Advances in cardiac CT imaging: 64-slice scanner
Background: Clinical progress by the development of multi-slice CT (MSCT) technology beyond 16 slices can more likely be expected from further improved spatial and temporal resolution rather than from a mere increase in the volume coverage speed. We present an evaluation of a recently introduced 64-slice CT (64SCT) system, which makes use of a periodic motion of the focal spot in the longitudinal direction (z-flying focal spot) to double the number of simultaneously acquired slices. Materials and Methods: A recently introduced 64SCT system (SOMATOM Sensation 64, Siemens Medical Solutions, Forchheim, Germany) is being described and tested in first clinical practice, applying the following parameters: z-flying focal spot technology, 64×0.6 mm slices; spatial resolution, 0.4 × 0.4 × 0.4 mm; gantry rotation time, 330 ms; temporal resolution, 83–165 ms. Various phantom studies and first clinically implemented protocols are being described, to evaluate the full spectrum of possible applications for this scanner type, with a focus on cardiac imaging. Results:ECG-gated cardiac scanning with this 64-slice CT system benefits clearly from both the improved temporal resolution and improved spatial resolution. These benefits enable a more reliable assessment of mixed plaques, due to reduced partial-voluming and beam-hardening artefacts caused by calcifications, and holds great promise for the reliable assessment of in-stent stenoses, as stent lumen visibility is clearly improved as compared to earlier MSCT systems. With the increased volume coverage and acquisition speed of the 64SCT system, a comprehensive emergency protocol of the thorax becomes feasible within an acceptable breath-hold time, performing an ECG-gated CT angiography of the complete thoracic vasculature. This protocol enables a detailed assessment of the thoracic aorta, the pulmonary arteries and the coronary arteries in one single examination. Conclusion:64SCT Cardiac imaging provides an increased spatial resolution with an isotropic voxel size of 0.4 mm and an improved temporal resolution of 83–165 ms. These benefits hold great promise especially for fast-moving organs requiring detailed imaging, such as the heart and coronary arteries.
KeywordsTemporal Resolution Volume Coverage Stent Lumen Isotropic Voxel Size Mixed Plaque
Unable to display preview. Download preview PDF.
- Ohnesorge, B, Flohr, T, Becker, C, et al. 2000Cardiac imaging by means of electrocardiographically gated multisection spiral CT: initial experienceRadiology217(2)564571Google Scholar
- Fayad, ZA, Fuster, V, Nikolaou, K, Becker, C 2002Computed tomography and magnetic resonance imaging for noninvasive coronary angiography and plaque imaging: current and potential future conceptsCirculation10620262034Google Scholar
- Flohr, T, Bruder, H, Stierstorfer, K, Simon, J, Schaller, S, Ohnesorge, B 2002New technical developments in multislice CT, part 2: sub-millimeter 16-slice scanning and increased gantry rotation speed for cardiac imagingRofo Fortschr Geb Rontgenstr Neuen Bildgeb Verfahr17410221027CrossRefPubMedGoogle Scholar
- Dewey, M, Laule, M, Krug, L, et al. 2004Multisegment and halfscan reconstruction of 16-slice computed tomography for detection of coronary artery stenosesInvest Radiol39223229Google Scholar
- Nieman, K, Cademartiri, F, Lemos, P, Raaijmakers, R, Pattynama, P, Feyter, P 2002Reliable noninvasive coronary aniography with fast submillimeter multslice spiral computed tomographyCirculation10620512054Google Scholar
- Ropers, D, Baum, U, Pohle, K, et al. 2003Detection of coronary artery stenoses with thin-slice multi-detector row spiral computed tomography and multiplanar reconstructionCirculation107664666Google Scholar
- Nikolaou, K, Becker, CR, Muders, M, Loehrs, U, Reiser, M, Fayad, ZA 2004High resolution magnetic resonance and multi-slice CT imaging of coronary artery plaques in human ex vivo coronary arteriesAtherosclerosis174243252Google Scholar