A Mid-IR Pyrometer Calibrated with High-Temperature Fixed Points for Improved Scale Realization to 2,500°C
- 63 Downloads
Glass surface temperature can be measured using a radiation thermometer operating at a mid-IR wavelength, typically the 3–5 μm band, where the glass is opaque. For optical fiber preforms, the temperature measurement requirement may exceed 2,200°C. Scale realization at national measurement institutes at these temperatures is usually carried out at short wavelengths, typically less than 1 μm. The mismatch in wavelength can lead to significant uncertainties when calibrating a radiation thermometer working at 3–5 μm. To overcome this, a narrow band 3.95 μm radiation thermometer has been built that is designed to be used from 1,000 to 2,500°C. It is calibrated by measurement of high-temperature metal–carbon eutectic fixed-points. The instrument is based on silicon lenses, with a liquid nitrogen (LN2)-cooled InSb detector, and narrow-band interference filter. An anti-reflection coated objective lens/aperture stop focuses onto a field stop giving a 1 mm target, then a collimating lens, and glare stop. All parts visible to the detector, other than the target area, are either at LN2 temperature or are part of a temperature-stabilized housing. A relay-operated shutter that blocks the field stop is used to subtract the background. The size-of-source effect of the instrument has been measured. Gold-point measurements have been made to assess the stability. The device has been calibrated using high-temperature fixed points. A three-parameter fit has been applied and the resultant scale compared to an ITS-90 realization.
KeywordsEutectic fixed point High temperature InSb Radiation thermometer
Unable to display preview. Download preview PDF.
- 1.E. Schreiber, G. Neuer, B. Wörner, in Proceedings of TEMPMEKO '90, 4th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by H.K. Graubner (Multiprint Oy, Helsinki, 1990), pp. 292-306Google Scholar
- 4.D. Lowe, G. Machin, Metal-Carbon Eutectic Alloy High- Temperature (to 2500°C) Fixed-Points as Stable Reference Sources at the National Physical Laboratory, in Proceedings of 11th Int. Metrology Congress, College Francaise de Metrologie, Toulon (2003)Google Scholar
- 5.D. Lowe, H. C. McEvoy, G. Machin, in Temperature, Its Measurement and Control in Science and Industry, ed. by D. Ripple (AIP, New York, 2003), pp. 553- 558Google Scholar
- 6.H. Yoon, C. Gibson, D. Allen, R. Saunders, M. Litorja, S. Brown, G. Eppeldauer, K. Lykke, in Proceedings of TEMPMEKO 2004, 9th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by D. Zvidic (FSB/LPM, Zagreb, Croatia, 2005), pp. 59-71Google Scholar
- 7.E. Theocharous, J. Birch, Detectors for Mid- and Far-infrared Spectrometry: Selection and Use, in Handbook of Vibrational Spectroscopy, ed. by J.M. Chalmers, P.R.Griffiths (John Wiley and Sons, Chichester, 2002)Google Scholar
- 8.M. Sadli, J. Fischer, Y. Yamada, V. Sapritsky, D. Lowe, G. Machin, in Proceedings of TEMPMEKO 2004, 9th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by D. Zvidic (FSB/LPM, Zagreb, Croatia, 2005), pp. 341-348Google Scholar
- 9.F. Sakuma, M. Kobayashi, in Proceedings of TEMPMEKO '96, 6th International Symposium on Temperature and Thermal Measurements in Industry and Science, ed. by P. Marcarino (Levrotto & Bella, Torino, 1997), pp. 305-310Google Scholar