Cylindrical Acoustic Resonator for the Re-determination of the Boltzmann Constant
- 177 Downloads
The progress towards re-determining the Boltzmann constant k B using two fixed-path, gas-filled, cylindrical, acoustic cavity resonators is described. The difference in the lengths of the cavities is measured using optical interferometry. Thus, a literature value for the density of mercury is not used, in contrast with the presently accepted determination of k B. The longitudinal acoustic resonance modes of a cylindrical cavity have lower quality factors Q than the radial modes of gas-filled, spherical cavities, of equal volume. The lower Qs result in lower signal-to-noise ratios and wider, asymmetric resonances. To improve signal-to-noise ratios, conventional capacitance microphones were replaced with 6.3 mm diameter piezoelectric transducers (PZTs) installed on the outer surfaces of each resonator and coupled to the cavity by diaphragms. This arrangement preserved the shape of the cylindrical cavity, prevented contamination of the gas inside the cavity, and enabled us to measure the longitudinal resonance frequencies with a relative standard uncertainty of 0.2 × 10−6. The lengths of the cavities and the modes studied will be chosen to reduce the acoustic perturbations due to non-zero boundary admittances at the endplates, e.g., from endplate bending and ducts and/or transducers installed in the endplates. Alternatively, the acoustic perturbations generated by the viscous and thermal boundary layers at the gas–solid boundary can be reduced. Using the techniques outlined here, k B can be re-determined with an estimated relative standard uncertainty of 1.5 × 10−6.
KeywordsBoltzmann constant Cylindrical acoustic resonator Two-color interferometry
Unable to display preview. Download preview PDF.
- 6.Moldover M.R., Boyes S.J., Meyer C.W., Goodwin A.R.H.: J. Res. Natl. Inst. Stand. Technol. 104, 11 (1999)Google Scholar
- 12.Moldover M.R., Trusler J.P.M., Edwards T.J., Mehl J.B., Davis R.S.: J. Res. Natl. Bur. Stand. 93, 85 (1988)Google Scholar
- 15.Morse P.M., Ingard K.U.: Theoretical Acoustics, vol. 606. McGraw-Hill Book Co, New York, pp. 554–557 (1968)Google Scholar
- 16.Trusler J.P.M.: Physical Acoustics and Metrology of Fluids. Adam Hilger, IOP Publishing Ltd, Bristol (1991)Google Scholar
- 17.Junger M.C., Feit D.: Sound, Structures, and Their Interaction, pp. 195–234. MIT Press, Cambridge (1986)Google Scholar
- 18.H. Lin, K.A. Gillis, J.T. Zhang, Int. J. Thermophys. (in press, 2010)Google Scholar
- 19.Gillis K.A., Lin H., Moldover M.R.: J. Res. Natl. Inst. Stand. Technol. 114, 263 (2009)Google Scholar
- 23.Trusler J.P.M.: Physical Acoustics and Metrology of Fluids, pp. 44–47. Adam Hilger, IOP Publishing Ltd, Bristol (1991)Google Scholar
- 26.E.D. Palik (ed.), Handbook of Optical Constants of Solids (Academic Press New York 1985), pp. 275–804Google Scholar