Factors affecting computed tomography image quality for assessment of mechanical aortic valves
- 161 Downloads
Evaluating mechanical valves with computed tomography (CT) can be problematic because artifacts from the metallic components of valves can hamper image quality. The purpose of this study was to determine factors affecting the image quality of cardiac CT to improve assessment of mechanical aortic valves. A total of 144 patients who underwent aortic valve replacement with mechanical valves (ten different types) and who underwent cardiac CT were included. Using a four-point grading system, the image quality of the CT scans was assessed for visibility of the valve leaflets and the subvalvular regions. Data regarding the type of mechanical valve, tube voltage, average heart rate (HR), and HR variability during CT scanning were compared between the non-diagnostic (overall image quality score ≤2) and diagnostic (overall image quality score >2) image quality groups. Logistic regression analyses were performed to identify predictors of non-diagnostic image quality. The percentage of valve types that incorporated a cobalt-chrome component (two types in total) and HR variability were significantly higher in the non-diagnostic image group than in the diagnostic group (P < 0.001 and P = 0.013, respectively). The average HR and tube voltage were not significantly different between the two groups (P > 0.05). Valve type was the only independent predictor of non-diagnostic quality. The CT image quality for patients with mechanical aortic valves differed significantly depending on the type of mechanical valve used and on the degree of HR variability.
KeywordsCardiac computed tomography Mechanical valves Aortic valve replacement Image quality Valve type
Compliance with ethical standards
Conflict of interest
All authors declare that there are no conflicts of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Informed consent was waived by approval of the Institutional Review Board of Severance Hospital.
- 4.Tsai IC, Lin YK, Chang Y, Fu YC, Wang CC, Hsieh SR, Wei HJ, Tsai HW, Jan SL, Wang KY, Chen MC, Chen CC (2009) Correctness of multi-detector-row computed tomography for diagnosing mechanical prosthetic heart valve disorders using operative findings as a gold standard. Eur Radiol 19(4):857–867CrossRefPubMedGoogle Scholar
- 8.Park YJ, Kim YJ, Lee JW, Kim HY, Hong YJ, Lee HJ, Hur J, Nam JE, Choi BW (2012) Automatic tube potential selection with tube current modulation (APSCM) in coronary CT angiography: comparison of image quality and radiation dose with conventional body mass index-based protocol. J Cardiovasc Comput Tomogr 6(3):184–190CrossRefPubMedGoogle Scholar
- 18.Chenot F, Montant P, Goffinet C, Pasquet A, Vancraeynest D, Coche E, Vanoverschelde JL, Gerber BL (2010) Evaluation of anatomic valve opening and leaflet morphology in aortic valve bioprosthesis by using multidetector CT: comparison with transthoracic echocardiography. Radiology 255(2):377–385CrossRefPubMedGoogle Scholar
- 19.Horiguchi J, Fujioka C, Kiguchi M, Yamamoto H, Kitagawa T, Kohno S, Ito K (2009) Prospective ECG-triggered axial CT at 140-kV tube voltage improves coronary in-stent restenosis visibility at a lower radiation dose compared with conventional retrospective ECG-gated helical CT. Eur Radiol 19(10):2363–2372CrossRefPubMedGoogle Scholar