The T-lock: automated compensation of radio-frequency induced sample heating
- 112 Downloads
Modern high-field NMR spectrometers can stabilize the nominal sample temperature at a precision of less than 0.1 K. However, the actual sample temperature may differ from the nominal value by several degrees because the sample heating caused by high-power radio frequency pulses is not readily detected by the temperature sensors. Without correction, transfer of chemical shifts between different experiments causes problems in the data analysis. In principle, the temperature differences can be corrected by manual procedures but this is cumbersome and not fully reliable. Here, we introduce the concept of a “T-lock”, which automatically maintains the sample at the same reference temperature over the course of different NMR experiments. The T-lock works by continuously measuring the resonance frequency of a suitable spin and simultaneously adjusting the temperature control, thus locking the sample temperature at the reference value. For three different nuclei, 13C, 17O and 31P in the compounds alanine, water, and phosphate, respectively, the T-lock accuracy was found to be <0.1 K. The use of dummy scan periods with variable lengths allows a reliable establishment of the thermal equilibrium before the acquisition of an experiment starts.
KeywordsTemperature control Calibration Chemical shifts NMR
We thank Professor Philip H. Bolton for providing 17O-labeled water and Gregory Heffron and Charles Sheahan for help with the NMR measurements. This research was supported in part by NIH (grants GM47467, GM075879 and EB00206) and the Swiss National Science Foundation.
- Ammann C, Meier P, Merbach AE (1982) A simple multinuclear NMR thermometer. J Magn Reson 46:319–321Google Scholar
- Bax A, Clore GM, Gronenborn AM (1990) 1H–1H correlation via isotropic mixing of 13C magnetization, a new 3-dimensional approach for assigning 1H and 13C spectra of 13C-enriched proteins. J Magn Reson 88:425–431Google Scholar
- Bornais J, Brownstein S (1978) A low-temperature thermometer for 1H, 19F, and 13C. J Magn Reson 29:207–211Google Scholar
- Braunschweiler L, Ernst RR (1983) Coherence transfer by isotropic mixing—application to proton correlation spectroscopy. J Magn Reson 53:521–528Google Scholar
- Cavanagh J, Fairbrother WJ, Palmer AGIII, Skelton NJ (1996) Protein NMR Spectroscopy: principles and practice. Academic Press, San DiegoGoogle Scholar
- Farrar TC, Sidky E, Decatur JD (1990) An inexpensive precise gas flow temperature controller. J Magn Reson 86:605–612Google Scholar
- Ikura M, Kay LE, Tschudin R, Bax A (1990) Three-dimensional NOESY-HMQC spectroscopy of a 13C-labeled protein. J Magn Reson 86:204–209Google Scholar
- Keiichiro H (1991) Patent application 3-156394, JapanGoogle Scholar
- Levy GC, Bailey FT, Wright DA (1980) A sensitive NMR thermometer for multinuclei FT NMR. J Magn Reson 37:353–356Google Scholar
- Shaka AJ, Barker PB, Freeman R (1985) Computer-optimized decoupling scheme for wideband applications and low-level operation. J Magn Reson 64:547–552Google Scholar
- Wüthrich K (1986) NMR of proteins and nucleic acids. Wiley, New YorkGoogle Scholar