Development of A Textile Capacitive Proximity Sensor and Gait Monitoring System for Smart Healthcare
- 221 Downloads
Gait is not only one of the most important functions and activities in daily life but is also a parameter to monitor one‘s health status. We propose a single channel capacitive proximity pressure sensor (TCPS) and gait monitoring system for smart healthcare. Insole-type TCPS (270 mm in length) was designed consisting of three layers including two shield layers and a sensor layer. Analyzing the step count and stride time are the basic indicators in gait analysis, thus they were selected as evaluation indicators. A total of 12 subjects participated in the experiment to evaluate the resolution of our TCPS. To evaluate the accuracy of TCPS, step count and its error rates were simultaneously detected by naked eye, ZIKTO Walk (ZIKTO Co., Korea), and HJ-203-K pedometer (Omron Co., Japan) as reference. Results showed that the error rate of 1.77% in TCPS was lower than those of other devices and correlation coefficient was 0.958 (p-value = 0.000). ZIKTO Walk and pedometer do not provide information on stride time, therefore it was detected by F-scan (Tekscan, USA) to evaluate the performance of TCPS. As a result, error rate of stride time measured by TCPS was found to be 1% and the correlation coefficient was 0.685 (p-value = 0.000). According to these results, our proposed system may be helpful in development of gait monitoring and measurement system as smart healthcare.
KeywordsGait Situation Smart Healthcare Capacitive Proximity Sensor Conductive Textile
Textile Capacitive Proximity Sensor
This research was supported by the Soonchunhyang University Research Fund.
Se Dong designed the framework of research and performed to anlyze gait data. And, Changwon assisted experiment for measuring gait data. Doo-Soon developed the gait monitoring application and Jong Hyuk contributed to verifying of proposed research method.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
All procedures performed in this studyinvolving human participants were in accordance with the ethical standards of the Soonchunhyang University of Korea, and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Informed consent was obtained from all individual participants included in the study.
- 1.Committee on Quality of Health Care in America and Institute of Medicine, Crossing the Quality Chasm: A New Health System for the 21st Century. National Academy Press, March, 2001.Google Scholar
- 2.Warren, S., Craft, R. L., Designing smart health care technology into the home of the future. Proceedings of The first Joint BME/EMBS Conf. Serving Humanity, Advancing Technology, Oct, 1999.Google Scholar
- 3.Hartrison, R., Clayton, W., and Wallace, P., Can telemedicine be used to improve communication between primary secondary care? BMJ 313:30, 1996.Google Scholar
- 5.Paradiso, R., Loriga, G., and Taccini, N., wearable system for vital signs monitoring. Stud. Health Technol. Inform. 108:253–259, 2004.Google Scholar
- 6.May, C., Mort, M., Mair, F., Ellis, N. T., and Gask, L., Evaluation of new technologies in healthcare systems: what’s the context?. Health Informatics, 2000.Google Scholar
- 7.Oliver, N., Flores-Mangas, F., Health Gear: A Real-time Wearable system for monitoring and analyzing physiological signals. Technical Report, Microsoft Research, Microsoft Corp., 182, 2005.Google Scholar
- 8.Liutkus, A., Scale-space peak picking. Research Report Inria Nancy – Grand Est (Villers-les-Nancy, France), https://hal.inria.fr/hal-01103123v2, 2015.
- 9.Muro-de-la-Herran, A., Garcia-Zapirain, B., and Mendez-Zorrilla, A., Gait analysis methods: An overview of wearable and non-wearable systems. highlighting clinical applications. Sensors. https://doi.org/10.3390/s140203362, 2014.
- 10.Plotnik, M., Giladi, N., and Hausdorff, J. M., A new measure for quantifying the bilateral coordination of human gait: effects of aging and Parkinson’s disease. Exp. Brain Res. 181:561–570, 2007.Google Scholar
- 11.Plotnik, M., Giladi, N., and Hausdorff, J. M., Bilateral coordination of walking and freezing of gait in Parkinson’s disease. Eur. J. Neurosci. 27(8):1999–2006, 2008. https://doi.org/10.1111/j.1460-9568.2008.06167.x.
- 12.Takeda, R., Tadano, S., Todoh, M., Morikawa, M., Nakayasu, M., and Yoshinari, S., Gait analysis using gravitational acceleration measured by wearable sensors. J. Biomech. 42(3):223–233, 2009.Google Scholar
- 13.Hook, W. R., Gait analysis using angular rate reversal. United States Patent Application Publication, 0131555 A1, 2013.Google Scholar
- 14.Liu, T., Inoue, Y., and Shibata, K., Development of a wearable system for quantitative gait analysis. Measurement 42:978–988, 2009.Google Scholar
- 15.Sejdić, E., Lowry, K. A., Bellanca, J., Perera, S., Redfern, M. S., and Brach, J. S., Extraction of stride events from gait accelerometry during treadmill walking. IEEE J.Translat. Eng. Health Med. 4, 2016. https://doi.org/10.1109/JTEHM.2015.2504961.
- 16.Lin, F., Wang, A., Zhuang, Y., Tomita, M. R., and Xu, W., Smart insole: A wearable sensor device for unobtrusive gait monitoring in daily life. IEEE Trans. Ind. Inf. 12(6):2281–2291, 2016.Google Scholar
- 17.Cantoral-Ceballos, J. A., Nurgiyatna, N., Wright, P., Vaughan, J., Brown-Wilson, C., Scully, P. J., and Ozanyan, K. B., Intelligent carpet system, based on photonic guided-path tomography, for gait and balance monitoring in home environments. IEEE Sens. J. 15(1):279–289, 2015.Google Scholar
- 18.Park, S. K., and Kim, W. K., Electronic and smart textiles. Polym. Sci. Technol. 24(1):38–44, 2013.Google Scholar
- 19.Marozas, V., Petrenas, A., Daukantas, S., and Lukosevicius, A., A comparison of conductive textile-based and silver/silver chloride gel electrodes in exercise electrocardiogram recordings. J. Electrocardiol. 44(2):189–194, 2011.Google Scholar
- 20.Akita, J., Shinmura, T., Sakurazawa, S., Yanagihara, K., Kunita, M., Toda, M., and Iwata, K., Wearable electromyography measurement system using cable-free network system on conductive fabric. Artif. Intell. Med. 42(2):99–108, 2008.Google Scholar
- 21.Lofhede, J., Seoane, F. and Thordstein, M., Soft textile electrodes for EEG monitoring, Information Technology and Applications in Biomedicine (ITAB), 2010 10th IEEE International Conference on. Nov. 2010.Google Scholar
- 22.Meyer, J., Arnrich, B., Schumm, J., and Troster, G., Design and modeling of a textile pressure sensor for sitting posture classification. IEEE Sens. J. 10(8):1391–1398, 2010.Google Scholar
- 23.Giovanelli, D., and Farella, E., Force sensing resistor and evaluation of technology for wearable body pressure sensing. J. Sens. 2016:1–13, 2016.Google Scholar
- 24.Chen, S., Lach, J., Lo, B., and Yang, G.-Z., Toward pervasive gait analysis with wearable sensors: A systematic review. IEEE J. biomed. Health Inform. 20(6):1521–1537, 2016.Google Scholar
- 25.Titaqnova, E. B., Mateev, P. S., and Tarkka, I. M., Footprint analysis of gait using a pressure sensor system. J. Electromyogr. Kinesiol. 14:275–281, 2004.Google Scholar
- 26.Abinaya, B., Latha, V., and Suchetha, M., An advanced gait monitoring system based on air pressure sensor embedded in a shoe. Process. Eng. 38:1634–1643, 2012.Google Scholar
- 27.Wilson, R. R., Oudekerk, D. R., Wilson, D. P., Fogel, K. M., Rebecca Neth Townsend, United States Patent, 8, 628,485 B2, 2014.Google Scholar
- 28.Wang, L., Tan, T., Ning, H., and Hu, W., Silhouette Analysis-based gait recognition for human identification. IEEE Trans. Pattern Anal. Machine Intel. 25(12):1505–1518, 2003.Google Scholar
- 29.Fang, P., Yang, Y., Fan, X., Li, S., Li, Y., Li, D., and Fan, Y., Optimal estimation of total plantar force for monitoring gait in daily life activities with low-price insole system. J. Mech. Med. Biol. 14(3), 2014. https://doi.org/10.1142/S0219519414500377.
- 30.Gerlach, C., Krumm, D., Illing, M., Lange, J., Kanoun, O., Odenwald, S., and Hubler, A., Printed MWCNT-PDMS-composite pressure sensor system for plantar pressure monitoring in ulcer prevention. IEEE Sens. J. 15(7):3647–3656, 2015.Google Scholar
- 31.Lai, D. T. H., Levinger, P., Begg, R. K., Gilleard, W. L., and Palaniswami, M., Automatic recognition of gait patterns exhibiting patellofemoral pain syndrome using a support vector machine approach. IEEE Trans. Inf. Technol. Biomed. 13(5):810–817, 2009.Google Scholar
- 32.Costilla-Reyes, O., Scully, P., and Ozanyan, K. B., Temporal Pattern Recognition in gait activities recorded with a footprint imaging sensor system. IEEE Sens. J. 16(24):8815–8822, 2016.Google Scholar
- 33.Huang, T.-C., Hsu, C., and Ciou, Z.-J., Systematic methodology for excavating sleeping beauty publications and their princes from medical and biological engineering studies. J. Med. Biol. Eng. 35(6):749–758, 2015.Google Scholar
- 34.Chen, L.-S., and Cai, S.-J., Neural-Network-Based Resampling method for detecting diabetes mellitus. J. Med. Biol. Eng. 35(6):824–832, 2015.Google Scholar
- 35.Andreu-Perez, J., Poon, C. C. Y., Merrifield, R. D., Wong, S. T. C., and Yang, G.-Z., Big data for health. IEEE J. Biomed. Health Inf. 19(4):1193–1208, 2015.Google Scholar
- 36.Hao, F., and Park, D.-S., Sang Yeon Woo, Se Dong Min, and Sewon Park, Treatment planning in smart medical: a sustainable strategy. J. Inf. Process. Syst. 12(4):711–723, 2016.Google Scholar
- 37.Sato, A., Huang, R., and Yen, N. Y., Design of fusion technique-based mining engine for smart business. human-centric Comput. Inform. Sci. 5(23), 2015. https://doi.org/10.1186/s13673-015-0036-z.
- 38.Lin, C.-C., Chen, C.-C., Lin, P.-S., Lee, R.-G., Huang, J.-S., Tsai, T.-H., and Chang, Y.-C., Development of home-based frailty detection device using wireless sensor networks. J. Med. Biol. Eng. 36(2):168–177, 2016.Google Scholar
- 39.Tawalbeh, L. A., Mehmood, R., Benkhlifa, E., and Song, H., Mobile cloud computing model and big data analysis for healthcare applications. IEEE Access, 4, https://doi.org/10.1109/ACCESS.2016.2613278, 2016.