Aerosol Profile Extracted from Spacers as a Determinant of Actual Dose
- 55 Downloads
We propose a novel method to evaluate the efficacy of a pressurized metered dose inhaler (pMDI) in combination with a spacer, by not only considering the total dose extractable from the spacer but also the dependence of dose on the volume available for aerosol inhalation.
We studied volume-dependence of aerosol concentration during extraction from two commonly used plastic spacers (150 ml AerochamberPlus; 750 ml Volumatic) after a single puff of a 100 μg salbutamol pMDI (HFA-Ventolin), using laser photometric measurements.
After a delay of 1s in each spacer, the aerosol peak dose for AerochamberPlus was 2-fold that for Volumatic (p < 0.001), with the peak appearing well within the first 0.5 L even for the largest spacer. The opposite dose relationship is reached when considering total cumulative dose, which was 2-fold higher for Volumatic than for AerochamberPlus (p < 0.001); >95% of total cumulative dose was extracted well within 3 L for the largest spacer. The 2-fold cumulative dose relationship was confirmed by chemical assay on an absolute filter [AerochamberPlus: 21.4 ± 3.2 (SD) μg; Volumatic: 43.8 ± 9.1 (SD) μg].
Actual aerosol dose available to patients during inhalation via spacers can only be done on the basis of a quantification of aerosol peak dose and cumulative dose as a function of extracted volume.
Key words:aerosol dose metered dose inhalers spacers
Unable to display preview. Download preview PDF.
- 1.1. P. W. Barry and C. O’Callaghan. In vitro comparison of the amount of salbutamol available for inhalation from different formulations used with different spacer devices. Eur. Respir. J. 10:1345–1348 (1997).Google Scholar
- 2.2. W. D. Bennett, J. S. Brown, K. L. Zeman, S. C. Hu, G. Scheuch, and K. Sommerer. Targeting delivery of aerosols to different lung regions. J. Aerosol Med. 15:179–188 (2002).Google Scholar
- 3.3. J. B. West. Respiratory Physiology: The Essentials. Lippincott Williams & Wilkins, Philadelphia, 2000.Google Scholar
- 4.4. J. P. Mitchell, M. W. Nagel, and J. L. Rau. Performance of large-volume versus small volume holding chambers with chlorofluorocarbon-albuterol and hydrofluoroalkane-albuterol sulfate. Respir. Care 44:38–44 (1999).Google Scholar
- 5.5. J. H. Wildhaber, M. J. Hayden, N. D. Dore, S. G. Devadason, and P. N. LeSouef. Salbutamol delivery from a hydrofluoroalkane pressurized metered-dose inhaler in pediatric ventilator circuits: an in vitro study. Chest 113:186–191 (1998).Google Scholar
- 6.6. J. H. Wildhaber, S. G. Devadason, E. Eber, M. J. Hayden, M. L. Everard, Q. A. Summers, and P. N. LeSouef. Effect of electrostatic charge, flow, delay and multiple actuations on the in vitro delivery of salbutamol from different small volume spacers for infants. Thorax 51:985–988 (1996).Google Scholar
- 7.7. F. Pierart, J. H. Wildhaber, I. Vrancken, S. G. Devadason, and P. N. Le Souef. Washing plastic spacers in household detergent reduces electrostatic charge and greatly improves delivery. Eur. Respir. J. 13:673–678 (1999).Google Scholar
- 8.8. E. Berg. In vitro properties of pressurized metered dose inhalers with and without spacer devices. J. Aerosol Med. 8(Suppl 3):S3–10 (1995).Google Scholar
- 9.9. M. J. Clarke, J. Peart, S. Cagnani, and P. R. Byron. Adhesion of powders for inhalation: an evaluation of drug detachment from surfaces following deposition from aerosol streams. Pharm. Res. 19:322–329 (2002).Google Scholar
- 10.10. J. H. Wildhaber, S. G. Devadason, M. J. Hayden, R. James, A. P. Dufty, R. A. Fox, Q. A. Summers, and P. N. LeSouef. Electrostatic charge on a plastic spacer device influences the delivery of salbutamol. Eur. Respir. J. 9:1943–1946 (1996).Google Scholar
- 11.11. E. Dompeling, A. M. Oudesluys-Murphy, H. M. Janssens, W. Hop, J. G. Brinkman, R. N. Sukhai, and J. C. de Jongste. Randomised controlled study of clinical efficacy of spacer therapy in asthma with regard to electrostatic charge. Arch. Dis. Child. 4:178–182 (2001).Google Scholar
- 12.12. J. Heyder. Gravitational deposition of aerosol particles within a system of randomly oriented tubes. Aerosol Sci. 6:133–137 (1976).Google Scholar
- 13.13. H. Bisgaard, J. Anhoj, B. Klug, and E. Berg. A non-electrostatic spacer for aerosol delivery. Arch. Dis. Child. 73:226–230 (1995).Google Scholar
- 14.14. E. Berg, J. Madsen, and H. Bisgaard. In vitro performance of three combinations of spacers and pressurized metered dose inhalers for treatment in children. Eur. Respir. J. 12:472–476 (1998).Google Scholar
- 15.15. C. O’Callaghan, J. Lynch, M. Cant, and C. Robertson. Improvement in sodium cromoglycate delivery from a spacer device by use of an antistatic lining, immediate inhalation, and avoiding multiple actuations of drug. Thorax 48:603–606 (1993).Google Scholar