Evaluation of a filling system for binary pediatric solutions.
- 42 Downloads
The aim of this study was to assess the use of an automatic filling system (Siframix™ M31 and M32 system) to prepare pediatric parenteral nutrition. Volumetric accuracy was measured for each siframix™ system loads cells (< 5% for 5 ml with the Siframix™ M32 and < 5% for 9 ml with the Siframix™ M31) with sterile water for injection. The minimal 20 ml of flushing sterile water of the common tubulure of the Siframix™ M32 (p=0.211 for 20 ml and p=0.75 for 500 ml), the use of viscous solutions (70 % dextrose) on the Siframix™ M31 (p=0.28 for 20 ml and p=0.12 for 500 ml) and the use of a special tubulure for using E.V.A. Luer‐lock bags (p=0.89 for 20 ml and p=0.103 for 500 ml) do not modify the accuracy. Changing bags or bottles during the filling operation modify the accuracy (p=0.004 for 20 ml and p=0.009 for 500 ml). A flushing operation is necessary to lower the risk of electrolytic pollution for the filling of little bags. The filling speed for each module was also measured (the maximal filling speed was five liters per minute). The Siframix™ system allows one to prepare pediatric parenteral nutrition bags when volumes are above 4 ml and with adapted source solutions in terms of concentration and conditioning volumes.
Unable to display preview. Download preview PDF.
- 1.Choux C, Arnaud P, Brion F. Validation d'une pompe péristaltique et d'un filtre à usage unique dans la technique de filtration stérilisante sous flux laminaire pour la fabrication de mélanges binaires pour la nutrition parentérale conditionnés en poches. Act Pharm Biol Clin 1990;5: 461–2.Google Scholar
- 4.Yu AY, Willcox GS, Jeffrey LP. Effectiveness of a volumetric pump for preparing sterile products. Am J Hosp Pharm 1983;40: 821–5.Google Scholar
- 5.Heide PE. Maschinelle herstellung von mischinfusionen: erfahrungen mit den geräten MC Gaw Hyper former, Medimix 4001 und Sifra-Mix M31 und 32. Krankenhauspharmazie 1993;11:505–8.Google Scholar
- 7.Davis NM. Unprecedented procedural safeguards needed with the use of automated I.V. compounders. Hosp Pharm 1992;27:488.Google Scholar
- 8.Maroun C, Moura R, Velez E. Applying continuous quality improvement to the preparation of parenteral nutrition solutions using automated compounders. 27th A.S.H.P. Midyear Clinical Meeting. December 1992.Google Scholar
- 9.Stafford SE, Thomson CL, Schmidt GL, Magee MJ. Proactive approach to assure accurate neonatal parenteral nutrition solutions using refractive index measurements. 50th A.S.H.P. Annual Meeting. June 1993.Google Scholar
- 10.Murphy C. Ensuring accuracy in the use of automatic compounders. Am J Hosp Pharm 1993;50:60.Google Scholar
- 12.Gura KM, Chan L, McRae M, Petrosillo L. Impact of bar code technology on a computerized automated TPN microcompounding system. 50th A.S.H.P.Annual Meeting. June 1993.Google Scholar