High-Speed Frequency Modulation of a 460-GHz Gyrotron for Enhancement of 700-MHz DNP-NMR Spectroscopy
- 220 Downloads
The high-speed frequency modulation of a 460-GHz Gyrotron FU CW GVI (the official name in Osaka University is Gyrotron FU CW GOI) was achieved by modulation of acceleration voltage of beam electrons. The modulation speed f m can be increased up to 10 kHz without decreasing the modulation amplitude δf of frequency. The amplitude δf was increased almost linearly with the modulation amplitude of acceleration voltage ΔV a. At the ΔV a = 1 kV, frequency spectrum width df was 50 MHz in the case of f m < 10 kHz. The frequency modulation was observed as both the variation of the IF frequency in the heterodyne detection system measured by a high-speed oscilloscope and the widths of frequency spectra df measured on a frequency spectrum analyzer. Both results well agree reasonably. When f m exceeds 10 kHz, the amplitude δf is decreased gradually with increasing f m because of the degradation of the used amplifier in response for high-speed modulation. The experiment was performed successfully for both a sinusoidal wave and triangle wave modulations. We can use the high-speed frequency modulation for increasing the enhancement factor of the dynamic nuclear polarization (DNP)-enhanced nuclear magnetic resonance (NMR) spectroscopy, which is one of effective and attractive methods for the high-frequency DNP-NMR spectroscopy, for example, at 700 MHz. Because the sensitivity of NMR is inversely proportional to the frequency, high-speed frequency modulation can compensate the decreasing the enhancement factor in the high-frequency DNP-NMR spectroscopy and keep the factor at high value. In addition, the high-speed frequency modulation is useful for frequency stabilization by a PID control of an acceleration voltage by feeding back of the fluctuation of frequency. The frequency stabilization in long time is also useful for application of a DNP-NMR spectroscopy to the analysis of complicated protein molecules.
KeywordsGyrotron Second harmonic DNP NMR Sub-THz Frequency modulation
This work was partially supported by Grants in Aid for Scientific Research (B) (No. 24360142) and for Challenging Exploratory Research (No. 25630134) from Japan Society for Promotion of Science (JSPS) and SENTAN project of JST. This work was performed under the Cooperative Research Programs of Institute for Protein Research, Osaka University and Research Center for Development of Far Infrared Region, University of Fukui (FIR UF).
- 1.M. Thumm, State-of-the-Art of High Power Gyro-Devices and Free Electron Masers, KIT Scientific Reports 7662 (2013).Google Scholar
- 8.O. Watanabe, H. Tsuchiya, H. Mori, La Agusu, S. Mitsudo, I. Ogawa,T. Saito, T. Idehara, Development of a Tera Hertz Gyrotron as a Radiation Source, Plasma and Fusion Research: Rapid Communications 2, 024 (2007).Google Scholar
- 9.T. Idehara, T. Tatsukawa,I. Ogawa, Y. Shimizu, N. Nishida and K. Yoshida, Development and applications of submillimeter wave gyrotrons, Proc. 3rd Int. Workshop on Strong Micrawaves in Plasmas, Nizhny Novgorod, 1997, Vol.2, 634–659.Google Scholar
- 11.Y. Bykov, A.F.L. Goldenberg, V. A. Flyagin, The possibilities of material processing by intense millimeter-wave radiation, Mat. Res. Soc. Symp. Proc., 169, 41–42 (1991).Google Scholar
- 14.Yoh Matsuki, Keisuke Ueda, Toshitaka Idehara, Ryosuke Ikeda, Isamu Ogawa, Shinji Nakamura, Mitsuru Toda, Takahiro Anai, Toshimichi Fujiwara, Helium-cooling and -spinning dynamic nuclear polarization for sensitivity-enhanced solid-state NMR at 14 T and 30 K, Journal of Magnetic Resonance 225, 1–9 (2012).CrossRefGoogle Scholar
- 15.Kevin J. Pike, Thomas F. Kemp, Hiroki Takahashi, Robert Day, Andrew P. Howes, Eugeny V. Kryukov, James F. MacDonald, Alana E. C. Collis, David R. Bolton, Richard J. Wylde, Marcella Orwick, Kosuke Kosuga, Andrew J. Clark, Toshitaka Idehara, Anthony Watts, Graham M. Smith, Mark E. Newton, Ray Dupree, Mark E. Smith, A spectrometer designed for 6.7 and 14.1 T DNP-enhanced solid-state MAS NMR using quasi-optical microwave transmission, Journal of Magnetic Resonance 215, 1–9 (2012).CrossRefGoogle Scholar
- 16.S. Alberti, J. Ph. Ansermet, F. Braunmuller, P. Cuanillon, J. Dubray, D. Fasel, J. Ph. Hogge, A. Macor, E. de Rijk, M.Q. Tran, T.M. Tran, Q. Vuillemin, Experimental results on a modular gyrotron operating at 0.26 THz for 400 MHz DNP/NMR spectroscopy application, 38th Int. Conf. on Infrared. Millimeter and Terahertz Waves (IRMMW-THz 2012), Wollongong, Australia, Sept 23–28, 2012, Thu-A-3.Google Scholar
- 24.T. Idehara, Y. Tatematsu, Y. Yamaguchi, E.M. Khutoryan, A.N. Kuleshov, K. Ueda,Y. Matsuki and T. Fujiwara, Development of 460 GHz gyrotrons for 700 MHz DNP-NMR spectroscopy, J. Infrared, Millimeter and Terahertz Waves, under review.Google Scholar