An improved torque type gravity gradiometer with dynamic modulation
Traditional torque type gravity gradiometer has an important pole in gravity gradient measurements, while it is relatively inefficient and with the precision of about 1 E mainly caused by the static operating mode. In this paper, we develop an improved torque type gravity gradiometer to improve the measuring efficiency, which is based on the dynamic modulation. The dynamic modulation keeps the gradiometer rotating on a turntable steadily, measures the deflection angle of the torsion pendulum continuously and then obtains the gravity gradients. The result shows that after using the improved gradiometer, the gradients W xz and W yz are obtained with precisions of 0.45 E and 0.32 E respectively in a cycle of 20 min.
KeywordsImproved torque type gradiometer Dynamic modulation Measuring efficiency Gravity gradient
This work is partially supported by the National Natural Science Foundation of China (Grant Nos. 11575160, 91636221, 11605065).
- Barton DC (1931) Gravity measurements with the Eotvos torsion balance. Consulting Geologist and Geophysicist, Houston, Texas. Physics of the Earth 77:167–188Google Scholar
- Choi KY (2006) A new equivalence principle test using a rotating torsion balance. Ph.DGoogle Scholar
- Dehlinger P (1978) Marine gravity, vol 22. Elsevier, AmsterdamGoogle Scholar
- ELCOMAT 3000, Germany Moeller-Wedel Optical GmbH. https://www.haag-streit.com/de/moeller-wedel-optical/
- Goodfellow Cambridge Limited, Huntingdon PE29 6WR, England. http://www.goodfellow.com/contact-us/
- Landau LD (ed.) (2013) The classical theory of fields, vol 2. Elsevier, Amsterdam Google Scholar
- Schweydar W (1918) Die Bedeutung der Drehwaage von Eötvös für die geologische Forschung nebst Mitteilung der Ergebnisse einiger Messungen. Zeitschrift für praktische Geologie 26:157–162Google Scholar
- SP-400, Shanghai Shanjin Vacuum Equipment Co., Ltd., Shanghai, ChinaGoogle Scholar
- Su Y (1992) A new test of the weak equivalence principle. Ph.DGoogle Scholar