Response of timepix detector with GaAs:Cr and Si sensor to heavy ions
- 36 Downloads
The response of the Timepix detector to neon ions with kinetic energy 77 and 158.4 MeV has been studied at the cyclotron U-400M of the JINR Flerov Laboratory of Nuclear Reaction. Sensors produced from gallium arsenide compensated by chromium and from silicon are used for these measurements. While in Timepix detector with Si sensor the well-known so-called “volcano effect” observed, in Timepix detector with GaAs:Cr sensor such effect was completely absent. In the work the behavior of the Timepix detector with GaAs:Cr sensor under irradiation with heavy ions is described in comparison with the detector based on Si sensor. Also the possible reason for absence of “volcano” effect in GaAs:Cr detector is proposed.
KeywordsGaAs Bias Voltage Method Phys Nucleus Letter Gallium Arsenide
Unable to display preview. Download preview PDF.
- 9.A. Butler, P. Butler, S. Bell, G. A. Chelkov, D. V. Dedovich, M. A. Demichev, V. G. Elkin, M. I. Gostkin, S. A. Kotov, D. A. Kozhevnikov, U. G. Kruchonak, A. A. Nozdrin, S. Yu. Porokhovoy, I. N. Potrap, P. I. Smolyanskiy, M. M. Zakhvatkin, and A. S. Zhemchugov, “Measurement of the energy resolution and calibration of hybrid pixel detectors with GaAs:Cr sensor and timepix readout chip,” Phys. Part. Nucl. Lett. 12, 59–73 (2015).CrossRefGoogle Scholar
- 11.J. Bouchami, A. Gutiérrez, T. Holy, A. Houdayera, J. Jakbek, C. Lebela, C. Leroy, J. Macanaa, J.-P. Martina, S. Pospíšil, S. Praka, P. Sabellaa, and C. Teyssiera, “Measurement of pattern recognition efficiency of tracks generated by ionizing radiation in a Medipix2 device,” Nucl. Instrum. Methods Phys. Res. A 633, 187–189 (2011).ADSCrossRefGoogle Scholar
- 14.ROOT data analysis framework. https://rootcernch/Google Scholar