Auditory brainstem implant (ABI) is used to provide auditory sensations in patients with neurofibromatosis type 2 who lost their hearing due to a surgical removal of the tumor. ABI surgery, implant activation and follow-up sessions present unique challenges including the exact placement of the electrode pad in the lateral recess of the IVth ventricle, identification of electrodes that trigger non-auditory sensation and their deactivation which lowers the number of electrodes responsible for hearing, changes of T- and C-levels across follow-up sessions. We present a complete procedure using an example case starting from the surgical part with the detailed description of intraoperative eABR measurement as a guidance for pad placement to the ABI activation and first fitting sessions with auditory sensation assessment. Since the first ABI electrode pad position presented non-satisfactory intraoperative eABR results it was decided to move the pad slightly which resulted in better eABR (more electrodes with auditory responses). The discussed patient demonstrated great auditory and speech perception results after the first ABI fitting (which included three sessions over 2 consecutive days). Repositioning of the ABI electrode pad during the surgery was carried out taking into account the intraoperative eABR results and this led to an overall positive outcome for the patient. The placement of ABI electrode pad is crucial for later auditory results. This study provides detailed insight in this very specialized procedure that is not performed in every clinic and adds to the knowledge of intraoperative navigation using eABR measurements during ABI surgery.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Tax calculation will be finalised during checkout.
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
Tax calculation will be finalised during checkout.
Waring MD (1995) Intraoperative electrophysiologic monitoring to assist placement of auditory brain stem implant. Ann Otol Rhinol Laryngol Suppl 166:33–36
Waring MD (1996) Properties of auditory brainstem responses evoked by intra-operative electrical stimulation of the cochlear nucleus in human subjects. Electroencephalogr Clin Neurophysiol 100(6):538–548
Nevison B, Laszig R, Sollmann WP, Lenarz T, Sterkers O, Ramsden R, Fraysse B, Manrique M, Rask-Andersen H, Garcia-Ibanez E, Colletti V, von Wallenberg E (2002) Results from a European clinical investigation of the nucleus multichannel auditory brainstem implant. Ear Hear 23(3):170–183
Nevison B (2006) A guide to the positioning of brainstem implants using intraoperative electrical auditory brainstem responses. Adv Otorhinolaryngol 64:154–166. https://doi.org/10.1159/000094650
O’Driscoll M, El-Deredy W, Ramsden RT (2011) Brain stem responses evoked by stimulation of the mature cochlear nucleus with an auditory brain stem implant. Ear Hear 32(3):286–299. https://doi.org/10.1097/AUD.0b013e3181fc9d72
Herrmann BS, Brown MC, Eddington DK, Hancock KE, Lee DJ (2015) Auditory brainstem implant: electrophysiologic responses and subject perception. Ear Hear 36(3):368–376. https://doi.org/10.1097/AUD.0000000000000126
Puram SV, Herrmann B, Barker FG 2nd, Lee DJ (2015) Retrosigmoid craniotomy for auditory brainstem implantation in adult patients with neurofibromatosis type 2. J Neurol Surg B Skull Base 76(6):440–450. https://doi.org/10.1055/s-0034-1544121
Barber SR, Kozin ED, Remenschneider AK, Puram SV, Smith M, Herrmann BS, Cunnane ME, Brown MC, Lee DJ (2017) Auditory brainstem implant array position varies widely among adult and pediatric patients and is associated with perception. Ear Hear 38(6):e343–e351. https://doi.org/10.1097/AUD.0000000000000448
Colletti V, Shannon RV, Carner M, Veronese S, Colletti L (2010) Complications in auditory brainstem implant surgery in adults and children. Otol Neurotol 31(4):558–564. https://doi.org/10.1097/MAO.0b013e3181db7055
Behr R, Colletti V, Matthies C, Morita A, Nakatomi H, Dominique L, Darrouzet V, Brill S, Shehata-Dieler W, Lorens A, Skarzynski H (2014) New outcomes with auditory brainstem implants in NF2 patients. Otol Neurotol 35(10):1844–1851. https://doi.org/10.1097/MAO.0000000000000584
Chatterjee M (1999) Effects of stimulation mode on threshold and loudness growth in multielectrode cochlear implants. J Acoust Soc Am 105(2 Pt 1):850–860
Galvin JJ 3rd, Fu QJ (2009) Influence of stimulation rate and loudness growth on modulation detection and intensity discrimination in cochlear implant users. Hear Res 250(1–2):46–54. https://doi.org/10.1016/j.heares.2009.01.009
Sanna M, Di Lella F, Guida M, Merkus P (2012) Auditory brainstem implants in NF2 patients: results and review of the literature. Otol Neurotol 33(2):154–164. https://doi.org/10.1097/MAO.0b013e318241bc71
Deep NL, Choudhury B, Roland JT Jr (2019) Auditory brainstem implantation: an overview. J Neurol Surg B Skull Base 80(2):203–208. https://doi.org/10.1055/s-0039-1679891
Wong K, Kozin ED, Kanumuri VV, Vachicouras N, Miller J, Lacour S, Brown MC, Lee DJ (2019) Auditory brainstem implants: recent progress and future perspectives. Front Neurosci 13:10. https://doi.org/10.3389/fnins.2019.00010
Anwar A, Singleton A, Fang Y, Wang B, Shapiro W, Roland JT Jr, Waltzman SB (2017) The value of intraoperative EABRs in auditory brainstem implantation. Int J Pediatr Otorhinolaryngol 101:158–163. https://doi.org/10.1016/j.ijporl.2017.08.007
Moore JK (1987) The human auditory brain stem as a generator of auditory evoked potentials. Hear Res 29(1):33–43
Waring MD (1995) Auditory brain-stem responses evoked by electrical stimulation of the cochlear nucleus in human subjects. Electroencephalogr Clin Neurophysiol 96(4):338–347
Waring MD (1998) Refractory properties of auditory brain-stem responses evoked by electrical stimulation of human cochlear nucleus: evidence of neural generators. Electroencephalogr Clin Neurophysiol 108(4):331–344
Frohne C, Matthies C, Lesinski-Schiedat A, Battmer RD, Samii M, Lenarz T (2000) Extensive monitoring during auditory brainstem implant surgery. J Laryngol Otol Suppl 27:11–14
We would like to thank Tomáš Tichý from Cochlear Company for his assistance with intraoperative measurements and ABI activation sessions.
Conflict of interest
All authors declare that they have no conflict of interest.
Research involving human participants
All procedures performed in this study involving human participant were in accordance with the ethical standards of the institutional research committee (the local Institutional Ethics Committee Review Board, reference number AKBE/175/2018) and with the 1964 Helsinki Declaration and its later amendments or comparable ethical standards.
One patient participated in this study—the patient has given full written and informed consent, both for the participation and for the publication of the report.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Lachowska, M., Pastuszka, A., Mikołajewska, L. et al. Detailed insight in intraoperative eABR measurements to assist auditory brainstem implantation in a patient with neurofibromatosis type 2. Acta Neurol Belg 120, 1371–1378 (2020). https://doi.org/10.1007/s13760-019-01248-7
- Hearing loss
- Evoked potentials
- Cochlear nerve
- Cochlear nucleus
- Auditory brainstem implant