Emerging Technologies for Measuring Flow Rate and Density of Bulk Flows of Particulate Solids
- 151 Downloads
The operating principles and characteristics of three devices for measuring or monitoring flows of bulk solids are outlined. These are a flowrate meter and a density meter for material in gravity flow, both based on load cell sensors, and a system, based on audio-frequency acoustic sensing, for measuring the flow rate of dilute flows in a pneumatic conveying pipeline. Common features are mechanical simplicity and robustness, ease of calibration, and high precision. These technologies have all been demonstrated under industrial conditions performing reliably and consistently, and are compatible with the wide ranging process requirements of different industrial sectors, including the minerals industry.
KeywordsBulk solids Flow rate Bulk density Measurement Pneumatic conveying
The developments reported here were carried out over several years with many contributors. Special thanks are to Nicola Brown, Kristina Fenton, John Grey, Bob Lankshear and Stephen Tallon. Much of the development of the acoustic flow meter was supported by funding from Foundation for Research Science and Technology, Wellington, New Zealand. The author thanks the Organizing Committee of MPT 2016 for support and encouragement and for the opportunity to contribute this paper to the A K Biswas Memorial Volume.
- 1.Kuo Chin S, The Mineral Industry of India. http://minerals.usgs.gov/minerals/pubs/country/2012/myb3-2012-in.pdf.
- 2.Knoema, (2016) https://knoema.com/nwnfkne/world-gdp-ranking-2016-data-and-charts-forecast. 14 October 2016.
- 3.Davies C E, in Book of Abstracts, International Seminar on Mineral Processing Technology, TCS Sahyadri Park Campus, Hinjewadi. 5–7 January (2016).Google Scholar
- 4.Coulson, J M, and Richardson, J F, Chemical Engineering Volume 1, Pergamon Press, Oxford (1965), p 125.Google Scholar
- 6.Brown R L, and Richards J C, Trans Inst Chem Eng 38 (1960) 243.Google Scholar
- 8.Dhodapkar S, Jacob K V, and Kodam M, Determining Discharge Rates of Particulate Solids, Chemical Engineering Progress, May (2016) 50.Google Scholar
- 9.Davies C E, and Harris B, IPENZ Trans 20 (1993) 42.Google Scholar
- 10.Davies C E, and Harris B, in Fluidization VII, Proceedings of Seventh Engineering Foundation Conference on Fluidization, Brisbane, (eds) Potter O E, and Nicklin D J, Engineering Foundation, New York (1992), 3–8 May 1992.Google Scholar
- 11.Davies C E, Dayal J, and Fenton K, in Proc APCChE & CHEMECA 93, 21st Australasian Chemical Engineering Conference, Melbourne, Australia (1993) p 223.Google Scholar
- 12.Davies C E, Tallon S, Grey, J, and Miller F, in Proc 5th International Conference on Bulk Materials Storage, Handling and Transportation, Newcastle, Australia (1995) 10–12 July, p 493.Google Scholar
- 13.Anon., NZS 3111: Method of Test for Water and Aggregate for Concrete, Standards Association of New Zealand, Wellington (1980).Google Scholar
- 14.Davies C E, Brown N, Tallon S, and Petersen M, in Proc. 6th World Congress of Chemical Engineering, Melbourne, Australia, 23–27 September (2001).Google Scholar
- 16.Tallon S, and Davies C E, in Advances in Fluidization and Fluid-Particle Systems, AIChE Symposium Series 317 93 (1997), p 136.Google Scholar
- 17.Tallon S, Davies C E, Levy E, and Yilmaz A, in Proceedings of 2001 Joint Conference of SCENZ/FEANZ/EMG, Auckland, New Zealand (2001) 9 and 10 April.Google Scholar
- 18.Tallon S, and Davies C E, Powder Handl Process, 13 (2001) 55.Google Scholar