Patient-specific finite element analysis of popliteal stenting
- 260 Downloads
Nitinol self-expanding stents are used for the endovascular management of peripheral artery diseases of the popliteal artery, which is located behind the knee joint. Unfortunately, the complex kinematics of the artery during the leg flexion leads to severe loading conditions, favouring the mechanical failure of the stent, calling for a specific biomechanical analysis. For this reason, in the present study we reconstruct by medical image analysis the patient-specific popliteal kinematics during leg flexion, which is subsequently exploited to compute the mechanical response of a stent model, virtually implanted in the artery by structural finite element analysis (FEA). The medical image analysis indicates a non-uniform configuration change of the artery during the leg flexion, leading to an increase of the vessel curvature above the knee. The computed mechanical response of the stent reflects the non-uniform configuration change of the artery as after the flexion the average stress is higher in the part of the stent located above the knee. Although the proposed analysis is limited to a case-study, it shows the capability of patient-specific FEA simulations to compute the mechanical response of a stent model subjected to the complex and severe loading conditions of the popliteal artery during leg flexion.
KeywordsFinite element analysis (FEA) Biomechanics Popliteal stenting Patient-specific Nitinol
This work is partially funded by: ERC Starting Grant through the Project ISOBIO: Isogeometric Methods for Biomechanics (No. 259229); Ministero dell’Istruzione, dell’Universitá e della Ricerca through the Project No. 2010BFXRHS; iCardioCloud project by Cariplo Foundation (No. 2013-1779) and Lombardy Region (No. 42938382; No. 46554874). The authors acknowledge MD Mauro Ferrari (clinical support), Eng. Stefania Marconi (medical image elaboration) and the support of Regione Lombardia and CINECA Consortium through the grant LISA 2013 (high-performance computing).
Compliance with ethical standards
Conflicts of interest
The authors have no commercial, proprietary, or financial interest in any products or companies described in this paper.
- 6.Huang Y, Gloviczki P, Noel AA, Sullivan TM, Kalra M, Gullerud RE, Hoskin TL, Bower TC (2007) Early complications and long-term outcome after open surgical treatment of popliteal artery aneurysms: Is exclusion with saphenous vein bypass still the gold standard? J Vasc Surg 45(4):706–715.e1. doi: 10.1016/j.jvs.2006.12.011 CrossRefGoogle Scholar
- 7.Tielliu IF, Zeebregts CJ, Vourliotakis G, Bekkema F, van den Dungen JJ, Prins TR, Verhoeven EL (2010) Stent fractures in the hemobahn/viabahn stent graft after endovascular popliteal aneurysm repair. J Vasc Surg 51(6):1413–1418. doi: 10.1016/j.jvs.2009.12.071. http://www.sciencedirect.com/science/article/pii/S0741521410000364
- 9.Auricchio F, Conti M, Marconi S, Reali A, Tolenaar JL, Trimarchi S (2013) Patient-specific aortic endografting simulation: from diagnosis to prediction. Comput Biol Med 43(4):386–394. doi: 10.1016/j.compbiomed.2013.01.006. http://www.sciencedirect.com/science/article/pii/S0010482513000206
- 11.Conti M, Auricchio F, De Beule M, Verhegghe B (2009) Numerical simulation of Nitinol peripheral stents: from laser-cutting to deployment in a patient specific anatomy. In: Proceeding of ESOMAT 2009:06008. doi: 10.1051/esomat
- 13.Rebelo N, Walker N, Foadian H (2001) Simulation of implantable stents. Abaqus User’s Conference 143:421–434Google Scholar
- 19.Allie D, Hebert C, Walker C (2004) Nitinol stent fractures in the SFA. Endovasc Today 7:22–34Google Scholar
- 20.Scheinert D, Scheinert S, Sax J, Piorkowski C, Brunlich S, Ulrich M, Biamino G, Schmidt A (2005) Prevalence and clinical impact of stent fractures after femoropopliteal stenting. J Am Coll Cardiol 45(2):312–315. doi: 10.1016/j.jacc.2004.11.026. http://www.sciencedirect.com/science/article/pii/S0735109704022570
- 21.Kapnisis KK, Halwani DO, BrottPeter BC, Anderson G, Lemons JE, Anayiotos AS (2012) Stent overlapping and geometric curvature influence the structural integrity and surface characteristics of coronary Nitinol stents. J Mech Behav Biomed Mater 20:227–236. doi: 10.1016/j.jmbbm.2012.11.006 CrossRefGoogle Scholar
- 23.Wensing P, Scholten F, Buijs P, Hartkamp M, Mali W, Hillen B (1995) Arterial tortuosity in the femoropopliteal region during knee flexion: a magnetic resonance angiographic study. J Anat 187(1):133–139Google Scholar
- 25.Smouse H, Nikanorov A, LaFlash D (2005) Biomechanical forces in the femoropopliteal arterial segment what happens during extremity movement and what is the effect on stenting? Endovasc Today 4:60–66Google Scholar
- 26.Cheng CP, Wilson NM, Hallett RL, Herfkens RJ, Taylor CA (2006) In vivo MR angiographic quantification of axial and twisting deformations of the superficial femoral artery resulting from maximum hip and knee flexion. J Vasc Interv Radiol 17(6):979–987. doi: 10.1097/01.RVI.0000220367.62137.E8 CrossRefGoogle Scholar
- 28.Jonker F, Schlosser F, Moll F, Muhs B (2008) Dynamic forces in the SFA and popliteal artery during knee flexion. Endovasc Today 53:53–58Google Scholar
- 33.Ghriallais R, Bruzzi M (2013) Effects of knee flexion on the femoropopliteal artery: a computational study. Med Eng Phys 35(11):1620–1628. doi: 10.1016/j.medengphy.2013.05.015. http://www.sciencedirect.com/science/article/pii/S135045331300132X
- 38.Meoli A, Dordoni E, Petrini L, Migliavacca F, Dubini G, Pennati G (2013) Computational modelling of in vitro set-ups for peripheral self-expanding Nitinol stents: the importance of stent–wall interaction in the assessment of the fatigue resistance. Cardiovasc Eng Technol 4(4):474–484CrossRefGoogle Scholar
- 42.Arena FJ (2005) Arterial kink and damage in normal segments of the superficial femoral and popliteal arteries abutting Nitinol stents a common. Vasc Dis Manag 2(5):160–164Google Scholar