Thermal Noise Sensor Insulation Using Ultra-thin Ceramic Coatings
- 60 Downloads
The insulation resistance and interference rejection of several prototype thermal noise sensors were measured. It was found that twisting the connecting wires as a pair optimizes the magnetic interference rejection. High-temperature-resistant wire insulation was developed by vapor depositing a few micrometer thick ceramic coating. The best insulation was obtained with a coating of 1.25μm SiO2 + 1.5μm TiO2 + 0.25μm ZrO2. The insulation of four such coated pairs reached stable levels of 18 ± 3kΩ during a testing period of 1 week at 950°C under atmospheric conditions.
KeywordsCeramic coating Insulation resistance Interference rejection Noise thermometer Twisted pair
Unable to display preview. Download preview PDF.
- 1.D.R. White, R.S. Mason, P. Saunders, in Proceedings of TEMPMEKO 2001, 8th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by B. Fellmuth, J. Seidel, G. Scholz (VDE Verlag, Berlin, 2002), pp. 129–134Google Scholar
- 2.M.J. de Groot, J.F. Dubbeldam, H. Brixy, F. Edler, E. Tegeler, Noise thermometry measurements at fixed points. CCT document CCT/96-30 (1996)Google Scholar
- 5.H. Brixy, R. Hecker. K.F. Rittinghaus, H. Höwener, in Temperature: Its Measurement and Control in Science and Industry, vol. 6, Part 2, ed. by J.F. Schooley (AIP, New York, 1982), pp. 1225–1237Google Scholar
- 6.H. Brixy, in Sensors, A Comprehensive Survey, vol. 4, Thermal Sensors (VCH Verlagsgesellschaft, Weinheim, 1991), pp. 225–251Google Scholar
- 7.E. Zimmermann, W. Glaas, Rauschthermometer Typ 6, Interner bericht des Forschungszentrums Jülich, KFA-ZEL-IB-500399 (June 1999)Google Scholar
- 12.D.R. White, R.S. Mason, P. Saunders, in Proceedings of TEMPMEKO 2004, 9th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by D. Zvizdić, L.G. Bermanec, T. Veliki, T. Stašić (FSB/LPM, Zagreb, Croatia, 2004), pp. 485–490Google Scholar