State of the Prescription Process for Dynamic Ankle-Foot Orthoses
- 72 Downloads
Purpose of Review
Dynamic ankle-foot orthoses (AFOs) are assistive devices that can be prescribed to individuals with mobility limitations to support and align joints. Dynamic AFOs are a category of passive AFOs that can control ankle motion to address both ankle joint range of motion limitations and ankle muscle weakness. This control of motion is achieved through the dynamic AFO’s functional characteristics, namely bending stiffness, which need to be customized to each individual’s needs. However, current conventions for customizing dynamic AFOs for each individual are variable and often not clearly documented. The purpose of this review was to synthesize the current state of customizing functional characteristics of dynamic AFOs to provide a foundation for ultimately optimizing the prescription of AFOs.
Dynamic AFO bending stiffness, bending axis, alignment, and footplate design were identified as key functional characteristics that can be customized. Studies showed that customizing these dynamic AFO functional characteristics has the ability to alter gait, and customizing multiple functional characteristics at once is key to improving individual outcomes.
Researchers have continued to expand their knowledge on how dynamic AFO functional characteristics can impact individual outcomes. Continued research should work towards developing guidelines for prescribing dynamic AFO functional characteristics based on individuals’ level of needs. Additionally, researchers and clinicians need to work together to ultimately translate these scientific findings into clinical practice.
KeywordsAnkle-foot orthosis Stiffness Alignment Axis Footplate
Compliance with Ethical Standards
Conflict of Interest
Corey Koller declares no conflicts of interest.
Elisa S. Arch has a United States Patent No. 8,538,570 issued.
Human and Animal Rights and Informed Consent
All reported studies/experiments with human or animal subjects performed by the authors have been previously published and complied with all applicable ethical standards (including the Helsinki declaration and its amendments, institutional/national research committee standards, and international/national/institutional guidelines).
Papers of particular interest, published recently, have been highlighted as: • Of importance •• Of major importance
- 1.Courtney-Long E, Carroll D, Zhang Q, Stevens A, Griffin-Blake S, Armour B, et al. Prevalence of disability and disability type among adults — United States, 2013 [Internet]. 2015. Available from: https://www.cdc.gov/mmwr/preview/mmwrhtml/mm6429a2.htm. Accessed Oct 5 2017
- 6.AOPA Fact Sheet [Internet]. AOPA Am. Orthotic Prosthet. Assoc. Available from: http://www.aopanet.org/media/fact-sheet/. Accessed 5 Oct 2017.
- 8.DaVanzo J, El-gamil A, Heath S, Pal S, Li J, Luu P, et al. Projecting the adequacy of workforce supply to meet patient demand. [Internet]. 2015. Available from: http://www.iiofoandp.org/PDF/2015_Work_Study.pdf. Accessed 5 Oct 2017.
- 9.Whiteside S, Allen M, Barringer W, Beiswenger W, Brncick M, Bulgarelli T, et al. Practice analysis of certified practitioners in the disciplines of orthotics and prosthetics. Alexandria Am. Board Certif. Orthot. Prosthetics Inc. [Internet]. 2007. Available from: https://www.abcop.org/individual-certification/Documents/ABCPracticeAnalysisoftheDisciplineofOrthoticsandProsthetics.pdf. Accessed 5 Oct 2017.
- 14.Shorter KA, Xia J, Hsiao-Wecksler ET, Durfee WK, Kogler GF. Technologies for powered ankle-foot orthotic systems: possibilities and challenges. IEEE/ASME Trans. Mechatronics. 2013; p. 337–47.Google Scholar
- 17.Bregman DJJ, Rozumalski A, Koops D, de Groot V, Schwartz M, Harlaar J. A new method for evaluating ankle foot orthosis characteristics: BRUCE. Gait Posture [Internet]. 2009;30:144–9. Available from: http://www.sciencedirect.com/science/article/pii/S0966636209001465. Accessed Oct 5 2017
- 18.Sumiya T, Suzuki Y, Kasahara T. Stiffness control in posterior-type plastic ankle-foot orthoses: effect of ankle trimline. Part 2: orthosis characteristics and orthosis/patient matching. Prosthet Orthot Int [Internet]. 1996;20:132–7. Available from: http://www.ncbi.nlm.nih.gov/pubmed/8876008. Accessed Oct 5 2017
- 22.• Arch E, Reisman DS. Passive-dynamic ankle-foot orthoses with personalized bending stiffness can enhance net plantarflexor function for individuals post-stroke. J Prosthetics Orthot. 2016;28:60–7. This study investigated if dynamic AFO bending stiffness customized to account for an individual’s level of plantarflexor deficit could replace lost plantarflexor function for individuals post-stroke. This study indicated that while walking with dynamic AFOs with a customized bending stiffness, the net (individual + AFO) peak plantarflexion moment increased; however, typical moments were not fully reached. This study supports the feasibility of objectively personalizing dynamic AFO bending stiffness to assist with plantarflexor weakness in individuals post-stroke . CrossRefGoogle Scholar
- 25.Singer ML, Kobayashi T, Lincoln LS, Orendurff MS, Foreman KB. The effect of ankle-foot orthosis plantarflexion stiffness on ankle and knee joint kinematics and kinetics during first and second rockers of gait in individuals with stroke. Clin Biomech Elsevier Ltd. 2014;29:1077–80.CrossRefGoogle Scholar
- 26.Kobayashi T, Singer ML, Orendurff MS, Gao F, Daly WK, Foreman KB. The effect of changing plantarflexion resistive moment of an articulated ankle-foot orthosis on ankle and knee joint angles and moments while walking in patients post stroke. Clin Biomech Elsevier Ltd. 2015;30:775–80.CrossRefGoogle Scholar
- 28.Perry J, Burnfield JM. Gait analysis: normal and pathological function. 2nd ed. Thorofare: Slack Inc.; 2010.Google Scholar
- 30.Kerkum YL, Buizer AI, Van Den Noort JC, Becher JG, Harlaar J, Brehm MA. The effects of varying ankle foot orthosis stiffness on gait in children with spastic cerebral palsy who walk with excessive knee flexion. PLoS One. 2015;10.Google Scholar
- 31.• Kerkum YL, Harlaar J, Buizer AI, Van Den Noort JC, Becher JG, Brehm MA. An individual approach for optimizing ankle-foot orthoses to improve mobility in children with spastic cerebral palsy walking with excessive knee flexion. Gait Posture Elsevier BV. 2016;46:104–11. This study investigated the efficacy of a ventral dynamic AFO to reduce knee flexion and walking energy cost by customizing dynamic AFO stiffness in children with CP. Results showed a median reduction of 9% in net energy cost and reduced knee flexion when compared to a shoes-only condition. This study used objective parameters, such as peak knee extension angle during single support and walking energy cost, to customize stiffness.CrossRefGoogle Scholar
- 32.•• Kobayashi T, Orendurff MS, Hunt G, Lincoln LS, Gao F, LeCursi N, et al. An articulated ankle–foot orthosis with adjustable plantarflexion resistance, dorsiflexion resistance and alignment: a pilot study on mechanical properties and effects on stroke hemiparetic gait. Med Eng Phys Elsevier Ltd. 2017;44:94–101. This study analyzed the effect plantarflexion stiffness, dorsiflexion stiffness, and alignment had on ankle and knee joint kinematics and kinetics in individuals post-stroke. This study found that optimizing dynamic AFO alignment can improve the heel rocker to achieve normal kinematics and kinetics, tuning plantarflexion stiffness can optimize the heel rocker, and dorsiflexion stiffnesses can reduce peak push-off ankle power. This study customized two functional characteristics, bending stiffness and alignment.CrossRefGoogle Scholar
- 37.Stanhope S, Schrank E. Process and system for manufacturing a customized orthosis. United States Patent No 8,538,570; 2013.Google Scholar
- 38.Sumiya T, Suzuki Y, Kasahara T, Ogata H. Instantaneous centers of rotation in dorsi/plantar flexion movements of posterior-type plastic ankle-foot orthoses. J Rehabil Res Dev. 1997;34: 279–85.Google Scholar
- 42.Owen E. Shank angle to floor measures of tuned “ankle-foot orthosis footwear combinations” used with children with cerebral palsy, spina bifida and other conditions. Gait Posture. 2002;16:132–3.Google Scholar
- 44.Owen E. A clinical algorithm for the design and tuning of AFO footwear combinations based on shank kinematics. 12th World Congr. Int. Soc. Prosthetics Orthot; 2007, p. 181.Google Scholar
- 47.Fatone S, Gard SA, Malas BS. Effect of ankle-foot orthosis alignment and foot-plate length on the gait of adults with poststroke hemiplegia. Arch Phys Med Rehabil. the American Congress of Rehabilitation Medicine and the American Academy of Physical Medicine and Rehabilitation. 2009;90:810–8.CrossRefPubMedGoogle Scholar