Nanoparticle Surface Area Monitor (NSAM, TSI model 3550 and Aerotrak 9000) is an instrument designed to measure airborne surface area concentrations that would deposit in the alveolar or tracheobronchial region of the lung. It was found that the instrument can only be reliably used for the size range of nanoparticles between 20 and 100 nm. The upper size range can be extended to 400 nm, where the minimum in the deposition curves occurs. While the fraction below 20 nm usually contributes only negligibly to the total surface area and is therefore not critical, a preseparator is needed to remove all particles above 400 nm in cases where the size distribution extends into the larger size range. Besides limitations in the particle size range, potential implications of extreme concentrations up to the coagulation limit, particle material (density and composition) and particle morphology are discussed. While concentration does not seem to pose any major constraints, the effect of different agglomerate shapes still has to be further investigated. Particle material has a noticeable impact neither on particle charging in NSAM nor on the deposition curves within the aforementioned size range, but particle hygroscopicity can cause the lung deposition curves to change significantly which currently cannot be mimicked with the instrument. Besides limitations, possible extensions are also discussed. It was found that the tendencies of the particle deposition curves of a reference worker for alveolar, tracheobronchial, total and nasal depositions share the same tendencies in the 20–400 nm size range and that their ratios are almost constant. This also seems to be the case for different individuals and under different breathing conditions. By means of appropriate calibration factors NSAM can be used to deliver the lung deposited surface area concentrations in all these regions, based on a single measurement.
Nanoparticles Surface area NSAM Occupational health Nanotechnology EHS
This is a preview of subscription content, log in to check access
The authors wish to thank Nick Stanley of University of Minnesota for providing SMPS data for DOS particles and Joakim Pagels of Lund University for giving us the opportunity of testing the NSAM during actual welding work.
Dockery DW, Pope CA, Xu X, Spengler JD, Ware JH, Fay ME et al (1993) An association between air pollution and mortality in six U.S. cities. N Engl J Med 329:1753–1759. doi:10.1056/NEJM199312093292401Google Scholar
Kreyling WG, Semmler M, Erbe F, Mayer P, Takenaka S, Schulz H et al (2002) Translocation of ultrafine insoluble iridium particles from lung epithelium to extrapulmonary organs is size dependent but very low. J Toxicol Environ Health 65:1513–1530. doi:10.1080/00984100290071649CrossRefGoogle Scholar
Löndahl J, Pagels J, Swietlicki E, Zhou J, Ketzel M, Massling A et al (2006) A set-up for field studies of respiratory tract deposition of fine and ultrafine particles in humans. J Aerosol Sci 37:1152–1163. doi:10.1016/j.jaerosci.2005.11.004CrossRefGoogle Scholar
Löndahl J, Massling A, Pagels J, Swietlicki E, Vaclavik E, Loft S (2007) Size-resolved respiratory-tract deposition of fine and ultrafine hydrophobic and hygroscopic aerosol particles during rest and exercise. Inhal Toxicol 19:109–116. doi:10.1080/08958370601051677Google Scholar
Oberdörster E (2004) Manufactured nanomaterials (Fullerenes, C60) induce oxidative stress in the brain of juvenile largemouth bass. Environ Health Perspect 112:1058–1062PubMedGoogle Scholar
Oberdörster G, Sharp Z, Atudorei V, Elder A, Gelein R, Kreyling W et al (2004) Translocation of inhaled ultrafine particles to the brain. Inhal Toxicol 16:437–445. doi:10.1080/08958370490439597Google Scholar