Dosing of the second-generation cryoballoon using acute time-to-pulmonary vein isolation as an indicator of durable ablation in a canine model
- 117 Downloads
Rigid time-based dosing protocol(s) currently used in the clinic for cryoballoon ablation of atrial fibrillation may be inadequate to guide the circumferential and transmural cryothermal energy transfer across the pulmonary vein (PV) and may result in injury to collateral tissues or electrical gaps between the PV and left atrium (LA).
A physiologic endpoint (e.g., acute time-to-PV isolation a.k.a. time-to-effect; TTE) may be effective in the determination of a transmural lesion formation and may allow for individualized ablation dosing across each PV.
Thirty PVs from 15 dogs were randomized into five dosing protocols, including (1) TTE + 60 s, (2) TTE + 90 s, (3) TTE + 120 s, (4) TTE + 150 s, and (5) 2 × 180 s. Ablations were conducted with a 23-mm second-generation cryoballoon, and TTE was assessed during a freeze by pacing from an inner balloon-lumen circular diagnostic catheter to a quadripolar diagnostic catheter in the coronary sinus. After ablation, animals were survived for 30 to 34 days, and repeat electrophysiology assessment of PV isolation was conducted after which animals were euthanized for gross anatomy and histological examination.
At study termination, efficacy endpoint evaluations were based on maintenance of PV electrical isolation, gross anatomy assessment of PV lesions, and histological examination of PVs. Five efficacy endpoint failures were noted, including the following: 1 PV in the TTE + 90 sec group; 2 PVs in the TTE + 120 sec group; 1 PV in the TTE + 150 s group; and 1 PV in the 2 × 180 s group. Regarding safety, one phrenic nerve injury was observed in the 2 × 180 s cohort. No other complications were observed.
In a canine model, effective PV isolation could be found even in the shortest duration dosing cohort (TTE + 60 s). One complication (phrenic nerve injury) was observed in the longest duration dosing group (2 × 180 s). Further studies will be required to correlate these results to a 28-mm cryoballoon (more commonly used in the cryoablation of a human LA); however, to date, this is the first reporting of a successful cryoablation using TTE + 60 s dosing (approximately 90 s total duration of freezing).
KeywordsArrhythmia Atrial fibrillation Catheter ablation Cryoablation Cryoballoon Pulmonary vein isolation Time-to-effect
phrenic nerve injury
pulmonary vein isolation
The authors would like to thank Hae Lim for helping in the manuscript preparation.
Compliance with ethical standards
This study was funded by Medtronic, Inc.
Conflict of interest
Dr. Wilber Su is a consultant for Medtronic Inc. Nicolas Coulombe, Nicole Kirchhof, Erin Grassl, and Dan Wittenberger are employees of Medtronic, Inc.
- 7.Miyazaki S, Hachiya H, Nakamura H, Taniguchi H, Takagi T, Hirao K, et al. Pulmonary vein isolation using a second-generation cryoballoon in patients with paroxysmal atrial fibrillation: one-year outcome using a single big-balloon 3-minute freeze technique. J Cardiovasc Electrophysiol. 2016;27(12):1375–80.CrossRefPubMedGoogle Scholar
- 9.Aryana A, Kenigsberg DN, Kowalski M, Koo CH, Lim HW, O'Neill PG, et al. Verification of a novel atrial fibrillation cryoablation dosing algorithm guided by time-to-pulmonary vein isolation: results from the Cryo-DOSING study (Cryoballoon-ablation DOSING based on the assessment of time-to-effect and pulmonary vein isolation guidance). Heart Rhythm. 2017;14(9):1319–25.CrossRefPubMedGoogle Scholar