skeletal structures during orthopedic protraction treatment were widely analyzed [6, 7, 25]. More recently, in a previous investigation stresses developed during maxillary protraction and their effects on FM structure were evaluated by means of FEM analysis . The absence of permanent plastic deformations and efficacy persistence of FM components highlighted the importance of a careful management of the device during treatment in order to grant its best performance. Although adverse effects on facial skin are known and very frequent in daily practice, however no data are available in literature regarding distributions and effects of tensile forces on facial skin. To our knowledge, both Petit and Delaire FM can lead to skin irritations caused by the plastic forehead and chin pads. Plastic supports of FM should to be adjusted to fit patient’s face maximizing the contact surface with the skin for a homogeneous distribution of the loads applied and avoiding skin wounds. Overall results of the present study highlighted higher tensile tensions at the level of the chin cup after the application of the Petit FM when compared with the Delaire FM (44 kPa and 29 kPa, respectively) (Table 1). More extended chin support of Delaire design reduced the intensity of the residual tensions and stresses developed on the patient skin face, while smaller chin support of Petit FM tended to increase the tensile tensions on the face exposing the patient to more skin irritation. Lower tensile tensions decrease the risk of skin irritations, but also maximize orthopedic forces transmission to the bone structure . Moreover, tensile tensions were localized on the whole chin cup area with Petit FM, while more extended chin cup pads of Delaire FM determined maximum stresses in correspondence of its upper border. In both FM models’ analysis, tensions registered on forehead region resulted particularly low and thus, clinically irrelevant, especially when Petit FM was tested. Maximum stresses ranged from 3 kPa (Petit FM) to 7 kPa (Delaire FM) and were mostly localized on the lower area, near to the eyebrows (Table 1). Color uniformity observed on the forehead support indicates the absence of significative stresses induced by applying Petit FM (Fig. 4B). Forehead caps can be considered negligible in terms of discomfort and fit if compared with the chin supports. Tensile strength analysis confirmed good adaptation for both FM designs. It can be supposed that all tensions and stresses developed during orthopedic protraction are maximized and focused on the chin area near the center of load application, especially in the Petit structure. The results of the present FEA investigation suggested that Petit FM appears clinically less comfortable than the Delaire model with a higher risk of skin irritation and mild swelling on the chin. On the other hand, a prolonged stress applied by the border of the chin cap could interfere with the periodontal health of the lower incisors . An external compression on the labial mental grove exerted by the chin pad could transmit a compression on the lower incisors’ marginal gingiva with resulting processes of gingival retraction. According to the findings of the present study, FM designs with more extended chin supports should be preferred to achieve wearing comfort. As previously suggested  a clinician-made customization of the chin cup using a poly-vinyl siloxane could be used to uniform the contact surfaces of the orthopedic device structure improving its performance and control of the load distribution.
Limitations of the current study were due to the 3D facial model used to perform FEM analysis of forces induced by different FM models. Standardization of 3D model did not provide data about how forces distribution could change on different facial biotype. Further clinical studies will be necessary to analyzed stresses and loads distribution of orthopedic forces induced by maxillary protraction considering different facial patterns and biotypes in order to grant optimum management and customized fit of the device.
Highest tensions were observed at the level of chin cup area than at the level of forehead cups for both Delaire and Petit FM. Stresses following Delaire FM application were mostly observed on the upper border of the chin cup at the level of the inferior lip and lower incisor marginal gingiva. Petit FM produced higher tensile tensions than the Delaire design on the chin cup with stresses more extensively distributed to the central area of the chin.
The authors wish to thank to Dr. Gabriele Scommegna for his advices in performing the experimental analysis and, Engineers Marouue J. and Gervasi G.L. for their competence and scientific support.
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the corresponding author on reasonable request.
FG analyzed and interpreted the data obtained, and was a major contributor in writing the manuscript. CP, RL, PC interpreted the data obtained and were a supervisor in writing the paper. All authors read and approved the final manuscript.
Ethics approval and consent to participate
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
- 3.Pavoni C, Mucedero M, Baccetti T, Franchi L, Polimeni A, Cozza P. The effects of facial mask/bite block therapy with or without rapid palatal expansion. Prog Othod. 2009;10:20–8.Google Scholar
- 5.Masucci C, Franchi L, Defraia E, Mucedero M, Cozza P, Baccetti T. Stability of rapid maxillary expansion and facemask therapy: a long-term controlled study. AJODO. 2011;140:493–500.Google Scholar
- 6.Gautam P, Valiathan A, Adhikari R. Maxillary protraction with and without maxillary expansion: a finite element analysis of sutural stresses. AJODO. 2009;136(3):361–6.Google Scholar
- 11.Petit HP. Adaptation following accelerated facial mask therapy. In: McNamara JA Jr, Ribbens KA, Howe RP, eds. Clinical Alteration of the Growing Face. 1983. Monograph 14, craniofacial growth series. Ann Arbor, Mich: Center for Human Growth and Development, University of Michigan;Google Scholar
- 13.Gazzani F, Pavoni C, Giancotti A, Cozza P, Lione R. Facemask performance during maxillary protraction: a finite element analysis (FEA) evaluation of load and stress distribution on Delaire facemask. Progr in Orthod 2018; 9;19(1):21;Google Scholar
- 16.Alcan T, Keles A, Erverdi N. The effects of a modified protraction headgear on maxilla. AJODO. 2000;117(1):27–38.Google Scholar
- 17.Lione R, Buongiorno M, Laganà G, Cozza P, Franchi L. Early treatment of class III malocclusion with RME and facial mask: evaluation of dentoalveolar effects on digital dental casts. Eur J Paediatr Dent. 2015;16(3):210–20.Google Scholar
- 20.Franchi L, Baccetti T, Tollaro I. Predictive variables for the outcome of early functional treatment of class III malocclusion. AJODO. 1997;112(1):80–6.Google Scholar
- 21.Nanda R, Kierl M. Prediction of cooperation in orthodontic treatment. AJODO. 1992;102(1):15–21.Google Scholar
- 23.Stocker B, Willmann JH, Wilmes B, Vasudavan S, Drescher D. Wear-time recording during early class III facemask treatment using TheraMon chip technology. AJODO. 2016;150(3):533–40.Google Scholar
- 24.Ozkalayci N, Cicek O. When Do Skeletal Class III Patients Wear Their Reverse Pull Headgears? Biomed Res. Int. 2017. https://doi.org/10.1155/2017/3546262;
- 26.Parenti S, Checchi V, Molinari C, Bonetti GA. Periodontal side effect during orthopedic face mask therapy. Inter J Orthod. 2015;26(4).Google Scholar
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.