The Thermal Hysteresis of a Standard Platinum Resistance Thermometer
- 76 Downloads
The ETS-25-type standard thermometers manufactured by a single company are investigated. It is shown that instability of the thermometers during calibration may be due to the occurrence of stresses in the platinum wire of the sensitive element, and not due to oxidation of the platinum.
Keywordsplatinum resistance thermometer stability thermal hysteresis calibration of the platinum thermometer
- 2.M. Zhao, X. Li, and D. Chen, “Experimental study of different fi lling gases on the stability of metal-sheathed sprts in temperature: its measurement and control in science and industry,” AIP Conf. Proc., D. C. Ripple et al. (eds.), Melville, New York (2003), Vol. 7, pp. 339–344.Google Scholar
- 3.J. Ancsin, “Oxidation of platinum resistance thermometers in temperature: its measurement and control in science and industry,” ibid., pp. 345–349.Google Scholar
- 4.H. Sakurai and O. Tamura, “Oxidation characteristics of standard platinum resistance thermometers,” Jpn. J. Appl. Phys., 50, 036601 (2011), DO1: 10.1143/JJAP.50.036601.
- 5.B. Trumic, D. Stankovic, and A. Ivanovic, “The impact of cold deformation, annealing temperatures and chemical assays on the mechanical properties of platinum,” J. Min. Metall. Sect. B-Metall., 46, No. 1 B, 51–57 (2010).Google Scholar
- 7.GOST R 8.571–98, GSI. Standard Platinum Resistance Thermometers of the 1st and 2nd Class. Methods of Checking.Google Scholar