Design and performance of a miniature TDCR counting system
- 113 Downloads
A portable liquid scintillation counting system based on the triple-to-double coincidence ratio (TDCR) method was developed at Sofia University “St. Kliment Ohridski”. The system consists of a miniature TDCR counter with cylindrical optical chamber and a specialized TDCR counting module named nanoTDCR. The nanoTDCR module is produced by the labZY company and provides several important new TDCR counting functionalities like: individual extending-type dead-time in each channel; simultaneous counting with two different extendable dead-times and two different coincidence windows and simultaneous TDCR counting and spectrum acquisition. The performance of the new system was tested in benchmark comparisons with the LNHB’s primary TDCR counting system of activity measurements of 241Am, 3H, 14C and 63Ni. Good agreement between the two systems was observed.
KeywordsTDCR counter Miniature TDCR system NanoTDCR Extending dead time Absolute activity measurement
This work is supported by the Bulgarian National Science Fund under Contract DFNI T02/13”POLYRAD”. The authors are indebted to Mr. Nikolay Markov and Mr. Todor Todorov for the mechanical work on the TDCR counter, which was manufactured and assembled in Bulgaria.
- 1.Malonda G, Carles G, L’Annunziata M (2012) Handbook of radioactivity analysis, vol 3. Academic Press, AmsterdamGoogle Scholar
- 3.Cassette P, Capogni M, Johansson L, Kossert K, Nähle O, Sephton J, De Felice P (2013) Development of portable liquid scintillation counters for on-site primary measurement of radionuclides using the triple-to-double coincidence ratio method. In: Proceedings of the 3rd International Conference on Advancements in Nuclear Instrumentation, Measurement Methods and their Applications (ANIMMA), Marseille, 2013, doi 10.1109/ANIMMA.2013.6727876
- 6.Triple-to-double coincidence ratio liquid scintillation counting system (TDCR) and MCA, http://www.labzy.com/index.html#nanoTDCR. Accessed 24 May 2017
- 8.Detection efficiency calculation for pure-beta radionuclides, LNHB program with short tutorial, 2012 version, TDCR07c.zip: http://www.nucleide.org/ICRM_LSCWG/icrmsoftware.htm. Accessed 24 May 2017