Determining the Fat Concentration of Fresh Raw Cow Milk Using Dielectric Spectroscopy Combined with Chemometrics
- 110 Downloads
Dielectric spectroscopy is usually used to obtain dielectric properties of materials. It not only has the advantages of high speed and little or no sample preparation but can also be used in in-line or in situ quality monitoring. To provide a quick and convenient method for milk fat concentration determination using dielectric spectroscopy, the dielectric spectra of 143 fresh raw cow milk samples were obtained from 20 to 4500 MHz. All samples were divided into calibration set and prediction set by using joint x-y distances sample set partitioning method. One hundred twenty-three, 9, and 40 dielectric variables were selected as effective variables (EVs) by applying uninformative variable elimination, successive projection algorithm, and stability competitive adaptive reweighted sampling methods, respectively. The models of partial least squares regression (PLSR), extreme learning machine (ELM), and least squares-supporting vector machine (LS-SVM) were built when the full dielectric spectra and selected EVs were used as inputs. The results showed that the PLSR model developed with full dielectric spectra was the best model on fat concentration determination with the root-mean-squares error of prediction set of 0.168%. The study tells that dielectric spectra have potential in exploring a milk fat concentration sensor used for in-line or in situ detection.
KeywordsMilk Fat Determination Dielectric spectra Dielectric spectroscopy
The study was financially supported by the project of National Natural Science Foundation of China (No. 31671935).
Compliance with Ethical Standards
Conflict of Interest
Xinhua Zhu declares that he has no conflict of interest. Wenchuan Guo declares that he has no conflict of interest. Dayang Liu declares that he has no conflict of interest. Kang Fei declares that he has no conflict of interest.
All experiments involving animals were conducted according to the principles of the Chinese Academy of Agricultural Sciences Animal Care and Use Committee (Beijing, China), which approved the study protocols.
Informed consent is not applicable in this study.
- Feng X, Su R, Xu N, Wang X, Yu A, Zhang H, Cao Y (2013) Portable analyzer for rapid analysis of total protein, fat and lactose contents in raw milk measured by non-dispersive short-wave near-infrared spectrometry. Chem Res Chin Univ 29(1):15–19. https://doi.org/10.1007/s40242-013-2191-y CrossRefGoogle Scholar
- Jiang H, Zhang H, Chen Q, Mei C, Liu G (2015) Identification of solid state fermentation degree with FT-NIR spectroscopy: comparison of wavelength variable selection methods of CARS and SCARS. Spectrochim Acta A Mol Biomol Spectrosc 149:1–7. https://doi.org/10.1016/j.saa.2015.04.024 CrossRefGoogle Scholar
- Melfsen A, Hartung E, Haeussermann A (2012b) Accuracy of milk composition analysis with near infrared spectroscopy in diffuse reflection mode. Biosyst Eng 112(3):210–217. https://doi.org/10.1016/j.biosystemseng.2012.04.003 CrossRefGoogle Scholar
- Tsenkova R, Atanassova S, Itoh K, Ozaki Y, Toyoda K (2000) Near infrared spectroscopy for biomonitoring: cow milk composition measurement in a spectral region from 1,100 to 2,400 nanometers. J Anim Sci 78(3):515–522. https://doi.org/10.2527/2000.783515x
- Woo YA, Terazawa Y, Chen JY, Iyo C, Terada F, Kawano S (2002) Development of a new measurement unit (MilkSpec-1) for rapid determination of fat, lactose, and protein in raw milk using near-infrared transmittance spectroscopy. Appl Spectrosc 56(5):599–604. https://doi.org/10.1366/0003702021955150 CrossRefGoogle Scholar