The Herschel-PACS photometer calibration
- 407 Downloads
This paper provides an overview of the PACS photometer flux calibration concept, in particular for the principal observation mode, the scan map. The absolute flux calibration is tied to the photospheric models of five fiducial stellar standards (α Boo, α Cet, α Tau, β And, γ Dra). The data processing steps to arrive at a consistent and homogeneous calibration are outlined. In the current state the relative photometric accuracy is ∼2 % in all bands. Starting from the present calibration status, the characterization and correction for instrumental effects affecting the relative calibration accuracy is described and an outlook for the final achievable calibration numbers is given. After including all the correction for the instrumental effects, the relative photometric calibration accuracy (repeatability) will be as good as 0.5 % in the blue and green band and 2 % in the red band. This excellent calibration starts to reveal possible inconsistencies between the models of the K-type and the M-type stellar calibrators. The absolute calibration accuracy is therefore mainly limited by the 5 % uncertainty of the celestial standard models in all three bands. The PACS bolometer response was extremely stable over the entire Herschel mission and a single, time-independent response calibration file is sufficient for the processing and calibration of the science observations. The dedicated measurements of the internal calibration sources were needed only to characterize secondary effects. No aging effects of the bolometer or the filters have been found. Also, we found no signs of filter leaks. The PACS photometric system is very well characterized with a constant energy spectrum νF ν = λF λ = const as a reference. Colour corrections for a wide range of sources SEDs are determined and tabulated.
KeywordsHerschel space observatory PACS Far-infrared Instrumentation Calibration Scan-map
We would like to thank the anonymous referee for the comments and suggestions that significantly improved the manuscript. Z. Balog, H. Linz and M. Nielbock are funded by the Deutsches Zentrum für Luft- und Raumfahrt e. V.
- 2.Billot, N., et al.: The Herschel-PACS 2560 bolometers imaging camera. SPIE Proc. 6265(62650D), 12 (2006)Google Scholar
- 3.Cohen, M., et al.: Stellar calibration in the infrared: extending the lagacy of the KAO, ISO, and MSX to SIRTF and beyond. In: Proceedings of the Conference The Calibration Legacy of the ISO Mission, ESA SP-481, vol. 135 (2003)Google Scholar
- 7.Ducati, J.R.: Catalogue of stellar photometry in Johnson’s 11-color system. In: CDS/ADC Collection of Electronic Catalogues, vol. 2237 (2002)Google Scholar
- 10.Gustafsson, B., et al.: A grid of model atmospheres for cool stars. In: Proceedings of the Conference Stellar Atmosphere Modeling, ASP Conference Series, vol. 288, p. 331 (2003)Google Scholar
- 11.Moór, e.t.al.: PACS photometer calibration block analysis. Exp. Astron. (2013)Google Scholar
- 12.Moreno, R.: Neptune and Uranus planetry brightness temperature tabulation. Tech. rep., ESA Herschel Science Centre. available from ftp://ftp.sciops.esa.int/pub/hsc-calibration/PlanetrayModels/ESA4/
- 13.Müller, T., et al.: Asteroid calibration. Exp. Astron. (2013)Google Scholar
- 14.Nielbock, M., et al.: The Herschel-PACS photometer calibration: a time dependent flux calibration for the PACS chopped photometry AOT mode. Exp. Astron. (2013)Google Scholar
- 18.Popesso, P., et al.: The effect of the high-pass filter data reduction technique on the Herschel-PACS photometer PSF and noise. A&A (2013). arXiv:1211.4257