Rapid on-site radionuclide screening of aqueous waste streams using dip-stick technologies and liquid scintillation counting
This paper describes an early-stage evaluation of a purpose-designed extraction/detection system that can be deployed by non-specialists either on-site or as part of a mobile laboratory. The system comprises three main components; (1) an optimised compact extraction system for recovery of radionuclides from the waste form; (2) an extraction test strip designed to recover the radionuclides from the waste digest; (3) a scintillation-based detection system capable of quantification of alpha, low energy beta and high energy beta emitting radionuclides. Data are presented on the preliminary assessment of the extraction/detection system for the measurement of 90Sr and 99Tc in aqueous wastes.
Keywords90Sr 99Tc Aqueous waste streams Rapid screening Dip-stick technology
Robust characterization of nuclear wastes is critical in underpinning nuclear decommissioning programmes and its importance has been recently highlighted by Emptage et al., . Traditionally nuclear waste characterisation has relied on specialist laboratory radioanalytical support which, although providing high quality data, can result in delays to the decommissioning programme. Although such support is essential, many areas of waste characterization can be accelerated through careful integration of detailed laboratory characterization with quicker, less sensitive on-site screening capabilities. Screening can provide valuable information on the presence/absence of contamination, stability of the waste stream composition and radionuclide vectors and early indication of significant deviations from expected operational parameters. For example, the procedures have been developed for rapid screening of potable waters in the event of an intentional assault on the drinking water supply system . On-site screening has been widely adopted for easy-to-measure (ETM; categorization ) gamma-emitting radionuclides. Development of on-site techniques for difficult-to-measure (DTM) radionuclides is desirable and techniques such as autoradiography have been applied to identify contaminated objects and surfaces (e.g., [4, 5]). However, extension of this concept to nuclide-specific measurement of radionuclides such as 90Sr and 99Tc has been limited, mainly as a result of the technical difficulties associated with the robust detection of these radionuclides outside of the laboratory environment.
Dip-stick technology applied to rapid screening of radionuclides is a concept currently being developed by Raddec International Ltd and is expected to lead to practical benefits. The basic principle is well established in other fields such as screening measurements of pH and chemical and biochemical species. The current approach incorporates an analyte-specific extractant that will interact with the target analyte (usually by adsorbing the analyte to a degree proportional to its concentration in the solution). The dip-stick can then be recovered for subsequent measurement. The dip-stick deployment and recovery can be readily achieved by non-specialists with minimal infrastructure or equipment. The dip-sticks can be designed to target specific analytes through selection of appropriate commercially available materials or purpose-designed extractants such as calixarenes/crown ethers, ion-imprinted polymers or metal organic frameworks. The approach is also ideal for the deployment of nano-particle extractants as the extractant is easily recoverable following deployment. The geometry of the dip-stick allows measurement using conventional liquid scintillation counting or using plate readers. Incorporation of scintillators, either integrated with the extractant or by using scintillating support materials, would permit further simplification of the detection systems. Dip-sticks can incorporate a single extractant to target one analyte (single extractant sticks—SES). Alternatively, multiple extractants can be incorporated in discrete zones (multiple extractant stick—MES), with measurement achieved using a spatially-resolved detector system to permit multi-radionuclide detection. The dip-stick technology does rely on the sample being in solution and some form of digestion is required for solid samples, although this can be simplified for screening purposes. It is therefore envisaged that the dip-stick technology could be applied to rapid screening of both aqueous and solid waste streams.
The aim of the current study was to evaluate the feasibility of using dip-stick technologies for rapid radionuclide screening in support of nuclear waste characterization. Two extractant systems were evaluated; (1) Sr-resin SES for 90Sr determination in acid leachates and (2) TEVA-resin SES for 99Tc determination in neutral solutions. Key operational parameters were assessed including the mass of resin loaded onto the dip-stick, the rate of analyte uptake and the correlation between activity on the dip-stick and activity concentration of the analyte in solution. The sensitivity of the dip-stick systems was compared with the operational limits associated with nuclear waste characterization.
To determine the mass of resin loaded onto the dip-stick (m s), ~0.5 g of TEVA resin was equilibrated with ~5000 Bq of 99Tc. The mixture was filtered and the Tc-loaded TEVA resin was dried and reweighed. Aliquots of the Tc-loaded TEVA resin were accurately weighed into a 22 mL polythene vial and 20 mL Gold Star scintillation cocktail was added. The 99Tc activity per gram of TEVA resin was then calculated. The Tc-loaded TEVA resin was then used to prepare a series of dip-sticks. Once prepared, the 99Tc activity on the dip-stick was measured using liquid scintillation counting and the mass of resin present on the dip-stick was calculated from the measured activity.
The rate of radionuclide uptake was assessed for Sr-SES and TEVA-SES dip-sticks. Five dip-sticks were prepared for each extractant using 100–150 µm resin. The effect of particle size on rate of uptake was also assessed for Sr-resin using dip-sticks prepared using 50–100 µm resin. Test solutions were prepared by spiking the target solution with known activities of 90Sr or 99Tc (NPL, UK). The dip-sticks were exposed to a test solution of either 8 M HNO3 (90Sr), or water at pH 7 (99Tc) containing a known activity concentration of the target radionuclide. The solutions were unstirred during exposure. The dip-stick was placed into the solution for a specified period of time, recovered and any adhering solution removed by quickly rinsing the test stick in a matrix-matched solution and drying with paper towel. The activity on the stick was measured by liquid scintillation counting. The rate of change in effective distribution coefficient (dk/dt) was then assessed.
The ratio of activity associated with the dip-stick to the initial activity concentration in solution (b t) was determined for Sr-SES and TEVA-SES dip-sticks. Dip-sticks were exposed to the test solutions containing the target radionuclide at a range of activity concentrations for a defined time. The activity adsorbed on the dip-stick was then measured and the correlation between the activity on the dip-stick and the activity concentration in solution was assessed. The passive adsorption of the radionuclide on an inert medium was also assessed by preparing a dip-stick using prefilter material (the support material used in the extractive resins) and exposing the ‘blank’ dip-stick to 90Sr solution. The limit of detection for each dip-stick was also calculated and compared with operational limits associated with nuclear waste characterization.
The activities of the radionuclides on the dip-stick and in the test solution were measured using Wallac 1220 Quantulus liquid scintillation counters. The dip-stick was transferred to a 22 mL polythene vial and 20 mL Gold Star scintillation cocktail (Meridian, UK) was added. Aliquots of test solution (typically 1 mL) were also sampled and mixed with 19 mL scintillation cocktail. Samples were typically counted along with a background sample for 3600 s.
Results and discussion
Mass of extractive-resin on dip-stick
Rate of uptake on Sr, and TEVA-SES sticks
Uptake coefficients for rate of uptake
Particle size (μm)
k eq (ml g−1)
Rate coefficient, r (s−1)
3.9 (8) × 10−4
2.0 (5) × 10−4
1.5 (5) × 10−4
Response of dip-stick to varying aqueous activity concentration
The sensitivity of the dip-stick system will depend on the deployment time and count time. It is envisaged that the deployment time will be fixed and specified for a given target sensitivity. For a 240 min deployment and 60 min count time, the limit of detection is <0.1 Bq/mL of solution. For solid samples, this would equate to 1 Bq/g assuming that 0.5 g of solid sample is leached with 5 mL of reagent. This detection limit compares favourably with the out-of-scope limits of 1 Bq/g for 90Sr and 99Tc . Further developments will be aimed at reducing the exposure time to permit more rapid deployment. The sensitivity of the system could be further improved by increasing the active surface area of the dip-stick or by increasing counting times. It is also noted that consideration must be given to the effect of interfering radionuclides on test stick response. For example, the Sr-SES stick would be expected to respond to 210Pb in solution . Likewise, tetra-valent actinides would co-extract with TcO4 − on the TEVA-SES resin . The use of multiple extractant zones would assist in deconvolving the signal from multiple radionuclides and barrier films could be used to prevent interfering radionuclides from interacting with the extractant. In addition, the observed response may be impacted by the presence of certain matrix elements in solution which could compete for adsorption sites or suppress uptake of the target analyte. These will be evaluated in future research.
The concept of the dip-tick for the assessment of radionuclide concentrations in aqueous solution has been successfully demonstrated using Sr-resin and TEVA-resin based dip-sticks. The quantity of radionuclide (either 90Sr or 99Tc) adsorbed onto the active area of the stick is proportional to the initial activity concentration present in solution. Further research is now required to assess the impact of competing ions on the performance of the test stick and to evaluate the effects of geometry, surface area and uptake kinetics and to optimize the operation of the test stick system accounting for these factors. The applicability of the test stick approach will be further extended through the development and evaluation of other extractant systems and to the application of multiple-extractant sticks measured using plate reader technologies. Finally, the adoption of these systems will require further development of simplified sample preparation procedures and the testing of the systems using operationally-exposed samples. However, it is envisaged that the successful development of such a testing approach could offer significant advantages for waste sentencing and facilitate more rapid screening at the point of waste production.
The research has been undertaken as a preliminary study in support of a Nuclear Decommissioning Authority, UK PhD Bursary allocated to the authors. We also acknowledge the collaboration of Raddec International Ltd as originators of the Dip Stick concept for radionuclide analysis.
- 1.Emptage M, Loudon D, McLeod R, Milburn H (2016) Characterisation: challenges and opportunities—a UK perspective. In: Proceedings of the PREDIC 2016 conference, 16–18th February 2016, Lyon, FranceGoogle Scholar
- 3.IAEA (2009) Determination and use of scaling factors for waste characterization in nuclear power plants. IAEA Nuclear Energy Series Report NW-T-1.18. IAEA, Vienna, AustriaGoogle Scholar
- 5.Haudebourg R, Fichet P (2016) A non-destructive and on-site digital autoradiography-based tool to identify contaminating radionuclide in nuclear wastes and facilities to be dismantled. J Radioannal Nucl Chem 309:551–561Google Scholar
- 7.Horwitz P, Dietz M, Chiarizia R, Diamond H, Maxwell SL, Nelson MR (1995) Separation and preconcentration of actinides by extraction chromatography using a supported liquid anion exchanger: application to the characterization of high-level nuclear waste solutions. Anal Chim Acta 310:63–78CrossRefGoogle Scholar
- 8.IAEA (2004) Application of the concepts of exclusion, exemption and clearance. Safety guide RS-G-1.7. IAEA, ViennaGoogle Scholar
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.