There is an increasing demand for the large-scale force standard machine with a capacity higher than 20 MN. In this paper, the calibration method and procedure were developed according to a proposed traceability route and used to calibrate a 60 MN force standard machine. One 7.5 MN and six 20 MN force transducers and two 60 MN build-up systems were used as the force transfer standards. Uncertainty models of the build-up system and force standard machine were also developed. The calibration results show that it is possible to calibrate a 60 MN build-up force standard machine with a relative expanded measurement uncertainty smaller than 0.035% (k = 2) by two build-up systems, which were compared with themselves. The 60 MN build-up system, which worked as the force transfer standard, should be calibrated to confirm its systematic error and corrected by the error. Its uncertainty also could be estimated independently by the developed model.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
A. Schäfer, Examples and proposed solutions regarding the growing importance of calibration of high nominal forces, in: IMEKO 21st conference on measurement of force, mass and torque, Pattaya, Chonburi, Thailand, 2010.
R. Kumme, F. Tegtmeier, D. Röske, A. Barthel, A. Germak and P. Averlant, Force traceability within the Meganewton range, in: IMEKO 22nd conference on the measurement of force, mass and torque, Cape Town, Republic of South Africa, 2014, pp. 1–2.
L. J. B. Maybank and A. J. Knott, Uncertainty of force measurement in build-up procedures, in: XV IMEKO world congress measurement to improve the quality of life in the 21st century - measurement helps to coordinate nature with human activities, Osaka, Japan, 1999.
M. Wagner, A. Tsiporenko and F. Tegtmeier, Development of a 15 MN build-up system for calibrations in UKRAINE´s heavy industry, in: XXI IMEKO world congress measurement in research and industry, Prague, Czech Republic, 2015. http://www.imeko.org/publications/wc-2015/IMEKO-WC-2015-TC3-090.pdf.
S. L. Yaniv and T. W. Bartel, Force metrology at NIST, in: XVI IMEKO world congress, Vienna, AUSTRIA, 2000.
T. W. Bartel, R. L. Seifarth and S. L. Yaniv, Force measurement services at NIST: equipment, procedures, and uncertainty, in: 1997 NCSL workshop and symposium, West Yorkshire, UK, 1997.
J. H. Yao, Q. Z. Li, H. Xu, H. Chi and J. H. Lin, Design of 60 MN build-up force standard machine, in: IMEKO 22nd conference on the measurement of force, mass and torque, Cape Town, Republic of South Africa, 2014. http://www.imeko.org/publications/tc3-2014/IMEKO-TC3-2014-002.pdf.
J. Yao, A. Knott, Q. Li, H. Xu and H. Li, Intercomparison between large force standard machines in China and UK, in: XXI IMEKO world congress measurement in research and industry, Prague, Czech Republic, 2015.
W. Liang, F. Tegtmeier, J. Yao and H. Xu, Force comparison between FJIM and PTB up to 16.5 MN, in: IMEKO 23rd conference on the measurement of force, mass and torque measurement facing new challenges, Helsinki, Finland, 2017.
EURAMET/cg -04/v.01, Calibration guide of uncertainty of force measurements.
D. Kang, H. Song, J. Lee, Y. Park and J. Kim, Traceability of large force standards in Korea, in: XVII IMEKO World congress metrology in the 3rd millennium, Dubrovnik, Croatia, 2003, pp. 393–396.
D. I. Kang, H. K. Song and J. T. Lee, Establishment of large force standards based on built-in force transducers and build-up system, in: XVI IMEKO World congress measurement - supports science - improves technology - protects environment and provides employment - now and in the future, Torino, Italy, 2000.
C. Ferrero, C. Marinari and E. Martino, Development and metrological characterisation of a build-up force standard up to 3 MN, in: XVII IMEKO World congress metrology in the 3rd millennium, Dubrovnik, Croatia, 2003, pp. 251–254.
M. Wagner and F. L. Tegtmeier, Processing and evaluation of build-up system measurement data, in: XXI IMEKO world congress measurement in research and industry, Prague, Czech Republic, 2015.
F. Tegtmeier, M. Wagner and R. Kumme, Investigations of transfer standards in the highest range up to 50 MM within EMRP Project SIB 63, in: XXI IMEKO world congress measurement in research and industry, Prague, Czech Republic, 2015.
F. Tegtmeier, D. Rosk and W. Liang, Practical applications of an enhanced uncertainty model for build-up systems, in: IMEKO 23rd conference on the measurement of force, mass and torque measurement facing new challenges, Helsinki, Finland, 2017. http://www.imeko.org/publications/tc3-2017/IMEKO-TC3-2017-012.pdf.
ISO 376-2011, Metallic materials - calibration of force proving instruments used for the verification of uniaxial testing machines.
JJG 1116-2015, Calibration regulation of build-up force standard machine.
H. Chi, The 60 MN build-up force delivery system based on compensating structure, Acta Metrol. Sin., 36 (2015) 221–224. http://dx.doi.org/10.3969/j.issn.1000-1158.2015.02.24.
Publishable JRP Summary Report for SIB63 Force Force traceability within the meganewton range. Last accessed: 13 Feb 2017.
This work was supported by Science and Technology Planning Project of Quality Technical Supervision and Inspection and Quarantine Administration [Grant Numbers: 2015QK295]. The authors kindly thank PTB for their support of the force comparison up to 50 MN.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Liang, W., Yang, X., Yao, J. et al. Calibration of Force Standard Machine by Build-up Force Transfer System in the Highest Range Up To 60 MN. MAPAN 35, 151–164 (2020). https://doi.org/10.1007/s12647-019-00362-4
- Force standard machine
- Build-up system