Pedicle screw fixation in thoracolumbar and lumbar spine assisted by lateral fluoroscopic imaging: a study to evaluate the accuracy of screw placement
- 305 Downloads
The purpose of this study was to evaluate the accuracy of pedicle screw placement, its advantages, and limitations in posterior instrumentation of thoracolumbar and lumbar burst fractures assisted only by lateral fluoroscopic imaging.
Materials and methods
Pre- and postoperative computerized tomographic (CT) scans of 117 patients with thoracolumbar and lumbar burst fractures, who underwent posterior instrumentation with pedicle screw fixation, were prospectively analyzed. Accuracy of screw placement, reconstruction of the vertebral height, and correction of the kyphotic angle were studied. Position of the pedicle screws were determined, and cortical breach was graded on the postoperative axial CT scans. Percentage of vertebral height reconstruction and kyphotic angle correction were calculated from the postoperative midsagittal CT scans.
Four hundred and sixty-eight pedicle screws in 234 motion segments were included in this study. 427 screws were centrally placed with an accuracy rate of 91.24%. Out of the 41 (8.76%) screws that breached the pedicle wall, 32 (6.84%) screws had violated the medial wall, while 9 (1.92%) screws breached the lateral wall. There were no “air-ball” screws. No screw penetrated the anterior wall. Postoperatively, none of the patients deteriorated neurologically, and no screw required revision. Postoperatively, there was significant restoration of vertebral height and correction of kyphosis (P < 0.05).
Pedicle fixation performed on a Relton-Hall frame is relatively simple and, when performed carefully using only lateral fluoroscopic imaging, has a lower potential for complications due to cortical breach.
KeywordsSpinal injury Burst fractures Pedicle screws Computed tomography
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Declaration of Helsinki and its later amendments or comparable ethical standards. Institutional ethical clearance was obtained before starting the study.
Informed consent was obtained from all individual participants included in the study.
Human and animal rights
This article does not contain any studies with animals performed by any of the authors.
- 1.Steffee AD, Biscup RS, Sitkowskj DJ (1986) Segmental spine plates with pedicle screw fixation a new internal fixation device for disorders of the lumbar and thoracolumbar spine. Clin Orthop Relat Res 203:45–53Google Scholar
- 7.Amiot LP, Lang K, Zippel H, Labelle H (1998) Comparative accuracy between conventional and computer-assisted pedicle screw installation. Presented at the annual meeting of the international society for the study of the lumbar spine, Brussels, BelgiumGoogle Scholar
- 12.Cobb JR (1948) Outline for the study of scoliosis. In: Edwards JW (ed) Instructional course lectures, vol 5. The American Academy of Orthopedic Surgeons, Ann Arbor, pp 261–275Google Scholar
- 18.Zou D, Yoo JU, Edwards WT, Donovan DM, Chang KW, Bayley JC, Fredrickson BE, Yuan HA (1993) Mechanics of anatomic reduction of thoracolumbar burst fractures| comparison of distraction versus distraction plus lordosis, in the anatomic reduction of the thoracolumbar burst fracture. Spine 18(2):195–203CrossRefPubMedGoogle Scholar