Quantitation of Saccharin and Cyclamate in Tabletop Formulations by Portable Raman and NIR Spectrometers in Combination with Partial Least Squares Regression
- 182 Downloads
Rapid, direct, and reagent-free screening tools using vibrational spectroscopy in combination with partial least squares regression (PLSR) were developed for the determination of sodium saccharin and sodium cyclamate in tabletop formulations. The four vibrational spectroscopic instruments employed were a portable Raman spectrometer, a NIR handheld device, and two FT-NIR benchtop spectrometers. Wavenumber ranges and type of spectral pretreatment were optimized for each PLSR calibration model using an independent validation set. Each sweetener model provided reliable predictions (low errors in validation and r 2 above 0.90) for both saccharin and cyclamate samples. Optimized models were tested with four commercially available tabletop formulations in order to simulate the application of the developed models towards routine sweetener analysis. With the exception of one model, the sweetener concentration predictions in commercial tabletop formulations using the portable devices were not significantly different from those based on spectra collected on the benchtop spectrometers. PLSR-predicted mean sweetener concentrations were within 80–120% of the label declared values, while the sweetener without a label declaration had consistent concentrations across the analytical methods used. As shown by the good agreement between spectroscopic and chromatographic analyses, the portable spectrometers offer an alternative to traditional chromatographic methods. To our knowledge, this is the first time portable Raman and handheld NIR devices with PLSR calibration models have been employed to evaluate sweeteners, and these analytical methods hold potential to be used for rapid screening of tabletop formulations for quality assurance and for regulatory labeling verification.
KeywordsSweetener Saccharin Cyclamate Partial least squares regression (PLSR) Vibrational spectroscopy Portable devices
SRK and SF acknowledge the support provided by an appointment to the Research Participation Program at the Center for Food Safety and the Applied Nutrition, administered by the Oak Ridge Institute for Science and Education through an interagency agreement between the US Department of Energy and the US Food and Drug Administration (US FDA).
TJM and KH were funded by the University of Maryland Joint Institute for Food Safety and Applied Nutrition through a cooperative agreement with the FDA, no. FDU001418.
Compliance with Ethical Standards
Conflict of Interest
Sanjeewa R. Karunathilaka declares that he has no conflict of interest. Betsy Jean Yakes declares that she has no conflict of interest. Samantha Farris declares that she has no conflict of interest. Tara Jade Michael declares that she has no conflict of interest. Keqin He declares that she has no conflict of interest. Jin Kyu Chung declares that he has no conflict of interest. Romina Shah declares that she has no conflict of interest. Magdi M. Mossoba declares that he has no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
- Bergamo AB, Da Silva JAF, De Jesus DP (2011) Simultaneous determination of aspartame, cyclamate, saccharin and acesulfame-K in soft drinks and tabletop sweetener formulations by capillary electrophoresis with capacitively coupled contactless conductivity detection. Food Chem 124:1714–1717CrossRefGoogle Scholar
- Buchgraber M, Wasik A (2008) Determination of nine intense sweeteners in foodstuffs by high-performance liquid chromatography and evaporative light-scattering detection: interlaboratory study. J AOAC Int 92:208–222Google Scholar
- Karunathilaka SR, Farris S, Mossoba MM, Moore JC, Yakes BJ (2016a) Characterizing variances of milk powder and instrumentation for development of a nontargeted, Raman spectroscopy and chemometrics detection method for the evaluation of authenticity. Food Addit Contam Part A 33:921–932CrossRefGoogle Scholar
- Lin-vein D, Colthup NB, Fateley WG, Grasselli JG (1991) The handbook of infrared and Raman characteristics frequencies of organic molecules. Academic Press, San DiagoGoogle Scholar
- Shah R, De Jager LS (2016) Recent analytical methods for the analysis of sweeteners in food: a regulatory perspective. In: Spizzirri UG, Cirillo G (eds) Food safety: innovative analytical tools for food safety assessment, 1st edn. John Wiley & Sons, Inc. Hoboken, NJ and Scrivener publishing LLC, Beverly, p 13–32Google Scholar
- U.S Food and Drug Administration (2015) Food addtitives and ingredients. Available from http://www.fda.gov/Food/IngredientsPackagingLabeling/FoodAdditivesIngredie nts/ucm397725.htm (accessed 15.02.2017)
- Vongsvivut J, Miller MR, McNaughton D, Heraud P, Barrow CJ (2014) Rapid discrimination and determination of polyunsaturated fatty acid composition in marine oils by FTIR spectroscopy and multivariate data analysis. Food Bioprocess Tech 7:2410-2422Google Scholar
- Williams PC (2001) Implementation of near-infrared technology. In: Williams P, Norris K (eds) Near-infrared technology in the agricultural and food industries, 2nd edn. American association of cereal chemists, Inc. St. Paul, p 145–169Google Scholar