Rapid method for plutonium-241 determination in soil samples
- 548 Downloads
A simple and rapid procedure for the determination of plutonium isotopes in the environment is presented. The procedure combines alpha spectrometry, solvent extraction and liquid scintillation measurements to ensure that both alpha- and beta-emitting isotopes are determined. Of five tested extractants, bis-(2-ethylhexyl) phosphoric acid was found to be the best choice. The procedure was applied to soil samples contaminated with Chernobyl fallout.
KeywordsPlutonium 241Pu Liquid extraction Liquid scintillation spectrometry Alpha spectrometry Soil samples
Alpha spectrometry is the most widely used method for the determination of plutonium isotopes in environmental samples. However, this method does not permit the determination of beta-emitting 241Pu, a radionuclide abundant in the environment. This isotope decays to 241Am, another radiotoxic hazard . The best way to determine 241Pu in an environmental sample is by liquid scintillation (LS) after the isotope has been removed from the sample by a radiochemical procedure. The following procedure can be used for the simultaneous determination of alpha and beta plutonium activity [2, 3, 4]. An important consideration is choosing an extracting agent that is both efficient and minimizes quenching effects during LS measurements.
The aim of this study is to illustrate a rapid method for the determination of 241Pu in soil samples using a simple solvent extraction method followed by LS measurements. The procedure described here is applied after the determination of other Pu isotopes in the sample by a technique of alpha spectrometry.
There are many commonly known extracting agents for the extraction of heavy metal traces, such as: trioctylphosphine oxide, TOPO [5, 6, 7], bis-(2-ethylhexyl) phosphoric acid, HDEHP [8, 9], methyltrioctylammonium chloride (Aliquat 336) [10, 11], N,N-diethyldodecanamide, DEDA  or thenoyltrifluoracetone, HTTA . These compounds have been tested to find those of high plutonium recovery from the environmental samples. This test should precede the final LS measurements.
The ultra low-level spectrometer Quantulus 1220 (Wallac, Perkin-Elmer) was used for LS measurements. Permablend III (Packard), consisted of 91 % PPO and 9 % bis-MSB dissolved in toluene was applied as a scintillation cocktail. Samples were measured for 5 h, after 24 h stabilization in darkness in the apparatus to avoid chemiluminescence. Standard solutions of 242Pu (~0.02 Bq) in 3 M nitric acid were used as tracers.
Alpha spectrometric measurements were performed using the alpha spectrometer (model 7401 Canberra) with PIPS detector of 17 keV FWHM resolution, the 1520 mixer-router, S-100 multichannel analyzer, and Genie™ 2000 software for quantitative analysis (Canberra).
Before the LS measurements, the experiments were performed to choose a proper extractant which is characterized by the best efficiency and low quenching effects. The following extractant solutions were tested: 0.2 M HDEHP dissolved in toluene, 0.2 M TOPO in cyclohexane, 0.05 M Aliquat 336 in xylene, 0.3 M DEDA in toluene and 0.01 M HTTA in toluene. The extraction procedure was described previously [14, 15].
For the verification of the procedure, soil samples contaminated with Chernobyl fallout were analyzed. The samples were collected in 1991 near Bragin village from two successive layers of soil (0–5 and 5–10 cm). The samples were ashed, co-precipitated, separated by ion exchange and then electrodeposited on small steel plates, as described recently . After the alpha-spectrometry measurements, the electro-deposited plutonium was washed off of the steel plates with nitric acid and extracted into the organic phase of 0.2 M HDEHP solution in toluene. An aliquot of the organic phase was mixed with the scintillation cocktail and its activity was measured with a Quantulus spectrometer.
Results and discussion
The characteristics of a useful extractant include the following: acceptable counting efficiency, resistance to decomposition by nitric acid, miscibility with the scintillation cocktail, low background counts, and low quenching characteristics. HDEHP had the best combination of these characteristics.
Figure 1 illustrates that the surface layer of soil (0–5 cm deep) displayed the following activities: 238Pu of 2.88 Bq/kg, 6.98 Bq/kg of 239+240Pu and 60.8 Bq/kg of 241Pu. Figure 2 illustrates that the deeper layer of soil (5–10 cm deep) displayed the following activities: 0.066 Bq/kg of 238Pu, 0.50 Bq/kg of 239+240Pu and 0.53 Bq/kg of 241Pu. The estimated uncertainties of these results are about 10 % for both soil depths.
Isotopic ratios of plutonium in analyzed Chernobyl contaminated samples (the last column ratio has been recalculated back on the date of the Chernobyl accident)
241Pu/239,240Pu in 1986
Table 1 shows the calculated isotopic ratios of the two soil layers. According to the literature data, the typical global fallout ratio 238Pu/239+240Pu is ranged 0.03–0.05, and for Chernobyl fallout is 0.3–0.65 . The observed ratio indicates that the upper soil layer is contaminated mainly with Chernobyl fallout and the lower rather with a global one.
The data from the literature is as follows, the ratio 241Pu/239+240Pu for global fallout is about 4.2 (based on the year 1986), and for the Chernobyl fallout it is about 94.8 and varies widely . As seen in Table 1, the calculated average ratio 241Pu/239+240Pu in the surface layer is 9.5 and after calculation on the date of the Chernobyl disaster it will be equal to 32. In the deeper layer the 241Pu/239+240Pu ratio was 1.1. It was therefore about 3.6 in 1986. Comparing these results with the literature data, one can conclude that in the deeper soil layer plutonium from global fallout predominates and in the surface layer the contribution of plutonium from Chernobyl fallout is undoubtedly the largest.
The presented procedure which combines the alpha spectrometry and the LS measurements after solvent extraction may be useful for the rapid determination of all plutonium isotopes present in the environment (238Pu, 239+240Pu and 241Pu).
It was found that among five various extractants, the HDEHP reveals the best properties to be applied as an extracting agent during the sample preparation to LS measurement.
The elaborated method used for the determination of plutonium in the samples contaminated with Chernobyl fallout shows its usefulness. Calculated isotopic ratios confirmed a large share of this source of contamination in the analyzed samples, especially in those taken from the upper layer of soil.
Study was performed with financial support of National Science Centre (Poland) by grant no. NN 204 110738.
- 1.Brown E, Firestone RB (1986) Table of radioactive isotopes. Wiley, New YorkGoogle Scholar
- 14.Komosa A, Piekarz M (2010) Nukleonika 55(2):137–141Google Scholar
- 15.Komosa A, Piekarz M (2011) In: Cassette P (ed) Radiocarbon. Arizona, TucsonGoogle Scholar
- 16.Orzel J, Komosa A (2013) J Radioanal Nucl Chem 299(1):643–649Google Scholar
- 17.Komosa A (2003) Physicochemical problems of determination and behavior of plutonium isotopes in environment, especially beta radiating 241Pu. UMCS, LublinGoogle Scholar
- 18.Mietelski JW (2003) Nuclear spectrometry in radionuclide bioavailability study of fuel like hot particles in forest environment. Nuclear physics institute report 1921/B, www.ifj.edu.pl/reports/2003.html
- 19.Kierepko R, Mietelski JW (2010) Nukleonika 55(2):201–204Google Scholar
- 20.Paatero J, Jaakkola T, Reponen A (1994) Radiochim Acta 64:139–144Google Scholar
Open AccessThis article is distributed under the terms of the Creative Commons Attribution License which permits any use, distribution, and reproduction in any medium, provided the original author(s) and the source are credited.