Fiber optic light sensor
- 280 Downloads
We describe a low-cost fiber optic sensor for measuring photosynthetically active radiation (PAR) in turbulent flow. Existing technology was combined in a novel way for probe development addressing the need for a small but durable instrument for use in flowing water. Optical components including fiber optics and a wide-spectrum light detector were used to separate light collection from electronic detection so that measurements could be completed in either the field or laboratory, in air or underwater. Connection of the detector to Arduino open-source electronics and a portable personal computer (PC) enabled signal processing and allowed data to be stored in a spreadsheet for ease of analysis. Calibration to a commercial cosine-corrected instrument showed suitable agreement with the added benefit that the small sensor face allowed measurements in tight spaces such as close to the streambed or within leafy or filamentous plant growth. Subsequently, we applied the probe in a separate study where over 35 experiments were successfully completed to characterize downward light attenuation in filamentous algae in turbulent flow.
KeywordsFiber optic sensor Photosynthetically active radiation PAR Light Meter Optics
The authors wish to thank the Montana-Wyoming USGS Science Center for loan of their Li-COR 192 m used in support of this work. The authors also wish to thank Kevin Chudyk for layout, assembly, and testing of the photodiode detector breakout board circuitry. Portions of this work were presented at SPIE Photonics Europe Innovation Village April 13–18, 2014.
- Crisp, J., & Elliot, B. (2005). Introduction to fiber optics (3rd ed.). Boston: Elsevier.Google Scholar
- Fielder, P. & Comeau, P. (2000). Construction and testing of an inexpensive PAR sensor. Research Branch British Columbia Ministry of Forests Working paper 53, Victoria, B.C. ISBN: 0-7726-4392-X.Google Scholar
- Flynn, K. F. (2014). Methods and mathematical approaches for modeling Cladophora glomerata and river periphyton, Ph.D. thesis, Dept. of Civil & Env. Eng., Tufts University, Medford, MA.Google Scholar
- Flynn K. F., Chudyk, W. A., Chapra, S. C., & Watson, V. (2014). Influence of biomass and velocity on light attenuation of Cladophora glomerata L. (Kuetzing) in rivers, in preparation.Google Scholar
- Fox, R. W., Pritchard, P. J., & McDonald, A. T. (2009). Introduction to fluid mechanics (7th ed.). Hoboken: Wiley. ISBN 9780471742999.Google Scholar
- Jerlov, N. G. (1968). Optical oceanography. New York: Elsevier Publishing Company. ISBN 978-0444403209.Google Scholar
- Kühl, M. (2005). Optical microsensors for analysis of microbial communities. Methods in Enzymology, 397, 166–199. doi: 10.1016/S0076-6879(05)97010-9.
- Kühl, M., Lassen, C., & Revsbech, N. P. (1997). A simple light meter for measurements of PAR (400 to 700 nm) with fiber-optic microprobes: application for P vs E-0(PAR) measurements in a microbial mat. Aquatic Microbial Ecology, 13, 197–207. doi: 10.3354/ame013197.
- LI-COR (2011). Radiation measurement instruments, www.licor.com. Accessed 2/14/2014.
- Parallax, Inc. (2014). PLX-DAQ, http://www.parallax.com/downloads/plx-daq. Accessed 21 Feb 2014.
- Pérez, G. L., Lagomarsino, L., & Zagarese, H. E. (2013). Optical properties of highly turbid shallow lakes with contrasting turbidity origins: the ecological and water management implications. Journal of Environmental Management, 130, 207–220. doi: 10.1016/j.jenvman.2013.09.001.CrossRefGoogle Scholar
- Taos (2007). TSL230RD, TSL230ARD, TSL230BRD Programmable light-to-frequency converters, www.taosinc.com. Accessed 9/7/2012.
- Thomson, J. (2010). Getting started with the TSL230R and Arduino, http://jethomson.wordpress.com/tsl230r-articles/getting-started-with-the-tsl230r-and-arduino. Accessed 21 Feb 2014.