Highly Sensitive Potassium-Doped Polypyrrole/Carbon Nanotube-Based Enzyme Field Effect Transistor (ENFET) for Cholesterol Detection
- 677 Downloads
Highly sensitive potassium (K)-doped carbon nanotube (CNT) and polypyrrole (PPy) nanocomposite membrane-based enzyme field effect transistor (ENFET) has been fabricated on indium tin oxide (ITO) for detection of cholesterol. P-type graphene has been deposited as substrate on ITO glass electrochemically. N-type graphene has been deposited in source and drain regions. Zirconium dioxide (ZrO2) has been deposited on the channel region as gate insulator. K/PPy/CNT composite has been deposited as sensing membrane on the top of ZrO2 layer; 1 μl of cholesterol oxidase (ChOx) has been immobilized on K/PPy/CNT membrane via physical adsorption technique. The response of K/PPy/CNT/FET has been studied using Agilent 3458A digital multimeter in presence of phosphate buffer saline (PBS) of 50 mM, pH 7.0 and 0.9 % NaCl contained in a glass pot. During measurement, 20 μl cholesterol solutions (0.5 to 25 mM) were poured into the pot through micropipette each time. It has been found that K/PPy/CNT/FET has linearly varied from 0.5 to 20 mM. The sensitivity of this FET has been found to be ~400 μA/mM/mm2 with regression coefficient (r) ~ 0.998. The proposed ENFET has response time of 1 s and stability up to 6 months. The experiment has been repeated 10 times, and only 2.0 % output variation has been observed. The limit of detection (LoD) and Michaelis-Menten constant (K m) were found to be ~1.4 and 2.5 mM, respectively. The results obtained in this work show negligible interference (3.7 %) with uric acid, glucose and urea.
KeywordsCarbon nanotube Cholesterol oxidase Enzyme field effect transistor Electrochemical deposition Potassium Graphene and zirconium dioxide
The authors thank the Council of Scientific and Industrial Research, India, for the award of Senior Research Fellowship to A. Barik (File No. 9/1099(0001)/2013-EMR-I). The authors are also thankful to Tezpur University and the University of Science and Technology, Meghalaya, for providing the facilities in the laboratory.
- 1.Baynes, J. W., & Dominiczak, M. (2005). Medical biochemistry (2nd ed.). Mosby Ltd: Elsevier.Google Scholar
- 2.White, A., Handler, P., Smith, E. L., Hill, R. L., & Lehman, I. R. (1987). Principles of biochemistry (6th ed.). New York: McGraw-Hill Book.Google Scholar
- 5.Nauck, M., Graziani, M. S., Bruton, D., et al. (1997). Multicenter evaluation of a homogeneous assay for HDL-cholesterol without sample pretreatment. Clinical Chemistry, 43, 1622–1629.Google Scholar
- 18.Sahoo, R., & Mishro, R. R. (2009). Simulations of carbon nanotube field effect transistors. International Journal of Electronic Engineering Research, 1, 117–125.Google Scholar
- 23.Neto, A. H. C., Guinea, F., Peres, N. M. R., et al. (2009). The electronic properties of graphene. Reviews of Modern Physics, 81, 1,109–163.Google Scholar
- 24.Stain, M. R., Unluer, D., & Ghosh, A. (2009). Graphene devices. Interconnect and Circuits Challenges and Opportunities. IEEE, 69–72.Google Scholar
- 30.Raicopol, M., Prună, A., Damian, C., et al. (2013). Functionalized single-walled carbon nanotubes/polypyrrole composites for amperometric glucose biosensors. Nanoscale Research Letters, 316, 1–8.Google Scholar
- 38.Norouzi, P., Faridbod, F., Esfahani, N. E., et al. (2010). Cholesterol biosensor based on MWCNTs-MnO2 nanoparticles using FFT continuous cyclic voltammetry. International Journal of Electrochemical Science, 5, 1008–1017.Google Scholar
- 42.Ali, M. A., Srivastava, S., Solanki, P. R., et al. (2012). Nanostructured anatase-titanium dioxide based platform for application to microfluidics cholesterol biosensor. Applied Physics Letters, 101(084105), 1–5.Google Scholar