Research on the Temperature and Emissivity Measurement of the Metallic Thermal Protection Blanket
- 85 Downloads
The temperatures and emissivities of the metallic thermal protection blanket at (900–1,300) °C are investigated experimentally by using a multi-wavelength pyrometer. A linear relation between the emissivity and true temperature at different wavelengths is assumed. Based on this assumption, the true temperatures and spectral emissivities of the metallic thermal protection blanket at the two temperature measurement points can be calculated simultaneously. Some experimental results for the practical data processing of measurements performed on the metallic thermal protection blanket show that the difference between the calculated temperature and the temperature measured by a standard thermocouple is within ± 10°C.
KeywordsEmissivity Metallic thermal protection blanket Multi-wavelength pyrometer True temperature
Unable to display preview. Download preview PDF.
- 1.M.L. Blosser, R. Chen, in Proceedings of the 40th Aerospace Sciences Meeting & Exhibit (Reno, Nevada, 2002), p. 20Google Scholar
- 2.Blosser M.L.: J. Phys. E 13, 306 (2000)Google Scholar
- 4.Dai J.M.: J. Infrared Millim. Waves 14, 461 (1995)Google Scholar
- 5.Coppa P.: High Temps.-High Press. 22, 479 (1988)Google Scholar
- 7.Dai J.M.: Acta Metrologica Sinica 20, 53 (1999)Google Scholar