A simple method for the determination of refractive indices of (rough) transparent solids
- 439 Downloads
Simple methods for the determination of refractive indices of transparent polymers and inorganic and organic solids of irregular geometry or with scratched or corrugated surfaces are rare. A classical procedure is based on the invisibility of a body immersed in a liquid with the same refractive index as that of the body. In order to avoid the laborious procedure connected with the search for a liquid with matching refractive index and to find an approach which is independent of the observation by eye, we describe here a modified immersion method which allows the ready determination of the refractive index of solids. The present method is based on the interpolation of the maximum transmission (nTmax) of a solid immersed in liquids with different, typically non-matching, refractive indices. Illustrations with quartz glass, crown glass and poly(vinylidene fluoride) (PVDF) films showed that nTmax can be determined with a reproducibility of ± 0.003. By comparison with refractive indices determined by ellipsometry, it was concluded that the refractive index of a solid can be determined with the modified immersion method within an accuracy better than ± 0.01 when systematic errors resulting from the fit method are also taken into consideration.
KeywordsQuartz Refractive Index Fluoride Systematic Error PVDF
Unable to display preview. Download preview PDF.
- 1.S. Singh, Phys. Scripta 65 (2002) 167.Google Scholar
- 2.N. G. SULTANOVA, I. D. NIKOLOV and C. D. IVANOV, Opt. Quant. Electronics 35 (2003) 21.Google Scholar
- 3.H. G. TOMPKINS, “A User’s Guide to Ellipsometry” (Academic Press, Boston, 1993).Google Scholar
- 4.RUDOLPH TECHNOLOGIES, http://www.rudolphtech.com/products/auto_el.html (July 30, 2002).
- 5.R. T. KERSTEN, Opt. Commun. 13 (1975) 327.Google Scholar
- 6.T. SHARDA, T. SOGA and T. JIMBO, J. Appl. Phys. 93 (2003) 101.Google Scholar
- 7.METRICON CORP., http://www.metricon.com/feature.htm (May 23, 2002).
- 8.ATOS, http://www.atos-online.de/Atos/Produkte/messtechnik/metricon.htm (April 7, 2002).
- 9.J. R. DE VORE and A. H. PFUND, J. Opt. Soc. Am. 37 (1947) 826.Google Scholar
- 10.H.-J. HöFERT, in “Ullmanns Encyklopädie der technischen Chemie, Band 2/1,” edited by W. Forest (Urban & Schwarzenberg, München, 1961) p. 488.Google Scholar
- 11.ISO 489, “Plastics—Determination of Refractive Index” (1999).Google Scholar
- 12.W. KLEBER, H.-J. BAUTSCH and J. BOHM, “Einführung in die Kristallographie,” 18th ed. (Verlag Technik, Berlin, 1998).Google Scholar
- 13.S. M. OJENA and P. R. DE FOREST, J. Forens. Sci. Soc. 12 (1972) 315.Google Scholar
- 14.B. J. BOZLEE, G. J. EXHAROS, A. E. JIMENEZ and S. L. VAN SWAM, J. Chem. Ed. 79 (2002) 619.Google Scholar
- 15.E. HECHT, “Optik” (R. Oldenbourg, München, 1999).Google Scholar
- 16.Mathematica Optica II, a product under development by Donald Barnhart and Wolfram Research.Google Scholar
- 17.D. R. LIDE (ed.), “Handbook of Chemistry and Physics” (CRC Press, Boca Raton, 1995).Google Scholar
- 18.A. FELDMAN and I. H. MALTISON, in “Recommended Reference Materials for the Realisation of Physicochemical Properties,” edited by K. N. Marsh (Blackwell Scientific Publications, Oxford, 1987).Google Scholar
- 19.D. R. LIDE (Editor in Chief of the CRC Handbook of Chemistry and Physics), Personal Communication (March 20, 2003).Google Scholar
- 20.W. S. RODNEY and R. J. SPINDLER, J. Res. Nation. Bur. Stand. 53 (1954) 185.Google Scholar
- 21.B. BRIXNER, J. Opt. Soc. Am. 57 (1967) 674.Google Scholar
- 22.S. FRANCKE (Custom Glass Department, Erie Scientific Department), Personal Communication (July 11, 2003).Google Scholar
- 23.J. C. SEFERIS, in “Polymer Handbook,” edited by J. Brandrup and E. H. Immergut (Wiley & Sons, New York, 1989).Google Scholar
- 24.A. Dogariu and G. D. Boreman, Opt. Lett. 21 (1996) 701.Google Scholar