Comparison System for the Calibration of Capsule-Type Standard Platinum Resistance Thermometers at NMIJ/AIST
- 77 Downloads
A new comparison system has been constructed using a Gifford- McMahon type cryogenic refrigerator for the calibration of capsule-type standard platinum resistance thermometers (CSPRTs) below 273.16 K at the National Metrology Institute of Japan (NMIJ). The system can compare six CSPRTs at once. A gold-plated comparison block, in which CSPRTs are mounted for calibration, is made from oxygen-free high-conductivity copper. The standard uncertainties related to the temperature control of the system are estimated to be 0.04 mK. The calibrated values for CSPRTs and a rhodium–iron resistance thermometer obtained using the comparison system are in good agreement with those obtained by the direct realization of the low-temperature fixed points of the ITS-90 within the combined standard uncertainty for the calibration using the comparison system.
KeywordsCalibration ITS-90 Low temperature Platinum resistance thermometer Uncertainty
Unable to display preview. Download preview PDF.
- 2.H. Sakurai, O. Tamura, in Proceedings of TEMPMEKO ‘90, 4th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by H.K. Graubner (Multiprint Oy, Helsinki, 1990), pp. 112–117Google Scholar
- 3.H. Sakurai, in Proceedings of TEMPMEKO 2001, 8th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by B. Fellmuth, J. Seidel, G. Scholz (VDE Verlag, Berlin, 2002), pp. 411–416Google Scholar
- 4.A.G. Steele, in Proceedings of Low Temperature Thermometry and Dynamic Temperature Measurement, ed. by A. Szmyrka-Grzebyk (DRUK,Wroclaw, Poland, 1997), pp. L48–L53Google Scholar
- 5.K.D. Hill, A.G. Steele, in Temperature: Its Measurement and Control in Science and Industry, vol. 7, Part 1, ed. by D.C. Ripple (AIP, Melville, New York, 2003), pp. 53–58Google Scholar
- 6.H. Sakurai, in Temperature: Its Measurement and Control in Science and Industry, vol. 7, Part 2, ed. by D.C. Ripple (AIP, Melville, New York, 2003), pp. 969–974Google Scholar
- 7.Nakano T., Tamura O., Sakurai H. (2002). T. SICE 38, 947Google Scholar
- 8.D. Ferri, D. Ichim, F. Pavese, I. Peroni, F. Sparasci, P.P.M. Steur, in Proceedings of TEMPMEKO 2004, 9th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by D. Zvizdić, L.G. Bermanec, T. Veliki, T. Stašić (FSB/LPM, Zagreb, Croatia, 2004), pp. 165–170Google Scholar
- 10.T. Shimazaki, K. Toyoda, O. Tamura, in Proceedings of International Cryogenic Engineering Conference 21, (ICEC 21), Vol. 1, ed. by G. G. Bauger, R. S. Safrata, V. Chrz (Icaris Ltd., Conference Management, Praha, 2007), pp. 629–632Google Scholar
- 11.T. Nakano, O. Tamura, H. Sakurai, in Proceedings of TEMPMEKO 2007, Int. J. Thermophys. 28, 1893 (2007), doi: 10.1007/s10765-007-0289-5
- 12.O. Tamura, S. Takasu, T. Nakano, H. Sakurai, in Proceedings of TEMPMEKO 2007, Int. J. Thermophys., doi: 10.1007/s10765-007-0319-3
- 13.Technical annex for the ITS-90 in the Mise en pratique for the definition of the kelvin, http://www.bipm.fr/en/committees/cc/cct/